- Technical notes
- Open Access

# Comparison of 2D and 3D calculation of left ventricular torsion as circumferential-longitudinal shear angle using cardiovascular magnetic resonance tagging

- Iris K Rüssel
^{1, 2}Email author, - Sandra R Tecelão
^{3}, - Joost PA Kuijer
^{1}, - Robert M Heethaar
^{1}and - J Tim Marcus
^{1}

**11**:8

https://doi.org/10.1186/1532-429X-11-8

© Rüssel et al; licensee BioMed Central Ltd. 2009

**Received: **19 December 2008

**Accepted: **20 April 2009

**Published: **20 April 2009

## Abstract

### Purpose

To compare left ventricular (LV) torsion represented as the circumferential-longitudinal (CL) shear angle between 2D and 3D quantification, using cardiovascular magnetic resonance (CMR).

### Methods

CMR tagging was performed in six healthy volunteers. From this, LV torsion was calculated using a 2D and a 3D method. The cross-correlation between both methods was evaluated and comparisons were made using Bland-Altman analysis.

### Results

The cross-correlation between the curves was *r*^{2} = 0.97 ± 0.02. No significant time-delay was observed between the curves. Bland-Altman analysis revealed a significant positive linear relationship between the difference and the average value of both analysis methods, with the 2D results showing larger values than the 3D. The difference between both methods can be explained by the definition of the 2D method.

### Conclusion

LV torsion represented as CL shear quantified by the 2D and 3D analysis methods are strongly related. Therefore, it is suggested to use the faster 2D method for torsion calculation.

## Keywords

- Cardiovascular Magnetic Resonance
- Shear Angle
- Left Ventricular Twist
- Left Ventricular Torsion
- Significant Positive Linear Relationship

## Background

Left ventricular (LV) torsion is a sensitive marker for both systolic and diastolic dysfunction [1–3], and is therefore a useful addition to other strain measures such as radial, circumferential and longitudinal strain. LV torsion can be assessed using several techniques like speckle tracking echocardiography [4, 5] and cardiovascular magnetic resonance (CMR) myocardial tagging [6, 7]. However, there is still a debate on how to describe and calculate torsion in an optimal way, as a gold standard is not yet available. A straightforward determination method for LV torsion will facilitate clinical use of this measure.

Calculating the CL shear angle in this way, is a fast approach that can be applied using only 2D short-axis (SA) image data. A method closer to the true CL shear angle, is the calculation of the local CL shear angle from extensive 3D strain analysis. However, this requires a dataset with SA and long-axis (LA) image data, from which 3D information on displacement of myocardium can be extracted. The additional acquisition time of the LA images and the large amount of additional post-processing time make 3D strain analysis time-consuming and less usable in clinical practice.

In this study, the results for CL shear strain from the 2D calculation and the 3D strain analysis methods are compared in healthy volunteers. The results will indicate whether it is sufficient to calculate torsion from only 2D SA CMR images.

## Methods

### Subjects

Six healthy male volunteers (26–56 years old, mean age: 43 years, ejection fraction: 56 ± 6%) with no history of cardiac disease were studied. Informed consent was obtained according to our institutional guidelines.

### Image acquisition

^{2}, the excitation flip angle 20°, repetition time (TR) = 4.7 ms, echo time (TE) = 2.3 ms, receiver bandwidth (BW) = 369 Hz/pixel, imaging matrix size = 256 × 78. Five SA slices, evenly distributed over the LV, as seen on an end-systolic 4-chamber image, were acquired with both horizontal and vertical tagging. For the 3D analysis, three additional LA planes uniformly distributed around the LV and perpendicular to the SA with the tagging direction parallel to the SA slices were acquired. Tag line distance was equal to 7 mm.

### Post processing

Harmonic magnitude (HARM) and harmonic phase (HARP) images were computed from the SA and LA CSPAMM images as described in [12]. LV endocardial and epicardial contours were drawn on the HARM images using a dedicated software package (MASS, Medis, Leiden, the Netherlands). The myocardial tissue inside the contours was tracked by applying the previously described automatic extended HARP tracking method to the HARP images [13].

### Calculation of the CL shear angle using 2D analysis

*Φ*) was averaged over the entire myocardium and set relative to the first timeframe. Two-dimensional torsion (

*T*

_{2D}) between two slices can then be calculated as follows:

*D*is the distance between the slices and

*ρ*the radius, calculated from the average pixel location inside the contours. In this way, the torsion can be interpreted as the global CL shear angle. The approach described in Eq. [1] has been previously applied by e.g. Aelen et al. and Delhaas et al. [10, 14]. Another approach is to calculate the difference in circumferential displacement directly, which is geometrically closer to the 3D definition of shear angle (Fig. 1):

generally resulting in larger values for *T*_{2D}, since in the normal situation, (*ϕ*_{apex} + *ϕ*_{base}) and (*ρ*_{apex} - *ρ*_{base}) will be positive.

Both definitions of torsion (Eqs. [1, 2]) use the small angle approximation (tan *x* ≈ *x*) for calculation of the shear angle and both will be used for comparison with 3D analysis. CL shear will be calculated at three levels: between base and apex, between base and mid, and between mid and apex.

### Calculation of the CL shear angle using 3D analysis

*E*was computed in RCL-coordinates with the knowledge of the displacements of the points, from which the CL shear angle (

*α*

_{CL}) could be computed:

where *E*_{ii} are diagonal elements and *E*_{ij} are off-diagonal elements of the strain tensor, *ε* is the axial strain and i and j are indices of the circumferential and longitudinal direction, respectively. Therefore, three-dimensional torsion (*T*_{3D}) is defined as: *T*_{3D} = *α*_{CL}.

The results were averaged over the entire 5 slices for base-apex torsion, the entire top 3 slices for base-mid torsion and the entire bottom 3 slices for mid-apex torsion to obtain a measure equivalent to that obtained with the 2D analysis.

### Comparisons and statistics

The 2D and 3D shear angle curves were compared to evaluate the difference between the two analysis methods. First, cross-correlations were calculated between the torsion curves obtained with both methods (base-apex, base-mid and mid-apex). The 2D and 3D curves were shifted with respect to each other. One time-lag was defined as the time between two subsequent cardiac phases (14 ms) as acquired during tagged CMR. Furthermore, paired T-tests and Bland-Altman analysis was performed on the global torsion curves. Limits of agreement for the Bland-Altman analysis were calculated using linear regression [17] in case of comparison with *T*_{2D}.

Results are presented as mean ± SD. P-values below 0.05 are regarded as statistically significant.

## Results

### Cross-correlations

*T*

_{2D},

*T**

_{2D}and

*T*

_{3D}calculation methods is presented in Fig. 3. The average maximum cross-correlation over all regions (base-apex, base-mid and mid-apex) and subjects between the

*T*

_{2D}and the

*T*

_{3D}was high (r

^{2}= 0.97 ± 0.02, see Table 1) and there was no time delay between the curves. The highest correlation between the two calculation methods was obtained for the base-apex torsion (r

^{2}= 0.99 ± 0.01).

r^{2} values and time-delays for the cross-correlations between the 2D and the 3D torsion calculation method.

r | r | Time delay (time lags) ( | |
---|---|---|---|

| 0.99 ± 0.01 | 0.98 ± 0.02 | 0.0 ± 0.0 |

| 0.97 ± 0.02 | 0.96 ± 0.03 | 0.0 ± 0.0 |

| 0.95 ± 0.02 | 0.93 ± 0.06 | 0.0 ± 0.0 |

Average | 0.97 ± 0.02 | 0.96 ± 0.04 | 0.0 ± 0.0 |

When using *T**_{2D}, the cross-correlation was slightly lower (r^{2} = 0.96 ± 0.04, see Table 1), although the time delay between the curves was still zero.

### Comparison of 2D and 3D torsion

*T*

_{2D}were significantly (p < 0.0001) higher than those obtained with

*T*

_{3D}(Table 2). Bland-Altman analysis revealed a significant positive linear relationship between the difference in torsion and the average torsion (Fig. 4, Table 2). This linear relationship was observed in all curves (base-apex: r = 0.67; base-mid: r = 0.71; mid-apex: r = 0.69; all p < 0.0001). The limits of agreement are therefore calculated as a regression line [17] and can be found in Table 2.

Comparison between the 2D and 3D torsion curves, top values: *T*_{2D} as 2D method, bottom values: *T**_{2D} as 2D method.

Average values (3D vs. 2D) (deg) | Regression line (3D vs. 2D) | Correlation coefficient (Difference vs. Average) | Limits of agreement regression line (top) (Difference vs. Average), Limits of agreement (bottom) (Difference) | |
---|---|---|---|---|

| 2.9 ± 1.9 vs. 3.4 ± 2.3 | y = 0.77*x + 0.21 | r = 0.67 | y = 0.23*x - 0.15 ± 1.01 |

2.9 ± 1.9 vs. 2.9 ± 2.1 | y = 0.86*x + 0.38 | r = 0.39 | 0.03 ± 1.14 | |

| 2.9 ± 1.9 vs. 3.7 ± 2.6 | y = 0.70*x + 0.32 | r = 0.71 | y = 0.32*x - 0.23 ± 1.37 |

2.9 ± 1.9 vs. 3.4 ± 2.4 | y = 0.75*x + 0.32 | r = 0.56 | 0.51 ± 1.71 | |

| 2.8 ± 1.8 vs. 3.5 ± 2.4 | y = 0.71*x + 0.31 | r = 0.69 | y = 0.30*x - 0.23 ± 1.31 |

2.8 ± 1.8 vs. 2.4 ± 2.0 | y = 0.82*x + 0.81 | r = 0.26 | -0.39 ± 1.61 |

*T**

_{2D}, the difference between the 2D and 3D method between the base-apex curves was no longer significant (p = 0.35). For the base-mid curves,

*T**

_{2D}was significantly higher (p < 0.0001) and for the mid-apex curves

*T**

_{2D}was significantly lower (p < 0.0001). In the Bland-Altman analysis, only weak correlations were found between the difference and the average of both methods (Fig. 5, Table 2); therefore the limits of agreement were no longer calculated as a regression line. Limits of agreement are slightly increased using

*T**

_{2D}, however (Table 2).

## Discussion

### Comparison of 2D and 3D measurement methods

This study shows that there is a high correlation between global torsion calculated as the CL shear angle using the 2D and the 3D methods, and that the curves are not delayed with respect to each other. However, when the 2D CL shear is calculated as the LV twist per unit length, multiplied by the average radius (*T*_{2D}, Eq. [1]), the difference between the 2D and 3D method increases as the torsion value increases. *T*_{2D} gives the highest values. When the 2D CL shear is calculated as the difference in circumferential displacement per unit length (*T**_{2D}, Eq. [2]), the linear trend in the difference is much smaller.

Considering that *T*_{3D} is obtained from information on local deformations following from 3D displacements, the *T*_{3D} is expected to be closest to the true torsion.

A strong relationship and no time delay between curves obtained with the 2D and the 3D method was found.

For *T**_{2D}, no linear relationship between the difference and average of the 2D and 3D method was observed. However, only when torsion was calculated between the basal and apical plane, was there no bias between both methods. For the other two longitudinal regions (base-mid, mid-apex), respectively an overestimation and an underestimation of torsion were found with respect to *T*_{3D}. For *T*_{2D} on the other hand, the difference between the 2D and 3D methods was relatively constant over the longitudinal regions.

This might be explained by the difference between *T*_{2D} and *T**_{2D}. Errors in the radius of the LV might be introduced by including too many trabeculae in the contours (underestimation of radius, usually on apical level), or by including pericardial fat in the contours (overestimation). Contours in tagged images are difficult to delineate, because of the low resolution of the HARM images. An overestimation of the radius in the basal slice or an underestimation of the radius in the apical slice will both lead to an overestimation of *T*_{2D}, regarding the (*ρ*_{base} - *ρ*_{apex}) term from Eq. [3], causing the relatively constant overestimation of torsion over the longitudinal regions. From error-propagation (assuming similar variance in radii, and constant *D* and *Φ*) it can be derived that the variance in *T**_{2D} is larger than in *T*_{2D}. Under- or overestimations of radii are directly reflected in the torsion value, but less so in *T**_{2D}, which also explains the slightly lower cross-correlation found with this method (Table 1).

Several other aspects might explain the differences between the 2D and 3D methods. Since the tetrahedrons in the 3D analysis can only be defined in the intersecting region of the contours of neighbouring image planes, the 3D analysis represents less myocardium than the 2D analysis [15]. The difference between the methods is probably not explained by the difference in the amount of myocardium used for averaging in the 2D and 3D methods, since the CL shear angle is thought to be constant over the transmural direction of the myocardial wall [8, 18]. However, the difference might be an explanation for the somewhat reduced correlation at the apical level between the 2D and 3D methods (Table 1).

Since *T*_{2D} is less sensitive to errors in the radius of the ventricle, its bias to *T*_{3D} is relatively constant, and it has been often used in literature, this is probably the best method to calculate 2D CL shear. Furthermore, when reference data obtained with the same method is present, the observed deviation between *T*_{2D} and *T*_{3D} is less important.

### Clinical implications

Torsion calculated as the 2D CL shear angle is very fast, both in acquisition as in post-processing, as compared to the 3D analysis. Acquisition of a horizontally and vertically tagged CSPAMM CMR slice requires up to 5 minutes with the protocol that was used in this study. Post-processing for the 2D method is fully automatic, except for the contours that have to be drawn manually. The calculation part of the post-processing requires only a few minutes on a standard PC. The main drawbacks of the 3D analysis are that additional images have to be acquired, and that the accompanying contours need to be drawn. These extra LA images also require additional post-processing.

Both 2D methods show strong (cross-)correlation with *T*_{3D}. The constant bias and narrower limits of agreement of *T*_{2D}, together with the fact that *T*_{2D} has already been used more often in literature, providing more reference data on torsion calculated in this way [8, 10, 19], demonstrates that this measure is suitable to be used in clinical practice.

### Limitations

In this study, no patients were included. It is known that torsion can be altered in several ways in patients with different diseases [2, 20–24]. Therefore, comparisons should be made in patients with different alterations in torsion. This might be a topic of future investigation.

## Conclusion

Global LV torsion represented as the CL shear angle quantified by a 2D method and a 3D method show a very strong relationship. Observed differences between both methods can be explained by the definition of the 2D method. Consequently, it is suggested to use the faster and easier 2D method for calculation of global LV torsion.

## Declarations

## Authors’ Affiliations

## References

- Delhaas T, Kotte J, Toorn Van der A, Snoep G, Prinzen FW, Arts T: Increase in Left Ventricular Torsion-to-Shortening Ratio in Children With Valvular Aortic Stenosis. Magn Res Med. 2004, 51: 135-139. 10.1002/mrm.10679.View ArticleGoogle Scholar
- Nagel E, Stuber M, Lakatos M, Scheidegger MB, Boesiger P, Hess OM: Cardiac rotation and relaxation after anterolateral myocardial infarction. Coron Artery Dis. 2000, 11: 261-267. 10.1097/00019501-200005000-00009.View ArticlePubMedGoogle Scholar
- Paetsch I, Föll D, Kaluza A, Luechinger R, Stuber M, Bornstedt A, Wahl A, Fleck E, Nagel E: Magnetic resonance stress tagging in ischemic heart disease. Am J Physiol: Heart Circ Physiol. 2005, 288: 2708-2714. 10.1152/ajpheart.01017.2003.Google Scholar
- Kim HK, Sohn DW, Lee SE, Choi SY, Park JS, Kim YJ, Oh BH, Park YB, Choi YS: Assessment of left ventricular rotation and torsion with two-dimensional speckle tracking echocardiography. J Am Soc Echocardiogr. 2007, 20: 45-53. 10.1016/j.echo.2006.07.007.View ArticlePubMedGoogle Scholar
- Takeuchi M, Nishikage T, Nakai H, Kokumai M, Otani S, Lang RM: The assessment of left ventricular twist in anterior wall myocardial infarction using two-dimensional speckle tracking imaging. J Am Soc Echocardiogr. 2007, 20: 36-44. 10.1016/j.echo.2006.06.019.View ArticlePubMedGoogle Scholar
- Axel L, Dougherty L: MR imaging of motion with spatial modulation of magnetization. Radiology. 1989, 171: 841-845.View ArticlePubMedGoogle Scholar
- Axel L, Dougherty L: Heart wall motion: improved method of spatial modulation of magnetization for MR imaging. Radiology. 1989, 172: 349-350.View ArticlePubMedGoogle Scholar
- Buchalter MB, Weiss JL, Rogers WJ, Zerhouni EA, Weisfeldt ML, Beyar R, Shapiro EP: Noninvasive quantification of left ventricular rotational deformation in normal humans using magnetic resonance imaging myocardial tagging. Circulation. 1990, 81: 1236-1244.View ArticlePubMedGoogle Scholar
- Sorger JM, Wyman BT, Faris OP, Hunter WC, McVeigh ER: Torsion of the left ventricle during pacing with MRI tagging. J Cardiovasc Magn Res. 2003, 5: 521-530. 10.1081/JCMR-120025227.View ArticleGoogle Scholar
- Aelen FW, Arts T, Sanders DG, Thelissen GR, Muijtjens AM, Prinzen FW, Reneman RS: Relation between torsion and cross-sectional area change in the human left ventricle. J Biomechanics. 1997, 30: 207-212. 10.1016/S0021-9290(96)00147-9.View ArticleGoogle Scholar
- Zwanenburg JJM, Kuijer JPA, Marcus JT, Heethaar RM: Steady state free precession with myocardial tagging: CSPAMM in a single breathhold. Magn Res Med. 2003, 49: 722-730. 10.1002/mrm.10422.View ArticleGoogle Scholar
- Osman NF, Kerwin WS, McVeigh ER, Prince JL: Cardiac motion tracking using CINE harmonic phase (HARP) magnetic resonance imaging. Magn Res Med. 1999, 42: 1048-1060. 10.1002/(SICI)1522-2594(199912)42:6<1048::AID-MRM9>3.0.CO;2-M.View ArticleGoogle Scholar
- Tecelao SR, Zwanenburg JJM, Kuijer JPA, Marcus JT: Extended harmonic phase tracking of myocardial motion: improved coverage of myocardium and its effects on strain results. J Magn Res Im. 2006, 23: 682-690. 10.1002/jmri.20571.View ArticleGoogle Scholar
- Delhaas T, Kroon W, Decaluwe W, Rubbens M, Bovendeerd P, Arts T: Structure and torsion of the normal and situs inversus totalis cardiac left ventricle. I. Experimental data in humans. Am J Physiol Heart Circ Physiol. 2008, 295: H197-201. 10.1152/ajpheart.00876.2007.View ArticlePubMedGoogle Scholar
- Tecelão SR, Zwanenburg JJ, Kuijer JP, de Cock CC, Germans T, van Rossum AC, Marcus JT: Quantitative comparison of 2D and 3D circumferential strain using MRI tagging in normal and LBBB hearts. Magn Res Med. 2007, 57: 485-493. 10.1002/mrm.21142.View ArticleGoogle Scholar
- Kuijer JPA, Marcus JT, Götte MJW, van Rossum AC, Heethaar RM: Three-dimensional myocardial strain analysis based on short- and long axis magnetic resonance tagged images using a 1D displacement field. Magn Reson Imaging. 2000, 18: 533-564. 10.1016/S0730-725X(00)00146-6.View ArticleGoogle Scholar
- Bland JM, Altman DG: Measuring agreement in method comparison studies. Stat Methods Med Res. 1999, 8: 135-160. 10.1191/096228099673819272.View ArticlePubMedGoogle Scholar
- Young AA, Imai H, Chang CN, Axel L: Two-dimensional left ventricular deformation during systole using magnetic resonance imaging with spatial modulation of magnetization. Circulation. 1994, 89: 740-752.View ArticlePubMedGoogle Scholar
- Fonseca CG, Dissanayake AM, Doughty RN, Whalley GA, Gamble GD, Cowan BR, Occleshaw CJ, Young AA: Three-Dimensional assessment of left ventricular systolic strain in patients with type 2 diabetes mellitus, diastolic dysfunction, and normal ejection fraction. Am J Cardiol. 2004, 94: 1391-1395. 10.1016/j.amjcard.2004.07.143.View ArticlePubMedGoogle Scholar
- Chung J, Abraszewski P, Yu X, Liu W, Krainik AJ, Ashford M, Caruthers SD, McGill JB, Wickline SA: Paradoxical increase in ventricular torsion and systolic torsion rate in type 1 diabetic patients under tight glycemic control. J Am Coll Cardiol. 2006, 47: 384-390. 10.1016/j.jacc.2005.08.061.View ArticlePubMedGoogle Scholar
- Nagel E, Stuber M, Burkhard B, Fischer SE, Scheidegger MB, Boesiger P, Hess OM: Cardiac rotation and relaxation in patients with aortic valve stenosis. Eur Heart J. 2000, 21: 582-589. 10.1053/euhj.1999.1736.View ArticlePubMedGoogle Scholar
- Sade LE, Demir Ö, Atar I, Müderrisoglu H, Özin B: Effect of Mechanical Dyssynchrony and Cardiac Resynchronization Therapy on Left Ventricular Rotational Mechanics. The American Journal of Cardiology. 2008, 101: 1163-1169. 10.1016/j.amjcard.2007.11.069.View ArticlePubMedGoogle Scholar
- Setser RM, Kasper JM, Lieber ML, Starling RC, McCarthy PM, White RD: Persistent abnormal left ventricular systolic torsion in dilated cardiomyopathy after partial left ventriculectomy. J Thorac Cardiovasc Surg. 2003, 126: 48-55. 10.1016/S0022-5223(03)00050-3.View ArticlePubMedGoogle Scholar
- Young AA, Kramer CM, Ferrari VA, Axel L, Reickek N: Three-Dimensional Left Ventricular Deformation in Hypertrophic Cardiomyopathy. Circulation. 1994, 90: 854-867.View ArticlePubMedGoogle Scholar
- Kuijer JPA, Marcus JT, Götte MJW, van Rossum AC, Heethaar RM: Three-dimensional myocardial strains at end-systole and during diastole in the left ventricle of normal humans. J Cardiovas Magn Reson. 2002, 4: 341-351. 10.1081/JCMR-120013299.View ArticleGoogle Scholar
- Lorenz CH, Pastorek JS, Bundy JM: Delineation of normal human left ventricular twist throughout systole by tagged cine magnetic resonance imaging. J Cardiovas Magn Reson. 2000, 2: 97-108. 10.3109/10976640009148678.View ArticleGoogle Scholar
- Rüssel IK, Götte MJW, Kuijer JPA, Marcus JT: Regional assessment of left ventricular torsion by CMR tagging. Journal of Cardiovascular Magnetic Resonance. 2008, 10: 26-10.1186/1532-429X-10-26.PubMed CentralView ArticlePubMedGoogle Scholar

## Copyright

This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.