Retrospective slice prescription compensation improves coronary cross-sectional area measurement by MRI
Journal of Cardiovascular Magnetic Resonance volume 13, Article number: P236 (2011)
To determine if vessel orientation can be estimated retrospectively, and if this information improves measurements of coronary luminal area.
Measurements of coronary cross-sectional area are utilized in vasomotor tone and endothelial function studies, which typically employ breath-held, 2-D multi-slice imaging protocols wherein slices are prescribed orthogonally to a linear vessel segment (Fig 1a) ,. An ovoid template is then manually fitted to the transverse vessel images (Fig 1b) to measure cross-sectional area. Projection through the slice leads to dependencies between the measured area and the prescription angle (between the slice and the vessel--ideally 0°). This adds bias and reduces repeatability, which is problematic when detecting subtle dilations .
Cardiac-gated, breath-held spiral coronary angiography with spectral-spatial excitation was performed using a GE Signa 3T scanner (5 mm slice thickness, 16 interleaves, 5 slices). Transverse views of a cylindrical vessel phantom (diameter=5 mm, field-of-view=5 cm, resolution=0.25 mm) and a linear segment of the right coronary artery (RCA) in a healthy volunteer (diameter=~3.5 mm, field-of-view=22 cm, resolution=0.7 mm) were acquired, along with field maps, for gridding reconstruction and off-resonance deblurring . We applied a segmented cylindrical model and estimated the prescription angle for each slice from the vessel displacement through the two neighboring slices. The measured cross-sectional areas were then geometrically compensated to correct any apparent ellipticity from non-orthogonal prescriptions. This process was repeated with several prescription angles ranging from 0° to approximately 30°.
Non-orthogonal prescription angles lead to slice-dependent elliptical projections of the cylindrical phantom (Fig 1c) and the RCA (Fig 1d). Ellipticity is also generated by changes in the local RCA shape, as seen in the 5°, slice 1 image. Our approach reduced area measurement error at every prescription angle (Fig 2). The approach is sensitive to angle estimation accuracy, and tends to underestimate area as seen in the high-angle in vivo results. A higher-order model of the vessel may improve estimation performance. Nonetheless, the error performance after correction was equivalent to acquiring with much smaller prescription angles (Tables 1,2). The median in vivo improvement in the sensitivity reduction was 16-fold.
Cross-sectional lumen area can be retrospectively compensated for non-orthogonally prescribed slices, thereby reducing the sensitivity to slice prescription and breath-hold repeatability. This approach should reduce operator dependence and shorten localization time.
Terashima , et al: JACC. 2005, 45 (1): 104-110.
Nguyen , et al: Circulation. 2006, 114 (11): 541-
Jain , et al: Circulation. 2006, 114 (18): II 540b-
Noll , et al: MRM. 1992, 25 (2): 319-333.
About this article
Cite this article
Smith, T., Nayak, K. Retrospective slice prescription compensation improves coronary cross-sectional area measurement by MRI. J Cardiovasc Magn Reson 13 (Suppl 1), P236 (2011). https://doi.org/10.1186/1532-429X-13-S1-P236
- Right Coronary Artery
- Elliptical Projection
- Neighboring Slice
- Prescription Angle
- Vessel Orientation