- Open Access
Bramwell-Hill modeling for local aortic pulse wave velocity estimation: a validation study with velocity-encoded cardiovascular magnetic resonance and invasive pressure assessment
© Westenberg et al; licensee BioMed Central Ltd. 2012
- Received: 5 August 2011
- Accepted: 9 January 2012
- Published: 9 January 2012
The Bramwell-Hill model describes the relation between vascular wall stiffness expressed in aortic distensibility and the pulse wave velocity (PWV), which is the propagation speed of the systolic pressure wave through the aorta. The main objective of this study was to test the validity of this model locally in the aorta by using PWV-assessments based on in-plane velocity-encoded cardiovascular magnetic resonance (CMR), with invasive pressure measurements serving as the gold standard.
Seventeen patients (14 male, 3 female, mean age ± standard deviation = 57 ± 9 years) awaiting cardiac catheterization were prospectively included. During catheterization, intra-arterial pressure measurements were obtained in the aorta at multiple locations 5.8 cm apart. PWV was determined regionally over the aortic arch and locally in the proximal descending aorta. Subsequently, patients underwent a CMR examination to measure aortic PWV and aortic distention. Distensibility was determined locally from the aortic distension at the proximal descending aorta and the pulse pressure measured invasively during catheterization and non-invasively from brachial cuff-assessment. PWV was determined regionally in the aortic arch using through-plane and in-plane velocity-encoded CMR, and locally at the proximal descending aorta using in-plane velocity-encoded CMR. Validity of the Bramwell-Hill model was tested by evaluating associations between distensibility and PWV. Also, theoretical PWV was calculated from distensibility measurements and compared with pressure-assessed PWV.
In-plane velocity-encoded CMR provides stronger correlation (p = 0.02) between CMR and pressure-assessed PWV than through-plane velocity-encoded CMR (r = 0.69 versus r = 0.26), with a non-significant mean error of 0.2 ± 1.6 m/s for in-plane versus a significant (p = 0.006) error of 1.3 ± 1.7 m/s for through-plane velocity-encoded CMR. The Bramwell-Hill model shows a significantly (p = 0.01) stronger association between distensibility and PWV for local assessment (r = 0.8) than for regional assessment (r = 0.7), both for CMR and for pressure-assessed PWV. Theoretical PWV is strongly correlated (r = 0.8) with pressure-assessed PWV, with a statistically significant (p = 0.04) mean underestimation of 0.6 ± 1.1 m/s. This theoretical PWV-estimation is more accurate when invasively-assessed pulse pressure is used instead of brachial cuff-assessment (p = 0.03).
CMR with in-plane velocity-encoding is the optimal approach for studying Bramwell-Hill associations between local PWV and aortic distensibility. This approach enables non-invasive estimation of local pulse pressure and distensibility.
- Cardiovascular Magnetic Resonance
- Aortic Arch
- Pulse Pressure
- Pulse Wave Velocity
- Aortic Distensibility
The pathophysiological processes of cardiovascular disease involve stiffening of the arterial vessel wall. Increased aortic wall stiffness results in an increased aortic pulse pressure and left ventricular afterload, restricting left ventricular filling during diastole, which eventually may lead to heart failure [1, 2]. Additionally, aortic stiffening is an important risk factor for end organ damage with coronary, renal or cerebral expression as the hemodynamic load on the end organs is increased with impaired damping of the systolic wave [3–10]. The growing awareness of the prognostic value of aortic stiffness for the prediction of cardiovascular morbidity and mortality, as highlighted in a meta-analysis published by Vlachopoulos et al. , increases the recognition of stiffness-assessment as a surrogate end point for cardiovascular disease in clinical research .
Assessment of regional PWV is of high interest in cardiovascular research as the independent prognostic value of regional PWV-assessment for outcome prediction in various patient populations has been recognized [2–4, 7–10]. Since the majority of the reservoir capacity of the arterial system resides in the proximal part of the aorta, stiffness assessment in this region will provide essential information on the aortic condition and function. Moreover, as the aorta changes in structure over its length considerably with age, regional identification of increased wall stiffening may provide valuable insight into the underlying pathology.
Several estimators - such as Young's modulus, distensibility or stiffness index - are currently in use to express aortic stiffness, all relating local blood pressure with the distention of the aorta (either by diameter or luminal area). Aortic distensibility is defined as the relative change in vessel diameter over local pulse pressure [13, 14]. A useful surrogate marker of aortic stiffness is the pulse wave velocity (PWV), which is defined as the velocity of the systolic pulse wave front propagating through the aorta. The PWV is increased when atherosclerotic wall degeneration and concomitant reduction of elastic recoil are present, and PWV has proven to be an independent and strong predictor of cardiovascular morbidity and mortality [15–19].
The gold standard for PWV-assessment is defined from invasive pressure measurements at consecutive locations in the aorta, from which the propagation speed of the systolic pressure wave front can be accurately determined. Cardiovascular Magnetic Resonance (CMR) with velocity-encoding (VE) is a validated alternative for measuring the PWV , globally for the whole aorta, regionally in specific aortic segments as well as locally at a specific position in the aorta .
The Bramwell-Hill model  theoretically links PWV, aortic distensibility and pulse pressure together. This model is derived from the Moens-Korteweg equation which, under modeling assumptions (i.e. vessel wall thickness is small compared to the diameter and the circulating fluid within the vessel is incompressible and nonviscous), relates arterial stiffness and PWV. Recently, Dogui et al.  tested the validity of this model and the associations between pulse pressure, PWV and aortic distensibility using VE CMR regionally in the aortic arch. However, in their experiments, no PWV or pulse pressure information was available locally in the aorta, whereas distensibility is an intrinsic local measure. In our present study, we aim to overcome the shortcomings of the study of Dogui et al. by using local PWV-assessment from in-plane VE CMR. Furthermore, invasive pressure measurements are obtained during catheterization to determine the local pulse pressure for accurate distensibility assessments, and to determine the gold standard for PWV-assessment. Therefore, the main objective of this study was to test the validity of the Bramwell-Hill model locally in the aorta by using PWV-assessments based on in-plane VE CMR-acquisitions, with invasive pressure measurements as the gold standard.
Data from the subjects reported in this study have been described earlier in studies validating PWV-assessment with VE-CMR [20, 21]. Research was carried out in compliance with the Helsinki Declaration. The study was approved by the local Medical Ethical Committee. A total of 17 patients (14 male and 3 female, mean age ± standard deviation (SD) = 57 ± 9 years) with suspected coronary artery disease awaiting elective cardiac catheterization were prospectively enrolled in the study after giving informed consent.
Patients underwent CMR examination to measure aortic PWV and aortic distention. The mean interval between catheterization and CMR was 17 ± 13 days. Exclusion criteria were general contraindications to CMR, evidence of aortic valve stenosis on ultrasonography, coarctation of the aorta, or other congenital heart disease, or a family history of Marfan syndrome.
with A being the luminal area and ΔA and ΔP the difference between maximal and minimal luminal area and blood pressure, respectively, during the cardiac cycle.
The relation between PWV, pulse pressure and distensibility will be tested by PWV-assessment from both invasive pressure measurements (gold standard) and VE CMR, with blood pressure assessment both invasively and from brachial cuff measurement, and assessment of aortic luminal area distention with CMR.
Invasive pressure measurements for PWVpressure
Invasive pressure-time curves and simultaneous ECG recordings were obtained immediately after vascular access, to avoid any interference by medication or performed procedures. A 6F JR4 pressure tip catheter (Cordis, Miami Lakes, FL) was introduced through a 6F sheet (Cordis) into either one of the femoral arteries and advanced through the aorta until just distal to the aortic valve. During stepwise pullback, pressure waves were recorded at consecutive positions spaced 5.8 cm apart. Pressure-time curves and ECGs were recorded with a sampling resolution of 2 kHz during at least 10 cardiac cycles at each position to take into account variations induced by respiration. PWVpressure is calculated as Δx/Δt (expressed in m/s), where Δx is the aortic path length between measurement sites, determined from catheter pullback, and Δt is the transit-time for the systolic pressure wave front to propagate between these sites. The onset of the systolic pressure wave front was automatically determined from the time point (relative to the R-wave) of minimal pressure prior to the upslope of the systolic pressure wave. Local pulse pressure was determined from the difference between maximal and minimal pressure. Offline analysis of the pressure-time curves was performed using custom-made software.
VE CMR for PWVt.p. and PWVi.p.
Imaging sequences were previously described [20, 21]. In short, first a stack of three consecutive double-oblique-sagittal slices was obtained capturing the full aorta from aortic root, arch to abdominal aorta, using cine-gradient-echo with retrospective gating and steady-state free-precession. This stack of slices covering the aorta was used for planning VE CMR-acquisitions and to measure the aortic path length of the aortic arch trajectory. Next, a single-slice one-directional through-plane VE CMR-acquisition was performed perpendicular to the aorta at the level of the pulmonary trunk, transecting both the ascending and proximal descending aorta distal to the aortic arch. A five-element cardiac surface coil was used for signal reception. Scan parameters were: 90% rectangular FOV 300 × 270 mm2, 8 mm slice thickness, TE 2.9 msec, TR 4.9 msec, flip angle α 20°, acquisition voxel size 2.3 × 2.1 × 8.0 mm3, sampling bandwidth 449 Hz, number of signal averages 2, retrospective gating with maximal number of phases reconstructed into one average cardiac cycle with arrhythmia rejection window set to acceptance threshold of 15%-variation. The velocity sensitivity was set to 150 cm/s. Acquisition was performed with free breathing without respiratory compensation and scan time was almost 4 minutes at a heart rate of 60 bpm.
Finally, PWVi.p. was assessed by means of two consecutive multi-slice two-directional in-plane VE CMR-acquisitions of the three-slice double-oblique-sagittal stack of the aorta. VE was performed in phase-encoding (i.e., anterior-posterior AP) direction and in frequency-encoding (i.e., feet-head FH) direction respectively with velocity sensitivity set to 150 cm/s in either direction. The body coil was used for signal reception. Scan parameters were: 60%-rectangular FOV 450 × 270 mm2, 10 mm slice thickness, TE 2.4 ms, TR 4.3 ms, α 10°, acquisition voxel size 3.5 × 2.1 × 10.0 mm3, sampling bandwidth 495 Hz, NSA 2, retrospective-gating with maximal number of phases reconstructed into one average cardiac cycle with arrhythmia rejection window set to acceptance of 15%-variation. Acquisition was performed with free breathing without respiratory compensation and scan time of a single acquisition amounted to 7 minutes 42 seconds at a heart rate of 60 bpm.
For CMR-assessed PWV, Δx/Δt was calculated with Δx the aortic path length between measurement sites and Δt the transit-time for the systolic velocity wave front to propagate between these sites. The aortic path length between the subsequent measurement sites in ascending and proximal descending aorta was manually assessed by drawing a poly-line along the centerline of the aortic arch within the scout images, using the in-house developed software package MASS (Leiden University Medical Center, Leiden, The Netherlands) . At each measurement site, maximal velocity-time curves were determined and used to evaluate the systolic wave propagation. These velocity-time curves were determined from aortic velocity maps which were analyzed with the in-house developed analytic software package FLOW (Leiden University Medical Center, Leiden, The Netherlands) , using automated contour detection. The onset of the systolic wave front, required for transit-time calculation, was automatically determined from the intersection point of a horizontal line modeling the constant horizontal diastolic flow and a line modeling the systolic upslope in the velocity-time curve. This line was modeled by linear regression of the flow velocity values within 20% and 80% of the total range of values along the systolic upslope. This algorithm was also used for calculating the transit-time for the propagation of the maximal velocity wave front, obtained with in-plane VE CMR. PWVi.p. was obtained as described before , from the two three-slice acquisitions with two-directional in-plane VE CMR. The full aorta was manually segmented with one set of contours that was subsequently copied to all slices, all phases and to both AP and FH encoding series. The aortic centerline was automatically determined from the contour set and 200 equidistantly-spaced sampling chords perpendicular to the centerline were defined. For each pixel within the aorta, the velocity in the direction parallel along the centerline was constructed from the velocity components in AP and FH direction. The aortic flow velocity was sampled along each chord and the maximal velocity per chord was determined. For each slice, this resulted in 200 velocity-time curves which were used to determine the transit-time as described above.
PWVt.p. and PWVi.p. were determined regionally in the aortic arch between measurement sites 1 and 2 and PWVi.p. was determined also locally around site 2, for an 11.6 cm trajectory starting 5.8 cm above site 2 to 5.8 cm below site 2. The aortic luminal distension was determined at site 2 from the maximal and minimal cross-sectional lumen area measured as described above with the automated contour detection on the velocity-insensitive magnitude gradient-echo images. Immediately after VE CMR-examination, brachial cuff blood pressure was obtained using a semi-automated sphygmomanometer (Dinamap, Critikon, Tampa, FL, USA), with the subject remaining in supine position on the CMR-table. Regional PWVpressure was determined over the aortic arch, from pressure measurements between sample points closest to measurement sites 1 and 2. Furthermore, local PWVpressure was determined for an 11.6 cm trajectory around site 2.
Statistical analysis was performed using SPSS for Windows (v. 12.0.1; Chicago, IL). All data are presented as mean values ± one SD, unless stated otherwise.
Associations of PWVt.p. and PWVi.p. with PWVpressure were evaluated using the Pearson's correlation coefficient (r), while variation with PWVpressure was studied with coefficients of variation (COV; defined as the standard deviation of the differences between the two series of measurements divided by the mean of both measurements). Also mean unsigned error (with PWVpressure as reference standard) and 95%-confidence intervals (95%-CI) were calculated. Statistical significance of differences between correlation coefficients was tested by stepwise linear regression analysis with the gold standard and the interaction between the gold standard and the tested methods as predictors. The approach described by Bland and Altman  was followed to study systematic trends in differences. Validity of the Bramwell-Hill model was tested by evaluating the association between regional and local PWV with local distensibility, using the pulse pressure measured both with brachial cuff-assessment at CMR examination (i.e. PPcuff) and intra-arterially during catheterization (i.e. PPcath). Furthermore, the estimated theoretical PWV from distensibility assessment was compared to the gold standard PWVpressure. Statistical significance on all tests was indicated by p < 0.05.
Mean ± SD
57 ± 9
34 - 75
173 ± 12
157 - 185
82 ± 11
62 - 100
Body mass index (kg.m -2 )
27 ± 4
22 - 35
Body surface area (m 2 )
2.0 ± 0.2
1.7 - 2.2
Systolic blood pressure (mmHg)
129 ± 21
97 - 166
Diastolic blood pressure (mmHg)
76 ± 13
53 - 100
2.2 ± 1.0
Regionally in the aortic arch, mean PWV from in-plane VE CMR was 6.7 ± 2.3 m/s, mean PWV from through-plane was 5.6 ± 1.2 m/s and mean PWV from invasive pressure measurements was 6.9 ± 1.5 m/s. Locally at site 2, mean PWV from in-plane VE CMR was 7.1 ± 2.6 m/s and from invasive pressure measurement 7.0 ± 1.6 m/s.
Associations between CMR- and pressure-assessed PWV.
PWVt.p. vs. PWVpressure
PWVi.p. vs. PWVpressure
PWVi.p. vs. PWVpressure
0.26 (p = 0.31)
0.69 (p = 0.002)
0.91 (p < 0.001)
Mean unsigned error
26% ± 13%
18% ± 12%
15% ± 8%
Mean difference (m/s)
-1.3 ± 1.7
-0.2 ± 1.6
0.2 ± 1.3
-2.1 - 0.5
-1.0 - 0.6
-0.4 - 0.8
Association between PWV and (distensibility)-1/2 according to Bramwell-Hill model.
Distensibility with PPcuff
Distensibility with PPcath
regional PWV t.p.
r = 0.40, p = 0.11
r = 0.34, p = 0.18
regional PWV i.p.
r = 0.63, p = 0.007
r = 0.62, p = 0.007
p-value comparison ↨
local PWV i.p.
r = 0.72, p = 0.001
r = 0.74, p = 0.001
p-value comparison ↨
local PWV pressure
r = 0.83, p < 0.001
r = 0.80, p < 0.001
p-value comparison ↨
regional PWV pressure
r = 0.45, p = 0.07
r = 0.57, p = 0.02
Association between PWV from Bramwell-Hill model (with brachial cuff and intra-arterial pressure) and pressure-assessed PWV.
0.82 (p < 0.001)
0.80 (p < 0.001)
Mean unsigned error
16% ± 9%
16% ± 9%
Mean difference (m/s)
-1.1 ± 0.9
-0.6 ± 1.1
-1.5 - -0.7
-1.1 - -0.1
In this study, the validity of the Bramwell-Hill model - which describes the relation between pulse pressure, pulse wave velocity and distensibility locally in the aorta - was tested on velocity-encoded CMR-acquisitions with invasive pressure measurements serving as the gold standard. PWV was assessed with VE CMR as well as with invasive pressure measurements (i.e. the gold standard), both locally at the proximal descending aorta and regionally in the aortic arch. The main findings of our study are: 1) correlation between the gold standard and regional PWV-assessment in the aortic arch with in-plane VE CMR is significantly stronger than with through-plane VE CMR; 2) local PWV-assessment in the proximal descending aorta with in-plane VE CMR is stronger correlated with the gold standard than regional PWV-assessment in the aortic arch; 3) the Bramwell-Hill model shows stronger association between PWV, distensibility and pulse pressure when applied to local PWV-assessment as compared to regional assessment; 4) local PWV modeled from local distensibility is significantly correlated with the gold standard, but shows a significant underestimation, with a higher mean error for pulse pressure measurement from non-invasive brachial cuff than from invasive pressure assessment.
Many pathophysiological processes of cardiovascular diseases involve increased stiffening of the arterial vessel wall [1, 2], which leads to abnormal wave propagation. Central aortic wall stiffening may be the starting point of a negative cascade of end organ damage and has been reported as an important risk factor for various expressions of end organ damage and dysfunction [3–10]. The Bramwell-Hill model describes the relation between PWV, distensibility and pulse pressure. Modeling assumptions on arterial vessel wall and flowing blood properties need to be taken into account, which implies that this model is not generally valid for all individuals. Furthermore, whereas distensibility and pulse pressure are assessed locally in the aorta, local PWV can only be estimated since it is determined from the wave propagation over a particular aortic trajectory, with the optimal estimation from in-plane VE CMR.
In a previous study by Westenberg et al. it was already shown that PWV-assessment from in-plane VE CMR results in a more accurate estimation for global PWV in the complete aorta . Our findings show that also PWV-assessment regionally in the aortic arch as well as locally at the proximal descending aorta correlates better with the gold standard when using in-plane VE CMR instead of through-plane VE CMR. The in-plane PWV-assessment is obtained from a highly dense spatial sampling of the pulse wave along the centerline of the aorta, which contributes significantly to the accuracy.
In this study, the validity of the Bramwell-Hill model was evaluated locally at the proximal descending aorta. This sampling location is technically attractive and pathophysiologically relevant. Motion blurring due to breathing and through-plane motion due to cardiac contraction in the ascending aorta may affect aortic lumen measurements, which are required for distensibility calculations. Respiratory motion will have a similar effect on the lumen assessment both in systole as in diastole. At the location of the descending aorta, the aorta will be subjected to minimal through-plane motion. Furthermore, it is known that the aorta stiffens with age. It was shown that global PWV increases in a nonlinear fashion with age and is most pronounced in the thoracic aorta , probably because of cumulative degradation of elastin fibers with age  that are most abundant in the thoracic part of the aorta.
To our knowledge, we are the first to test the validity of the Bramwell-Hill model applied at a single location in the aorta. We found a stronger association between PWV, distensibility and pulse pressure for local assessment than for regional assessment. Dogui et al. tested the model on a regional level in the aortic arch and acknowledged the inability to assess local values as a limitation to their study . Nevertheless they found good correlations between regional PWV in the aortic arch and aortic distensibility with correlation still between 0.7 and 0.8. Another discrepancy with the study from Dogui et al. is that in our study the local pulse pressure was assessed invasively and accurately, whereas Dogui et al. only used a global estimation by cuff measurements and carotid tonometry. These practical improvements in our study may possibly explain the higher correlations for the Bramwell-Hill model from our data.
The Bramwell-Hill model can be applied to calculate local PWV-values from distensibility measurements, or vice versa. Theoretically, local pulse pressure can be determined non-invasively from this model, by using local PWV- and aortic luminal distention assessments. Still, in our study a significant underestimation for calculated PWV-values were found when compared to PWV-measurements assessed with invasive pressure. However, this mean underestimation amounted to 0.6 m/s, which may not be clinically relevant since this is in order of 10% of normal values for aortic PWV. Several sources of error might explain this underestimation. First, the CMR and catheterization were not performed on the same day, which is a limitation of our study. The time interval between both examinations might be a possible explanation for the differences, since arterial compliance is subject to physiological day-to-day differences in blood pressure, blood flow and sympathic tone . Also, it is known that beta blocker agents and angiotensin converting enzyme inhibitors may increase aortic compliance and inversely decrease aortic PWV [2, 30]. In four patients of the current study population, medication of beta blocking agent was either commenced or increased in dosage between catheterization and CMR. Additionally, in four patients medication with angiotensin converting enzyme inhibitor was initiated or increased. It is unknown what the magnitude of change in aortic PWV will be after changing medication, and what the time interval should be for the medication to have this effect.
Our study has some additional limitations. Automated cuff blood pressure measurements were performed for brachial blood pressure assessment immediately after CMR. The gold standard for measuring brachial blood pressure noninvasively is the mercury manometer. In our study a Dinamap automated blood pressure measurement device was used. It has been described that automated sphygmomanometers are less accurate than mercury manometers. Beaubien et al. showed that only 59% of systolic and 56% of diastolic readings will be within 5 mmHg of the mercury manometer values . Due to practical considerations, the use of an automated sphygmomanometer was required in our study. In order to limit sympathic influences, patients remained in supine position on the CMR-table whilst blood pressure measurements were performed immediately after CMR. Furthermore, this issue is of limited concern since it potentially will introduce a systematic error of equal magnitude on all patients since the same sphygmomanometer was used for all subjects. Furthermore, in the 2006 expert consensus on arterial stiffness it was reported that brachial pulse pressure does not accurately represent aortic pulse pressure . Additionally, blood pressures measurements were obtained invasively during cardiac catheterization, which provide accurate local values of the pulse pressure, necessary for distensibility assessment. Still, the already acknowledged physiological day-to-day differences in blood pressure will restrict the interchangeable use of pressure measurements between examinations. Local invasive pressure should ideally be assessed simultaneously to aortic distention and PWV-assessment. The availability of an interventional CMR environment may facilitate further exploration of the Bramwell-Hill model. Alternatively, this model may also be applied inversely to calculate local pulse pressure from PWV, acquired non-invasively from VE CMR.
Finally, the small sample size is another limitation. This study described a validation of the Bramwell-Hill model for local and regional aortic PWV by VE CMR. No data on the prognostic value of PWV-assessment were presented. Large scale studies, both in patients and healthy volunteers, are needed for further testing the validity of the Bramwell-Hill model and to assess whether local PWV from in-plane VE CMR is superior to regional PWV-assessment for predicting outcome. Additionally, PWV-assessment from in-plane VE CMR has been proven to be accurate and reproducible , but image acquisition and analysis still remain time-consuming, which hampers widespread use in clinical practice. Carotid-femoral or sternal-femoral ultrasound data still provide a faster and more cost-effective estimation of PWV.
CMR with in-plane velocity-encoding is the optimal approach for studying the association between local PWV and aortic distensibility as described by the Bramwell-Hill model. This approach enables local pulse pressure and local distensibility estimation in a non-invasive manner.
Acknowledgements and funding
Funding for this study by the Netherlands Heart Foundation (Project 2006B138) is gratefully acknowledged. We would like to acknowledge Gerrit Kracht for graphic design.
- O'Rourke MF, Staessen JA, Vlachopoulos C, Duprez D, Plante GE: Clinical applications of arterial stiffness; definitions and reference values. Am J Hypertens. 2002, 15: 426-444. 10.1016/S0895-7061(01)02319-6.View ArticlePubMedGoogle Scholar
- Auseon AJ, Tran T, Garcia AM, Hardy CJ, Valavalkar P, Moeschberger M, Raman SV: Aortic pathophysiology by cardiovascular magnetic resonance in patients with clinical suspicion of coronary artery disease. J Cardiovasc Magn Reson. 2007, 9: 43-48. 10.1080/10976640600897377.View ArticlePubMedGoogle Scholar
- Boutouyrie P, Tropeano AI, Asmar R, Gautier I, Benetos A, Lacolley P, Laurent S: Aortic stiffness is an independent predictor of primary coronary events in hypertensive patients: a longitudinal study. Hypertension. 2002, 39: 10-15. 10.1161/hy0102.099031.View ArticlePubMedGoogle Scholar
- Mattace-Raso FU, van der Cammen TJ, Hofman A, van Popele NM, Bos ML, Schalekamp MA, Asmar R, Reneman RS, Hoeks AP, Breteler MM, Witteman JC: Arterial stiffness and risk of coronary heart disease and stroke: the Rotterdam Study. Circulation. 2006, 113: 657-663. 10.1161/CIRCULATIONAHA.105.555235.View ArticlePubMedGoogle Scholar
- van Elderen SG, Westenberg JJ, Brandts A, van der Meer RW, Romijn JA, Smit JW, de Roos A: Increased aortic stiffness measured by MRI in patients with type 1 diabetes mellitus and relationship to renal function. AJR Am J Roentgenol. 2011, 196: 697-701. 10.2214/AJR.10.4898.View ArticlePubMedGoogle Scholar
- Blacher J, Guerin AP, Pannier B, Marchais SJ, Safar ME, London GM: Impact of aortic stiffness on survival in end-stage renal disease. Circulation. 1999, 99: 2434-2439.View ArticlePubMedGoogle Scholar
- O'Rourke MF, Safar ME: Relationship between aortic stiffening and microvascular disease in brain and kidney: cause and logic of therapy. Hypertension. 2005, 46: 200-204. 10.1161/01.HYP.0000168052.00426.65.View ArticlePubMedGoogle Scholar
- Brandts A, van Elderen SG, Westenberg JJ, van der Grond J, van Buchem MA, Huisman MV, Kroft LJ, Tamsma JT, de Roos A: Association of aortic arch pulse wave velocity with left ventricular mass and lacunar brain infarcts in hypertensive patients: assessment with MR imaging. Radiology. 2009, 253: 681-688. 10.1148/radiol.2533082264.View ArticlePubMedGoogle Scholar
- van Elderen SG, Brandts A, van der Grond J, Westenberg JJ, Kroft LJ, van Buchem MA, Smit JW, de Roos A: Cerebral perfusion and aortic stiffness are independent predictors of white matter brain atrophy in type 1 diabetic patients assessed with magnetic resonance imaging. Diabetes Care. 2011, 34: 459-463. 10.2337/dc10-1446.PubMed CentralView ArticlePubMedGoogle Scholar
- Laurent S, Katsahian S, Fassot C, Tropeano AI, Gautier I, Laloux B, Boutouyrie P: Aortic stiffness is an independent predictor of fatal stroke in essential hypertension. Stroke. 2003, 34: 1203-1206. 10.1161/01.STR.0000065428.03209.64.View ArticlePubMedGoogle Scholar
- Vlachopoulos C, Aznaouridis K, Stefanadis C: Prediction of cardiovascular events and all-cause mortality with arterial stiffness: a systematic review and meta-analysis. J Am Coll Cardiol. 2010, 55: 1318-1327. 10.1016/j.jacc.2009.10.061.View ArticlePubMedGoogle Scholar
- Laurent S, Cockcroft J, Van Bortel L, Boutouyrie P, Giannattasio C, Hayoz D, Pannier B, Vlachopoulos C, Wilkinson I, Struijker-Boudier H, European Network for Non-invasive Investigation of Large Arteries: Expert consensus document on arterial stiffness: methodological issues and clinical applications. Eur Heart J. 2006, 27: 2588-2605. 10.1093/eurheartj/ehl254.View ArticlePubMedGoogle Scholar
- Herment A, Lefort M, Kachenoura N, De Cesare A, Taviani V, Graves MJ, Pellot-Barakat C, Frouin F, Mousseaux E: Automated estimation of aortic strain from steady-state free-precession and phase contrast MR images. Magn Reson Med. 2011, 65: 986-993. 10.1002/mrm.22678.View ArticlePubMedGoogle Scholar
- Grotenhuis HB, Westenberg JJ, Doornbos J, Kroft LJ, Schoof PH, Hazekamp MG, Vliegen HW, Ottenkamp J, de Roos A: Aortic root dysfunctioning and its effect on left ventricular function in Ross procedure patients assessed with magnetic resonance imaging. Am Heart J. 2006, 152 (5): 975.e1-8. 10.1016/j.ahj.2006.06.038.View ArticleGoogle Scholar
- Blacher J, Asmar R, Djane S, London GM, Safar ME: Aortic pulse wave velocity as a marker of cardiovascular risk in hypertensive patients. Hypertension. 1999, 33: 1111-1117.View ArticlePubMedGoogle Scholar
- Meaume S, Benetos A, Henry OF, Rudnichi A, Safar ME: Aortic pulse wave velocity predicts cardiovascular mortality in subjects >70 years of age. Arterioscler Thromb Vasc Biol. 2001, 21 (12): 2046-2050. 10.1161/hq1201.100226.View ArticlePubMedGoogle Scholar
- Sutton Sutton-Tyrrell K, Najjar SS, Boudreau RM, Venkitachalam L, Kupelian V, Simonsick EM, Havlik R, Lakatta EG, Spurgeon H, Kritchevsky S, Pahor M, Bauer D, Newman A, Health ABC Study: Elevated aortic pulse wave velocity, a marker of arterial stiffness, predicts cardiovascular events in well-functioning older adults. Circulation. 2005, 111: 3384-3390. 10.1161/CIRCULATIONAHA.104.483628.View ArticleGoogle Scholar
- Willum-Hansen T, Staessen JA, Torp-Pedersen C, Rasmussen S, Thijs L, Ibsen H, Jeppesen J: Prognostic value of aortic pulse wave velocity as index of arterial stiffness in the general population. Circulation. 2006, 113: 664-670. 10.1161/CIRCULATIONAHA.105.579342.View ArticlePubMedGoogle Scholar
- Cruickshank K, Riste L, Anderson SG, Wright JS, Dunn G, Gosling RG: Aortic pulse-wave velocity and its relationship to mortality in diabetes and glucose intolerance: an integrated index of vascular function?. Circulation. 2002, 106: 2085-2090. 10.1161/01.CIR.0000033824.02722.F7.View ArticlePubMedGoogle Scholar
- Grotenhuis HB, Westenberg JJ, Steendijk P, van der Geest RJ, Ottenkamp J, Bax JJ, Jukema JW, de Roos A: Validation and reproducibility of aortic pulse wave velocity as assessed with velocity-encoded MRI. J Magn Reson Imaging. 2009, 30: 521-526. 10.1002/jmri.21886.View ArticlePubMedGoogle Scholar
- Westenberg JJ, de Roos A, Grotenhuis HB, Steendijk P, Hendriksen D, van den Boogaard PJ, van der Geest RJ, Bax JJ, Jukema JW, Reiber JH: Improved aortic pulse wave velocity assessment from multislice two-directional in-plane velocity-encoded magnetic resonance imaging. J Magn Reson Imaging. 2010, 32: 1086-1094. 10.1002/jmri.22359.View ArticlePubMedGoogle Scholar
- Bramwell JC, Hill AV: The velocity of the pulse wave in man. Proc R Soc Lond B. 1922, 93: 298-306. 10.1098/rspb.1922.0022.View ArticleGoogle Scholar
- Dogui A, Kachenoura N, Frouin F, Lefort M, De Cesare A, Mousseaux E, Herment A: Consistency of aortic distensibility and pulse wave velocity estimates with respect to the Bramwell-Hill theoretical model: a cardiovascular magnetic resonance study. J Cardiovasc Magn Reson. 2011, 13: 11-10.1186/1532-429X-13-11.PubMed CentralView ArticlePubMedGoogle Scholar
- Stevanov M, Baruthio J, Gounot D, Grucker D: In vitro validation of MR measurements of arterial pulse-wave velocity in the presence of reflected waves. J Magn Reson Imaging. 2001, 14: 120-127. 10.1002/jmri.1161.View ArticlePubMedGoogle Scholar
- van der Geest RJ, de Roos A, van der Wall EE, Reiber JH: Quantitative analysis of cardiovascular MR images. Int J Card Imaging. 1997, 13: 247-258. 10.1023/A:1005869509149.View ArticlePubMedGoogle Scholar
- van der Geest RJ, Niezen RA, van der Wall EE, de Roos A, Reiber JH: Automated measurement of volume flow in the ascending aorta using MR velocity maps: evaluation of inter- and intraobserver variability in healthy volunteers. J Comput Assist Tomogr. 1998, 22: 904-911. 10.1097/00004728-199811000-00013.View ArticlePubMedGoogle Scholar
- Bland JM, Altman DG: Statistical methods for assessing agreement between two methods of clinical measurement. Lancet. 1986, 1: 307-310.View ArticlePubMedGoogle Scholar
- Taviani V, Hickson SS, Hardy CJ, McEniery CM, Patterson AJ, Gillard JH, Wilkinson IB, Graves MJ: Age-related changes of regional pulse wave velocity in the descending aorta using Fourier velocity encoded M-mode. Magn Reson Med. 2011, 65: 261-268. 10.1002/mrm.22590.View ArticlePubMedGoogle Scholar
- Schlatmann TJ, Becker AE: Histologic changes in the normal aging aorta: implications for dissecting aortic aneurysm. Am J Cardiol. 1977, 39: 13-20. 10.1016/S0002-9149(77)80004-0.View ArticlePubMedGoogle Scholar
- Groenink M, de Roos A, Mulder BJ, Spaan JA, van der Wall EE: Changes in aortic distensibility and pulse wave velocity assessed with magnetic resonance imaging following beta-blocker therapy in the Marfan syndrome. Am J Cardiol. 1998, 82: 203-208. 10.1016/S0002-9149(98)00315-4.View ArticlePubMedGoogle Scholar
- Beaubien ER, Card CM, Card SE, Biem HJ, Wilson TW: Accuracy of the Dinamap 1846 XT automated blood pressure monitor. J Hum Hypertens. 2002, 16: 647-652. 10.1038/sj.jhh.1001463.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.