- Poster presentation
- Open Access
The conundrum of ECG-gated carotid arterial imaging with navigator corrected respiratory motion
Journal of Cardiovascular Magnetic Resonance volume 12, Article number: P135 (2010)
T o determine whether ECG-gating is necessary in carotid artery imaging and if navigator-compensation for respiration improves image quality.
Black-blood carotid artery imaging offers excellent image quality with clear delineation of the vessel wall, but suffers from time inefficiency. 3D sequences combat this problem but swallowing  and respiratory motion  may cause motion artefact. Non-ECG gated imaging is reported to be qualitatively comparable to ECG-gated images in older people with thickened, less compliant vessel walls . We used a cross-pair navigator to track respiratory motion with and without ECG-gating to assess image quality.
Ten healthy volunteers were scanned on a Siemens 1.5 T scanner with phased-array carotid coils. The carotid artery bifurcation was imaged. The subject's head was stabilised using a customised head and face mask (figure 1) and instructed not to swallow during the scans. Each subject was scanned four times: ECG-gated, navigator-gated, ECG-and-navigator-gated, none (randomised order). Two blinded independent observers graded the images on a 1-5 scale (1 = carotid wall not distinguishable, 2 = markedly blurred image +/- motion artefact, 3 = moderate blurring +/- motion artefact, 4 = minimal blurring +/- motion artefact and 5 = arterial wall clearly demarcated, no motion artefact).
Imaging parameters: A T1-weighted 2D TSE dark-blood sequence with TE = 10 ms, ETL 9 and 200% phase-oversampling was used. Pixel size 0.4 × 0.4 mm and slice thickness 2 mm. Gated studies: TR set according to RR-interval. Non-gated studies: a simulated ECG set to each individual's average RR-interval. Accept/reject-algorithm navigators were incorporated into the study.
Results and discussion
There was borderline significant difference (p = 0.06) between scores from NE and E images: this may be because respiratory motion artefact affecting image quality is variable and occurs if it falls within the centre of K-space. A significant difference (p < 0.001) was found between E and “none” images, suggesting vessel wall sharpness is dependent on the cardiac cycle and arterial pulsatility. A borderline significant difference between N and “none” images was found (p > 0.07).
ECG-gated carotid artery imaging is important. Vessel pulsatility can affect image quality in a younger age group but we did not prove that the application of a respiratory navigator was useful in this study.
Chan CF: JMRI. 2009
Chan CF: Proceedings ISMRM. 2009
Mani V: JMRI. 2005
About this article
Cite this article
Chan, C.F., Gatehouse, P., Pennell, D.J. et al. The conundrum of ECG-gated carotid arterial imaging with navigator corrected respiratory motion. J Cardiovasc Magn Reson 12, P135 (2010). https://doi.org/10.1186/1532-429X-12-S1-P135
- Motion Artefact
- Respiratory Motion
- Excellent Image Quality
- Carotid Artery Bifurcation
- Respiratory Motion Artefact