- Oral presentation
- Open Access
Towards MR-guided EP interventions using an RF-safe approach
© Krueger et al; licensee BioMed Central Ltd. 2009
- Published: 28 January 2009
- Fiber Optic Temperature
- Intracardiac Electrogram
- Prototype Setup
- Tracking Coil
Various cardiac arrhythmias, e.g. atrial fibrillation and ventricular tachycardia, can be treated by electrophysiological (EP) interventions . Applying MR for guiding these interventions offers advantages like 3D visualization of the cardiac soft tissue in relation to the catheter and absence of ionizing radiation . In this work, a prototype MR-EP system and catheter for diagnostic EP-interventions is described, which integrates concepts for RF-safe MR-tracking  and EP diagnostics . The operation of the system is demonstrated in MR-guided EP experiments in pigs including mapping and pacing. RF-safety of the diagnostic MR-EP catheter prototype is shown and signal quality is compared to conventional EP catheters.
All experiments were performed on a clinical whole-body 1.5 T MR scanner (Achieva I/T, Philips Healthcare, Netherlands) equipped with an in-room display and an additional MR-EP-workstation (MR-EP-WS) including a standard EP-recorder (EP Tracer, CardioTek, Netherlands). This workstation, located next to the scanner, combines and displays incoming real-time 2D and 3D images and real-time tracking positions from the MR scanner as well as real-time EP-data from the EP-recorder.
Comparison MR-EP/conventional EP catheter
Conventional diagnostic EP catheters (Supreme Quad, JSN, 5F, St Jude, MN) and MR-EP catheters were compared under X-ray. Bipolar intracardiac electrograms (IEGM) were acquired with both catheters at corresponding locations (RA lateral wall, RV apex, TV ring, and HIS).
In-vivo proof of RF-safety
Temperature recordings during a typical real-time bFFE sequence (TR 2.4 ms, flip 65°, global SAR 4 W/kg) were performed for the MR-EP and the conventional EP catheter. The catheters were equipped with fiber optic temperature probes and were inserted into the RA.
The RA and RV were mapped using the MR-EP system and catheters. 3D bFFE and 3D CE-MRA datasets were acquired prior to catheterization of the animals. All MR and EP data can be combined and displayed on the MR-EP-WS for guidance, including a surface model of the cardiac vessels, reformatted slices at the catheter position either manually angulated or using the real-time MR imaging geometry.
Comparison with conventional EP catheter
In-vivo proof of RF-safety
In contrast, an increase of up to 7.5 K in only 80 s was observed at the tip of the conventional catheter (Fig. 3b).
EP recording under MRI
Furthermore, atrial and ventricular pacing was achieved via the MR-EP catheters. Successful stimulation was confirmed by a second MR-EP catheter and was also clearly visible in the surface ECG.
Recording of intracardiac electrograms is feasible with the MR-EP catheter. EP data quality is equivalent to conventional EP catheters. The combined use of highly resistive wires and a transformer-based transmission line for active tip tracking effectively suppresses RF-heating even during high SAR MRI.
The prototype setup of the MR-EP system provided excellent guidance and an efficient workflow for diagnostic MR-EP interventions.
- Singer I, (ed): Interv Electrophys. 2001, Lippincott Williams & Wilkins, 2Google Scholar
- Lardo AC: Pediatr Cardiol. 2000, 21: 80-98. 10.1007/s002469910010.View ArticlePubMedGoogle Scholar
- Weiss S, et al: Magn Reson Med. 2005, 54: 182-189. 10.1002/mrm.20543.View ArticlePubMedGoogle Scholar
- Wirtz D, et al: ISMRM. 2007, 738-Google Scholar
This article is published under license to BioMed Central Ltd.