- Meeting abstract
- Open Access
1141 Can every other heart-beat acquisition be better for the same breath-hold interval in MR tagging sequences?
Journal of Cardiovascular Magnetic Resonance volume 10, Article number: A266 (2008)
In cardiac MRI with segmented k-space acquisition, data is partially acquired through consecutive cardiac cycles. However, the available longitudinal magnetization in a given cardiac cycle is exhausted by the excitation radiofrequency pulses in previous cycles, which limits the available signal-to-noise ratio (SNR). One potential solution is to increase the repetition time (TR) to allow for partial magnetization recovery between the RF pulses. However, this solution is not practical in sequences with breath-hold acquisition as it increases the breath-hold time, which limits the applicability of the considered sequence. In this work, a new technique is proposed for improving the SNR without increasing the breath-hold time. The idea is based on acquiring the signal every other heartbeat within the same breath-hold time interval. This limits the number of the acquired k-space segments to a half but allow the magnetization a longer time to recover. This technique results in better SNR, compared to conventional (every heartbeat) segmented k-space acquisition, as evidenced by numerical simulations of the Bloch equation as well as phantom and in-vivo experiments. Two tagging pulse sequences are considered in this work: complementary-spatial-modulation-of-magnetization (CSPAMM)  and strain-encoding (SENC) MRI .
Numerical simulations of the Bloch equation were conducted to examine the effect of the imaging flip angle and TR on the resulting signal amplitude. Ramped flip angle scheme was implemented [1, 3]. The simulations were conducted for both every heartbeat and every-other heartbeat acquisition schemes.
To verify the simulations, a phantom experiment was conducted using a spiral acquisition . Two sets of CSPAMM and SENC images were acquired using the same breath-hold intervals. In the first set, the acquisition was done every heartbeat while for the second set it was every-other heartbeat. To compensate for the number of spiral trajectories that was halved in the second set, two modifications were done. First, the spiral acquisition window time was doubled. Second, a multi-shot acquisition with factor 2 per cardiac phase was used.
In vivo studies
Two normal volunteers were consented and scanned using the same sequences. SNR was measured from the images by defining a region of interest in the static tissue and the heart. The noise was measured from the background and adjusted for Rayleigh distribution.
Results and discussion
Simulation results are shown in Figs. 1 and 2. Fig. 3 shows the SNR gain of the every other heartbeat acquisition over every heartbeat acquisition. Fig. 3.a shows the gain in the phantom study using the two mentioned maneuvers. Fig. 3.b shows the SNR gain in-vivo using the first maneuver only. The results show that the every-other heartbeat acquisition performs better than every heartbeat acquisition. However, the first maneuver has a substantial increase in SNR (≈180%) at the nominal flip angle values (< 40° due to the SAR limitations during actual scans). Finally, Fig. 4 shows short-axis cardiac images acquired with every- and every-other heartbeat for different flip angles. We can easily distinguish the SNR enhancement in both the myocardium and the background noise.
In contrast to what is usually performed in routinely conducted CSPAMM and SENC studies, we have demonstrated that for the same breath-hold time interval, the every-other heartbeat acquisition gives substantially higher SNR than every heart-beat acquisition. This should be taken into consideration for routinely conducted studies to improve the quality of the resulting images.
Fischer : MRM. 30: 191-200.
Osman : MRM. 46: 324-334.
Stuber : MAGMA. 9: 85-91. 10.1007/BF02634597.
Ryf : MRM. 51: 237-242.
This work was supported in part by grants from the national institute of heart, lung and blood R01HL072704 and R01HL61912, and Donald W. Reynolds Foundation grant.
Rights and permissions
Open Access This article is published under license to BioMed Central Ltd. This is an Open Access article is distributed under the terms of the Creative Commons Attribution 2.0 International License (https://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Basha, T., Ibrahim, ES., Shehata, M.L. et al. 1141 Can every other heart-beat acquisition be better for the same breath-hold interval in MR tagging sequences?. J Cardiovasc Magn Reson 10 (Suppl 1), A266 (2008). https://doi.org/10.1186/1532-429X-10-S1-A266
- Flip Angle
- Bloch Equation
- Phantom Experiment
- Spiral Acquisition
- Excitation Radiofrequency