- Open Access
Comparison between cardiovascular magnetic resonance and transthoracic doppler echocardiography for the estimation of effective orifice area in aortic stenosis
© Garcia et al; licensee BioMed Central Ltd. 2011
- Received: 20 November 2010
- Accepted: 28 April 2011
- Published: 28 April 2011
The effective orifice area (EOA) estimated by transthoracic Doppler echocardiography (TTE) via the continuity equation is commonly used to determine the severity of aortic stenosis (AS). However, there are often discrepancies between TTE-derived EOA and invasive indices of stenosis, thus raising uncertainty about actual definite severity. Cardiovascular magnetic resonance (CMR) has emerged as an alternative method for non-invasive estimation of valve EOA. The objective of this study was to assess the concordance between TTE and CMR for the estimation of valve EOA.
Methods and results
31 patients with mild to severe AS (EOA range: 0.72 to 1.73 cm2) and seven (7) healthy control subjects with normal transvalvular flow rate underwent TTE and velocity-encoded CMR. Valve EOA was calculated by the continuity equation. CMR revealed that the left ventricular outflow tract (LVOT) cross-section is typically oval and not circular. As a consequence, TTE underestimated the LVOT cross-sectional area (ALVOT, 3.84 ± 0.80 cm2) compared to CMR (4.78 ± 1.05 cm2). On the other hand, TTE overestimated the LVOT velocity-time integral (VTILVOT: 21 ± 4 vs. 15 ± 4 cm). Good concordance was observed between TTE and CMR for estimation of aortic jet VTI (61 ± 22 vs. 57 ± 20 cm). Overall, there was a good correlation and concordance between TTE-derived and CMR-derived EOAs (1.53 ± 0.67 vs. 1.59 ± 0.73 cm2, r = 0.92, bias = 0.06 ± 0.29 cm2). The intra- and inter- observer variability of TTE-derived EOA was 5 ± 5% and 9 ± 5%, respectively, compared to 2 ± 1% and 7 ± 5% for CMR-derived EOA.
Underestimation of ALVOT by TTE is compensated by overestimation of VTILVOT, thereby resulting in a good concordance between TTE and CMR for estimation of aortic valve EOA. CMR was associated with less intra- and inter- observer measurement variability compared to TTE. CMR provides a non-invasive and reliable alternative to Doppler-echocardiography for the quantification of AS severity.
- Cardiovascular Magnetic Resonance
- Aortic Stenosis
- Left Ventricular Outflow Tract
- Effective Orifice Area
- Valve Orifice
Accurate assessment of valve stenosis severity is crucial for optimal management of patients with aortic stenosis (AS). The valve effective orifice area (EOA) is one of the most frequently used index to quantify stenosis severity and current ACC/AHA/ESC guidelines propose an EOA < 1.0 cm2 as the criteria to be utilized to identify severe AS [1, 2]. Given its non-invasive, radiation-free, low-cost, and versatility nature, transthoracic Doppler-echocardiography (TTE) is currently the method of choice to measure the valve EOA and grade AS severity. However, TTE has several limitations including: i) inability to obtain reliable measurements of EOA due to inadequate acoustic window and poor image quality in some patients; ii) potential for underestimation of flow velocity due to mis-alignment of Doppler beam with flow direction; iii) risk of underestimation of LV outflow (LVOT) diameter due to inadequate quality and/or positioning of image plane; iv) measurement variability related to manual tracing of flow velocity contours, etc . These limitations may significantly alter the performance of TTE to accurately quantify AS severity. Furthermore, the cardiologist if often confronted to discordant results among the different stenotic indices (i.e. EOA, transvalvular gradient, peak velocity, dimensionless velocity index) measured by Doppler-echocardiography or between the Doppler-echocardiographic evaluation of stenosis severity and the patient's clinical status . These discordances may raise some uncertainty about the actual severity of the stenosis and thus about the indication for aortic valve replacement if the patient is symptomatic. When Doppler-echocardiographic evaluation is inconclusive and/or discordant with other clinical findings, catheterization may be used to confirm valve EOA and gradients. However, left heart catheterization is an invasive method that may cause cerebral embolism . Cardiovascular magnetic resonance (CMR) has emerged as a non-invasive, radiation-free alternative modality to corroborate AS severity [5–10]. The majority of previous studies have, however, focused on the evaluation of the valvular anatomic (geometric) orifice area measured by planimetry on the images obtained by CMR or computed tomography [11–13]. From a physiologic standpoint, it is important to emphasize that the transvalvular pressure gradient and thus the LV workload are essentially determined by the valve EOA, i.e. the cross-sectional area of the vena contracta of the transvalvular flow jet, and not by the valve anatomic orifice area [14, 15]. And in this regard, it should be noted that the anatomic and effective orifice areas may differ markedly, depending on the magnitude of the flow contraction downstream of the valve.
The objective of this study was to assess the concordance between TTE and CMR for the estimation of valve EOA with use of the continuity equation method.
Seven (7) healthy control subjects and 31 patients with mild to severe AS (0.72 cm2 ≤ EOA ≤ 1.73 cm2) were included in this study. Exclusion criteria were: age < 21 years old, LV ejection fraction < 50%, atrial fibrillation, moderate or severe mitral or aortic regurgitation, poor TTE imaging quality and standard contra-indications to magnetic resonance imaging. All patients provided written informed consent. Initial AS severity classification at study entry was based on TTE-derived EOA: normal (EOA > 2.0 cm2), mild (1.5 cm2 < EOA ≤ 2.0 cm2), moderate (1.0 cm2 < EOA ≤ 1.5 cm2) and severe (EOA ≤ 1.0 cm2).
Where SVLVOT is the stroke volume measured in the LVOT, ALVOT is the cross-sectional area of the LVOT calculated assuming a circular shape: (LVOT diameter)2 × 0.785. and VTILVOT VTIAo are the velocity-time integrals of the LVOT and transvalvular flow, respectively.
Cardiovascular Magnetic Resonance
The peak flow velocity within the ROI was used to determine the instantaneous peak aortic velocity at the 6 and 10 mm positions (Figure 3D). The velocity-time integral of peak velocity during systole was calculated (VTIAo) and compared to the VTIAo measured by TTE. Given that slightly higher velocities were obtained at 6 mm versus 10 mm, we used the 6 mm position for estimation of VTIAo and EOA by CMR in this study.
Where SVCMR is the stroke volume using Simpson's rule to integrate systolic flow and VTIAo is the velocity-time integral of the peak aortic flow velocity measured at 6 mm downstream of the valve during systole.
To evaluate the intra- and inter- observer variability related to image analysis by CMR and TTE; the measurements of EOA were repeated in a subset of 15 studies (11 AS patients and 4 control subjects) by two blinded observers with the use of the same set of TTE and CMR images. To further evaluate the intra- and inter- observer- variability related to image acquisition and analysis by TTE and CMR, 5 AS patients were imaged twice within 4 weeks (including image acquisition and analysis).
Results are expressed as mean ± SD. CMR versus TTE measurements were compared by 2-tailed paired Student t-tests. Correlations and agreements between CMR and TTE measurements were assessed by Pearson's correlations and Bland-Altman comparisons, respectively. Statistical analysis was performed with SPSS 17 (SPSS, Chicago, IL).
62 ± 17
Male gender n (%)
Heart rate (bpm)
65 ± 12
76 ± 13
169 ± 10
Body surface area (m2)
1.88 ± 0.19
Body mass index (Kg/m2)
26 ± 3
Tricuspid n (%)
Bicuspid n (%)
Indeterminate n (%)
LVOT cross-sectional area
Comparison of Transthoracic Doppler-echocardiography (TTE) and Cardiovascular Magnetic Resonance (CMR) data
Mean ± SD
Mean ± SD
Heart rate (bpm)
65 ± 12
66 ± 11
LVOT area (cm2)
3.84 ± 0.8
4.78 ± 1.05
AP/RL diameter ratio
0.87 ± 0.08
LVOT VTI (cm)
21 ± 4
15 ± 4
80 ± 13
80 ± 18
Ao VTI (cm)
61 ± 22
57 ± 20
1.53 ± 0.67
1.59 ± 0.73
LVOT flow velocities and stroke volume
Aortic valve EOA
In the subset of 15 subjects (11 AS patients and 4 control subjects) in whom the analyses of EOA were repeated on the same set of images, the intra- and inter- observer variability of EOATTE was 5 ± 5% and 9 ± 5%, respectively, compared to 2 ± 1% and 7 ± 5% for CMR. In the subset of 5 patients in whom both the acquisition and analysis of images were repeated, the intra- and inter- observer variability of EOATTE was 10 ± 8% and 12 ± 5%, respectively, compared to 8 ± 8% and 9 ± 8%, for EOACMR.
Doppler-echocardiography is the method used to assess AS severity and guide therapeutic management in current practice. The valve EOA is one of the most frequently used Doppler-echocardiographic index to quantify stenosis severity. However, the measurement of valve EOA may not be feasible in a significant proportion of patients due to poor acoustic window and/or subvalvular flow acceleration. Moreover, given that the calculation of EOA requires the inclusion of 3 measures (LVOT diameter, LVOT VTI, and aortic jet VTI) in the continuity equation, this method may yield to relatively large measurement errors. Finally, there are often discordances between EOA and other Doppler-echocardiographic indices of stenosis severity, such as peak aortic jet velocity and transvalvular gradients. These discordances are particularly frequent in presence of low flow state conditions, where the gradients may be low despite the presence of a severe stenosis [18, 19]. Discordant or inconclusive Doppler-echocardiographic findings may raise some uncertainty about the actual severity of the stenosis and therefore about the therapeutic management of the patient. There is thus an important need for additional non-invasive and accurate methods to corroborate stenosis severity in patients for whom Doppler-echocardiography does not provide a definitive conclusion with regard to AS severity.
Multidetector computed tomography is a powerful imaging modality to measure dimensions, surfaces and volumes of cardiac chambers. However, this method does not allow measurement of flow velocity and thereby does not permit the determination of valve EOA. CMR is a non-invasive, radiation-free imaging modality that allows quantification of flow velocity in the LVOT and aorta. Moreover CMR has superior temporal resolution compared to computed tomography.
The main finding of this study is that there is a good agreement between CMR and Doppler-echocardiography for the estimation of valve EOA. This study also confirms the results of previous studies that reported that Doppler-echocardiography underestimates the LVOT cross-sectional area compared to computed tomography imaging [20–26]. This overestimation is essentially related to the fact that Doppler-echocardiography assumes a circular LVOT shape, whereas, in fact, it is oval in most patients. However, as opposed to what was previously believed, this underestimation of LVOT area does not necessarily translate into underestimation of LV stroke volume and valve EOA. Indeed, TTE overestimates the LVOT VTI compared to CMR, which thus compensates the underestimation of LVOT area and yields to concordant estimates of valve EOA. With TTE method, it is assumed that: i) the flow velocity profile in the LVOT is flat, i.e. mean velocity equals peak velocity, and ii) the flow velocity profile is homogenous, i.e. measurement of velocity with the pulsed-wave Doppler sample volume positioned in the center of the LVOT accurately reflects the average velocity throughout the whole LVOT cross-section. However, as illustrated in Figure 4, CMR reveals that flow velocity profile is not flat and is often skewed with higher velocities along the anterior and right aspects of the LVOT. Hence, tracing of the contour of the peak velocity envelopes obtained by pulsed-wave Doppler at the center of the LVOT overestimates the actual mean velocity and the VTI in the LVOT. The overestimation of VTILVOT by TTE somewhat counterbalances the underestimation of ALVOT. And consequently, the average stroke volume and EOA determined by TTE are similar to those determined by CMR.
Aortic valve EOA
Several previous CMR studies have focused on the measurement of the area of the aortic valve orifice by planimetry [5, 10–13]. However, it is important to underline that this "anatomic" orifice area (AOA) is not equivalent to the EOA. The latter indeed reflects the cross-sectional area of the vena contracta of the transvalvular flow jet [14, 15]. The EOA is generally smaller than the AVA because there is a contraction of the flow downstream of the valve orifice. From a physiological standpoint, the transvalvular pressure gradient and thus the LV workload are essentially determined by the EOA and the magnitude of flow rate. The ratio EOA/AOA, i.e. the contraction coefficient, may vary from 0.6 to 1.0 depending on the shape of the valve inflow and the geometry of the valve orifice [14, 15, 27]. Hence, the EOA is superior to the AOA to accurately quantify the LV hemodynamic burden associated with the stenosis. Doppler-echocardiography and CMR are the two sole methods capable of measuring the valve EOA.
Our results are consistent with those of Caruthers et al., who reported a very good correlation between EOA determined by CMR with the use of continuity equation and that obtained by TTE (r = 0.83, SEE = 0.22 cm2) . In a study where the stroke volume entered in the continuity equation was estimated by the Simpson method (i.e. LV end-diastolic volume minus LV end-systolic volume) instead of stroke volume measured in the LVOT, Yap et al. obtained an excellent correlation with TTE (r = 0.91, SEE = 0.17 cm2) . Hagui et al. also proposed a hybrid method using the stroke volume measured by CMR and the aortic jet VTI obtained by TTE in the continuity equation . This hybrid CMR-TTE method had a good agreement with the standard TTE method (bias = -0.01 cm2, limits of agreement: -0.36 to 0.34). The correlation between CMR- and TTE- derived EOAs reported in the present study appears to be better than those reported in previous studies. This may be due, at least in part, to differences in the population samples. Moreover, in the present study, we tested several locations for the measurement of the aortic jet VTI and found that highest velocities were obtained at 6 mm downstream to the valve orifice. These findings suggest that the vena contracta may actually be closer to the valve orifice compared to what was assumed (10 mm) in the previous studies [6–10].
In the present study, we also assessed the intra- and inter- observed variability: first, by repeating the EOA measurements with the use of the same sets of CMR and TTE images, and second, by repeating both acquisition and analysis of images. In both situations, CMR was found to have much less measurement variability compared to TTE, which lends further support to the reliability of this alternative imaging modality to confirm stenosis severity in the AS population.
Estimation of EOA by CMR should be contemplated when Doppler-echocardiographic measurement of EOA is not feasible or when the findings are discordant: e.g. valve EOA in the severe range (< 1.0 cm2) but mean transvalvular gradient in the moderate range (< 40 mmHg) or vice versa [18, 19]. Recent studies have revealed that these discordances are frequent [3, 28]. The first situation (small EOA and low gradient) is often found in presence of low transvalvular flow. The stroke volume and thus the transvalvular flow may indeed be significantly reduced not only in patients with low LV ejection fraction but also in those with preserved LVEF. This latter entity was recently described by our group and was termed: "paradoxical" low flow AS [3, 29]. This entity is characterized by pronounced LV concentric remodelling, small LV cavity with impaired LV filling and reduced stroke volume despite preserved LVEF. These patients with paradoxical low flow AS, who represent approximately 15-20% of AS population, often exhibit discordance between EOA and gradient and accurate determination of stroke volume and EOA is crucial in these patients. CMR may be particularly useful in these patients to corroborate stenosis severity and guide therapeutic management.
The main limitations of this study are the relatively small number of patients with severe AS and the absence of a gold standard reference method. Unfortunately there is no such method available for in vivo measurement of valve EOA. The determination of valve EOA by catheterization with the use of the Gorlin formula also has important limitations and cannot be considered as a gold standard reference method . Furthermore, this method is associated with increased risk of cerebral embolism .
Underestimation of ALVOT by TTE is compensated by overestimation of VTILVOT, thereby resulting in a good concordance between TTE and CMR for estimation of aortic valve EOA. CMR provides a non-invasive and reliable alternative to Doppler-echocardiography for the quantification of AS severity.
This work was supported by a NSERC grant (343165-07) and by a grant from the Fondation de l'Institut universitaire de cardiologie et de pneumologie de Québec. Dr. Pibarot is the director of the Canada Research Chair in Valvular Heart Diseases, Canadian Institutes of Health Research, Ottawa, Ontario, Canada. J. Garcia is supported by CONACYT (Mexico City, Mexico, grant 208171) at Laval University. Dr. Larose is a Clinical research scholar of the Fonds de la recherché en santé du Québec. We thank Isabel Fortin, Haïfa Mahjoub, Jocelyn Beauchemin, Romain Capoulade and Marc Amyot for their assistance on this study.
- Bonow RO, Carabello BA, Kanu C, et al: ACC/AHA 2006 guidelines for the management of patients with valvular heart disease: a report of the American College of Cardiology/American Heart Association Task Force on Practice Guidelines (writing committee to revise the 1998 Guidelines for the Management of Patients With Valvular Heart Disease): developed in collaboration with the Society of Cardiovascular Anesthesiologists: endorsed by the Society for Cardiovascular Angiography and Interventions and the Society of Thoracic Surgeons. Circulation. 2006, 114: e84-e231. 10.1161/CIRCULATIONAHA.106.176857.View ArticlePubMedGoogle Scholar
- Vahanian A, Baumgartner H, Bax J, et al: Guidelines on the management of valvular heart disease: The Task Force on the Management of Valvular Heart Disease of the European Society of Cardiology. Eur Heart J. 2007, 28 (2): 230-68.PubMedGoogle Scholar
- Hachicha Z, Dumesnil JG, Bogaty P, et al: Paradoxical low flow, low gradient severe aortic stenosis despite preserved ejection fraction is associated with higher afterload and reduced survival. Circulation. 2007, 115: 2856-2864. 10.1161/CIRCULATIONAHA.106.668681.View ArticlePubMedGoogle Scholar
- Omran H, Schmidt H, Hackenbroch M, et al: Silent and apparent cerebral embolism after retrograde catheterisation of the aortic valve in valvular stenosis: a prospective, randomised study. Lancet. 2003, 361 (9365): 1241-6. 10.1016/S0140-6736(03)12978-9.View ArticlePubMedGoogle Scholar
- Cawley PJ, Maki JH, Otto CM: Cardiovascular Magnetic Resonance Imaging for Valvular Heart Disease: Technique and Validation. Circulation. 2009, 119: 468-478. 10.1161/CIRCULATIONAHA.107.742486.View ArticlePubMedGoogle Scholar
- Caruthers SD, Lin SJ, Brown P, et al: Practical value of cardiac magnetic resonance imaging for clinical quantification of aortic valve stenosis: comparison with echocardiography. Circulation. 2003, 108: 2236-2243. 10.1161/01.CIR.0000095268.47282.A1.View ArticlePubMedGoogle Scholar
- Yap SC, van Geuns RJ, Meijboom FJ, et al: A simplified continuity equation approach to the quantification of stenotic bicuspid aortic valves using velocity-encoded cardiovascular magnetic resonance. J Cardiovasc Magn Reson. 2007, 9: 899-906. 10.1080/10976640701693717.View ArticlePubMedGoogle Scholar
- Waters EA, Caruthers SD, Wickline SA: Correlation Analysis of Stenotic Aortic Valve Flow Patterns Using Phase Contrast MRI. Annals of Biomed Eng. 2005, 33: 878-887. 10.1007/s10439-005-2865-9.View ArticleGoogle Scholar
- Hagui D, Suselbeck T, Fluechter S, et al: A hybrid approach for quantification of aortic valve stenosis using cardiac magnetic resonance imaging and echocardiography: comparison to right heart catheterization and standard echocardiography. Clin Res Cardiol. 2006, 95: 162-167. 10.1007/s00392-006-0355-1.View ArticleGoogle Scholar
- O'Brien KR, Gabriel RS, Greiser A, et al: Aortic valve stenotic area calculation from phase contrast cardiovascular magnetic resonance: the importance of short echo time. J Cardiovasc Magn Reson. 2009, 11: 49-10.1186/1532-429X-11-49.PubMed CentralView ArticlePubMedGoogle Scholar
- Malyar NM, Schlosser T, Barkhausen J, et al: Assessment of Aortic Valve Area in Aortic Stenosis Using Cardiac Magnetic Resonance Tomography: Comparison with Echocardiography. Cardiology. 2008, 109: 126-134. 10.1159/000105554.View ArticlePubMedGoogle Scholar
- Ropers D, Ropers U, Marwan M, et al: Comparison of dual-source computed tomography for the quantification of the aortic valve area in patients with aortic stenosis versus transthoracic echocardiography and invasive hemodynamic assessment. Am J Cardiol. 2009, 104 (11): 1561-7. 10.1016/j.amjcard.2009.07.024.View ArticlePubMedGoogle Scholar
- Bruder O, Jochims M, Hunold P, et al: Comparison of Aortic Valve Area Measured by Magnetic Resonance Imaging and Dual-Source Computed Tomography. Acta Radiol. 2009, 50 (6): 645-51. 10.1080/02841850902902540.View ArticlePubMedGoogle Scholar
- Garcia D, Kadem L: What do you mean by aortic valve area: geometric orifice area, effective orifice area, or gorlin area?. J Heart Valve Dis. 2006, 15 (5): 601-8.PubMedGoogle Scholar
- Pibarot P, Larose E: What our eyes see is not necessarily what our heart feels. Cardiology. 2008, 109 (2): 122-5. 10.1159/000105553.View ArticlePubMedGoogle Scholar
- Quinones MA, Otto CM, Stoddard M, et al: Doppler Quantification Task Force of the Nomenclature and Standards Committee of the American Society of E: Recommendations for quantification of Doppler echocardiography: a report from the Doppler Quantification Task Force of the Nomenclature and Standards Committee of the American Society of Echocardiography. J Am Soc Echocardiogr. 2002, 15: 167-184. 10.1067/mje.2002.120202.View ArticlePubMedGoogle Scholar
- Garcia J, Kadem L, Larose E, Pibarot P: In vivo Velocity and Flow Errors Quantification by Phase-Contrast Magnetic Resonance Imaging. Conf Proc IEEE Eng Med Biol Soc. 2008, 2008: 1377-1380.PubMedGoogle Scholar
- Pibarot P, Dumesnil JG: Assessment of aortic stenosis severity: when the gradient does not fit with the valve area. Heart. 2010, 96: 1431-1433. 10.1136/hrt.2010.195149.View ArticlePubMedGoogle Scholar
- Dumesnil JG, Pibarot P, Akins C: New Approches to Quantifying Aortic Stenosis Severity. Current Cardiology Reports. 2008, 10: 91-97. 10.1007/s11886-008-0017-1.View ArticlePubMedGoogle Scholar
- Johnson MA, Moss RR, Munt B: Determining Aortic Stenosis Severity: What to Do When Measuring Left Ventricular Outflow Tract Diameter Is Difficult. J Am Soc Echocardiogr. 2009, 22: 452-453. 10.1016/j.echo.2009.03.018.View ArticlePubMedGoogle Scholar
- Stolzmann P, Knight J, Desbiolles L, et al: Remodelling of the aortic root in severe tricuspid aortic stenosis: implications for trascatheter aoritc valve implantation. Eur Radiol. 2009, 19: 1316-1323. 10.1007/s00330-009-1302-0.View ArticlePubMedGoogle Scholar
- Leye M, Brochet E, Lepage L, et al: Size-adjusted left ventricular outflow tract diameter reference values: a safeguard for the evaluation of the severity of aortic stenosis. J Am Soc Echocardiogr. 2009, 22: 445-451. 10.1016/j.echo.2009.02.007.View ArticlePubMedGoogle Scholar
- Baumgartner H, Kratzer H, Helmreich G, et al: Determination of aortic valve area by Doppler echocardiography using the continuity equation: a critical evaluation. Cardiology. 1990, 77: 101-111. 10.1159/000174590.View ArticlePubMedGoogle Scholar
- Halpern EJ, Mallya R, Sewell M, et al: Differences in aortic valve area measured with CT planimetry and echocardiography (continuity equation) are related to divergent estimates of left ventricular outflow tract area. AJR Am Roentgenol. 2009, 192: 1668-1673. 10.2214/AJR.08.1986.View ArticleGoogle Scholar
- Schoenhagen P, Tuzcu EM, Kapadia SR, et al: Three-dimensional imaging of the aortic valve and aortic root with computed tomography: new standards in an era of transcatheter valve repair/implantation. Eur Heart J. 2009, 30: 2079-2086. 10.1093/eurheartj/ehp260.View ArticlePubMedGoogle Scholar
- Doddamani S, Grushko MJ, Makaryus AN: Demonstration of left ventricular outflow tract eccentricity by 64-slice multi-detector CT. Int J Cardiovasc Imaging. 2009, 25: 175-181. 10.1007/s10554-008-9362-9.View ArticlePubMedGoogle Scholar
- Gilon D, Cape GC, Handschumacher MD, Song JK, et al: Effect of Three-Dimensional Valve Shape on the Hemodynamics of Aortic Stenosis: Three-Dimensional Echocardiographic Stereolithography and Patient Studies. J Am Coll Cardiol. 2002, 40: 1479-86. 10.1016/S0735-1097(02)02269-6.View ArticlePubMedGoogle Scholar
- Minners J, Allgeier M, Gohlke-Baerwolf C, et al: Inconsistencies of echocardiographic criteria for the grading of aortic valve stenosis. Eur Heart J. 2008, 29 (8): 1043-8. 10.1093/eurheartj/ehm543.View ArticlePubMedGoogle Scholar
- Dumesnil JG, Pibarot P, Carabello B: Paradoxical low flow and/or low gradient severe aortic stenosis despite preserved left ventricular ejection fraction: implications for diagnosis and treatement. Eur Heart J. 2010, 31: 281-289. 10.1093/eurheartj/ehp361.PubMed CentralView ArticlePubMedGoogle Scholar
- Chambers J, Bach D, Dumesnil JG, et al: Crossing the Aortic Valve in Severe Aortic Stenosis: No Longer Acceptable?. J Heart Valve Dis. 2004, 12 (5): 344-6.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.