- Open Access
Single breath-hold assessment of cardiac function using an accelerated 3D single breath-hold acquisition technique - comparison of an intravascular and extravascular contrast agent
© Makowski et al.; licensee BioMed Central Ltd. 2012
- Received: 21 January 2012
- Accepted: 10 July 2012
- Published: 31 July 2012
Cardiovascular magnetic resonance (CMR) is the current gold standard for the assessment of left ventricular (LV) function. Repeated breath-holds are needed for standard multi-slice 2D cine steady-state free precession sequences (M2D-SSFP). Accelerated single breath-hold techniques suffer from low contrast between blood pool and myocardium. In this study an intravascular contrast agent was prospectively compared to an extravascular contrast agent for the assessment of LV function using a single-breath-hold 3D-whole-heart cine SSFP sequence (3D-SSFP).
LV function was assessed in fourteen patients on a 1.5 T MR-scanner (Philips Healthcare) using 32-channel coil technology. Patients were investigated twice using a 3D-SSFP sequence (acquisition time 18–25 s) after Gadopentetate dimeglumine (GdD, day 1) and Gadofosveset trisodium (GdT, day 2) administration. Image acquisition was accelerated using sensitivity encoding in both phase encoding directions (4xSENSE). CNR and BMC were both measured between blood and myocardium. The CNR incorporated noise measurements, while the BMC represented the coeffiancy between the signal from blood and myocardium . Contrast to noise ratio (CNR), blood to myocardium contrast (BMC), image quality, LV functional parameters and intra-/interobserver variability were compared. A M2D-SSFP sequence was used as a reference standard on both days.
All 3D-SSFP sequences were successfully acquired within one breath-hold after GdD and GdT administration. CNR and BMC were significantly (p < 0.05) higher using GdT compared to GdD, resulting in an improved endocardial definition. Using 3D-SSFP with GdT, Bland–Altman plots showed a smaller bias (95% confidence interval LVEF: 9.0 vs. 23.7) and regression analysis showed a stronger correlation to the reference standard (R2 = 0.92 vs. R2 = 0.71), compared to 3D-SSFP with GdD.
A single-breath-hold 3D-whole-heart cine SSFP sequence in combination with 32-channel technology and an intravascular contrast agent allows for the accurate and fast assessment of LV function.
The study was approved by the local research ethics committee (Study No. 07/Q0704/2) and was registered with the Medicines and Healthcare Products Regulatory Agency (MHRA Study No. 28482/0002/001–0001, EudraCTnumber 2006–007042).
- Cardiovascular magnetic resonance
- Cardiac function
- Balanced steady-state-free precession (SSFP)
- Gadofosveset trisodium
- Gadopentetate dimeglumine
- 32-channel coil
- Sensitivity encoding (SENSE)
The assessment of left ventricular (LV) volumes and function is of high importance in patients with cardiovascular disease as it has therapeutic and prognostic consequences . Cardiovascular magnetic resonance (CMR) is the gold standard method for the assessment of cardiac function and regional wall motion [1–3]. Currently used CMR techniques require the acquisition of multiple short-axis slices during repeated breath-holds to cover the ventricles from base to apex. For severely ill or non-compliant patients, performing multiple breath-holds can be challenging. Additionally, spatial misalignment of slices can occur if patients do not fully comply with breathing commands. A single breath-hold technique allowing the accurate assessment of cardiac volumes and function would significantly reduce scan time and increase patient comfort.
Recently, 3D time resolved cine steady-state-free precession (SSFP) sequences using sensitivity encoding (SENSE) undersampling were introduced for ventricular functional assessment within a single breath-hold . Other available approaches to accelerate image acquisition include parallel imaging  and undersampled reconstruction kt-techniques [6, 7]. All these time-accelerated acquisition techniques suffer from a reduction in blood to myocardium contrast, compared to standard multi-slice (M2D) sequences with repeated breath-holds. The reduced blood to myocardium contrast results from limited inflow of blood into the acquisition volume, which covers the complete ventricle . This reduced contrast can hamper a clear delineation of endocardial borders, which is necessary for the precise assessment of left ventricular volumes and function.
A recent study indicated, that the administration of an extravascular contrast agent immediately prior to image acquisition results in an increase in blood to myocardium contrast using a 3D-SSFP sequence . Extravascular contrast agents rapidly redistribute into extravascular tissues, thereby limiting the contrast between myocardium and blood. Additionally, this type of contrast agent has a 5- to 6-fold lower R1 relaxivity at 1.5 T compared to novel intravascular contrast agents .
In this study, the following hypothesis was tested: The administration of an intravascular contrast agent in combination with a single-breath-hold 3D-whole-heart cine SSFP sequence using a 32-channel cardiac coil and SENSE undersampling allows a more accurate assessment of the left ventricular function compared to an extravascular contrast agent.
Study population and design
Fourteen patients with congenital heart disease (age: 23 to 43 years; median: 36 years) were prospectively enrolled into this study. All patients were in sinus rhythm. Written informed consent was obtained from all patients. The study was approved by the institutional review board (IRB No. 07/Q0704/2) and was registered with the Medicines and Healthcare Products Regulatory Agency (MHRA Study No. 28482/0002/001-0001, EudraCT number 2006–007042). Only individuals without contraindications to CMR and MR contrast agents were included in this study. Any suspicion of kidney impairment resulted in exclusion from the study. Patients were given a take-home informed consent regarding MR contrast media.
Left Ventricular Functional Parameters
3D-SSFP + GdD
3D-SSFP + GdT
ED volume (ml)
ES volume (ml)
Ejection fraction (%)
Qualitative image analysis
Image quality scoring system
Poor-image quality: non diagnostic
Endocardial border barely or not delineable
Endocardial border moderately or partly delineable
Endocardial border well delineable
Endocardial border excellently delineable
Quantitative image analysis
Mean signal intensities (S) of the myocardium and blood pool were measured at end diastole. For assessment of the signal of the myocardium the region of interest (ROI) was placed in the septal myocardium of the mid-ventricle. The signal of blood was defined as the mean signal from a ROI drawn in the center of the left ventricle at enddiastole, excluding the papillary muscles. Noise (N) was determined by the standard deviation in the respective ROIs, as parallel imaging was used (SENSE) . The noise (N) for all measurements was determined by the standard deviation in the respective ROIs, as parallel imaging (SENSE) was used.
Ventricular volume analysis
Volumes derived from single-breath-hold 3D-whole-heart cine SSFP sequences (3D-SSFP) after the administration of GdD and GdT for the evaluation of LV volumes were compared to volumes derived from standard multi-slice 2D cine SSFP sequences (M2D-SSFP, reference standard). In a subgroup of patients standard M2D-SSFP, associated with multiple breath-holds, was performed for volumetric and functional ventricular assessment (n = 8). M2D-SSFP scans were performed on both days in each patient prior to the administration of the respective contrast agent. We included this group, to intra-individually compare M2D-SSFP (as clinical reference standard) to 3D-SSFP scans regarding ventricular function. The four chamber view was used to plan the short-axis stack. The most basal slice was defined when at least 50% of the LV myocardial circumference was visible at end-diastolic and end-systolic short axis slices. The end-diastole and end-systole were defined as the mid-ventricular temporal frame, in which the image showed the largest and smallest LV cavity area. Papillary muscles were excluded from the analysis of the ventricular volume . Measurements were performed by MRM and GFG. Left ventricular volume and ejection fraction were compared between both methods (Table 1).
For interobserver variability two independent observers analyzed 8 of the datasets. For intraobserver variability the same datasets were evaluated twice by the same observer. Datasets were used for the analysis of intraobserver and interobserver variability were randomly chosen for both analysis.
Continuous variables are reported as mean ± standard deviation. Paired and unpaired t-testing was used to compare continuous variables, as appropriate. The agreement between observers regarding image quality grading was assessed by the Wilcoxon signed rank test. A p-value < 0.05 was considered statistically significant. Linear regression analysis was used to assess the volumetric agreement of 3D-SSFP with M2D-SSFP sequences. Bland–Altman analysis was used to assess differences and bias .
Fourteen patients were enrolled into the study protocol. No side effects were correlated to the use of either contrast agent (> 1 year follow-up). There was no significant (p > 0.05) difference in heart rate between the first and second scan session.
Qualitative image analysis
Image quality of single-breath-hold 3D-whole-heart cine SSFP sequence (3D-SSFP) after the administration of GdD and GdT were compared (Figure 1). GdT improved image quality significantly (p < 0.05) in 3D-SSFP scans compared to GdD (Figure 2).
Quantitative image analysis
Results from the blood to myocardial contrast analysis (BMC) and contrast to noise ratios (CNR) are shown in Figure 3. After GdT administration, single-breath-hold 3D-whole-heart cine SSFP (3D-SSFP) images demonstrated a significantly (p < 0.05) higher BMC and CNR compared to 3D-SSFP images after GdD administration (Figure 3).
Left ventricular volume analysis
Left ventricular parameters for all scan groups are shown in Table 1. No significant difference between the M2D-SSFP and respective single-breath-hold 3D-whole-heart cine SSFP sequence (3D-SSFP) scans on the same day were measured for LVEDV, LVESV and LVEF.
95% Confidence interval (ml)
3D-SSFP + GdD
3D-SSFP + GdT
95% Confidence interval
M2D-SSFP + GdD
3D-SSFP + GdD
M2D-SSFP + GdT
3D-SSFP + GdT
Scan time for single-breath-hold 3D-whole-heart cine SSFP sequences (3D-SSFP) was significantly shorter compared to M2D-SSFP sequences (18-25 s vs. 4–5 mins, p < 0.05).
In this prospective study, the administration of an intravascular contrast agent in combination with a single-breath-hold 3D-whole-heart cine SSFP sequence (3D-SSFP) improved image quality significantly. This resulted in a more accurate assessment of LV function compared to the use of an extravascular contrast agent. The use of 32-channel-coil technology and a SENSE factor of 4, enabled imaging with a temporal and spatial resolution comparable to standard multi-slice 2D cine SSFP sequences (M2D-SSFP) . The use of GdT improved the delineation of left ventricular endocardial borders and enabled the accurate quantification of LV volumes and function within one breath-hold. The higher CNR and BMC measured after the administration of GdT, resulted from the higher relaxivity of the contrast agent and its prolonged intravascular half life, compared to the extravascular contrast agent used [15–23]. This study demonstrated that 3D assessment of ventricular function with high accuracy is feasible in a significantly shortened imaging time. Severely ill patients may particularly benefit from shorter scans times, while scanning costs are reduced.
Gadofosveset trisodium reversibly binds to serum albumin with a binding fraction of approximately 85% and therefore predominately remains in the blood pool. High contrast imaging was shown to be feasible up to 60 min after contrast agent administration . The binding to albumin lowers the rate of molecular tumbling of the gadolinium, resulting in a better rotational correlation with water protons. This yields a 5- to 6-fold greater R1 relaxivity at 1.5 T compared with conventional gadolinium contrast agents . With regard to NSF, no cases have been associated with the use of gadofosveset trisodium so far. Due to the greater relaxivity, gadofosveset can be administered at a 3-fold lower dose, while providing greater enhancement from the ventricular blood pool. The usefulness of intravascular contrast agents for the assessment of 3D vascular structures has already been demonstrated [24, 25].
A recently published study focused on the direct comparison of non-contrast-enhanced 2D-SSFP sequences vs. non-contrast-enhanced 3D acquisitions and gadopentetate dimeglumine enhanced 3D acquisitions . This study demonstrated that the CNR between blood and myocardium in 3D acquisitions increases directly after the administration of an extravascular contrast agent.
The administered contrast agents would also have had an effect on gradient echo (GRE) sequences. GRE profits from shortening the T1-relaxtion time of blood. In particular, GRE is often used to study myocardial function at higher field strength (e.g. 3 Tesla). However, a high blood contrast with GRE is only possible through in-flow enhancement in 2D imaging or after administration of a contrast agent. In spoiled GRE the image contrast is proportional to sqrt(TR/2 T1) for the optimal flip-angle and thus profits from shortening T1. However, even for a short T1 relaxation time the overall signal is usually lower than in SSFP imaging, in which case the contrast is proportional to 0.5*sqrt(T2/T1) resulting in a relatively high signal and thus SNR. Although a contrast agent is shortening both the T1 and T2 relaxation time, the T1 effect is usually higher. Therefore, in SSFP a signal increase can also be found after contrast agent administration. The signal enhancement can be calculated for a tissue (e.g. blood) for a given contrast agent concentration (assuming the T1 of the blood reduces from 1400 ms to 220 ms). In spoiled GRE the blood signal would increase from 0.035 to 0.09. In SSFP the blood signal would increase from 0.22 to 0.4 (taking also the small T2 shortening into account). Although the relative signal increase is larger in GRE the overall signal amplitude is higher in SSFP imaging resulting in a higher SNR of SSFP in comparison to GRE.
Other potential applications of 3D techniques include pharmacological stress MRI scans of the myocardium. Traditionally, three short axis slices with intra-slice gaps are planned through the LV . Each slice has to be acquired within a separate breath-hold during stress conditions. A 3D approach may offer an attractive alternative, as it is a fast acquisition method, which allows covering both ventricles within one breath-hold with high temporal resolution. The administration of an intravascular contrast agent may be beneficial, as a clearer delineation of the endocardial border is possible, which is especially important for the assessment of regional wall motion. Its use for myocardial perfusion stress testing is currently under investigation.
In this study we could demonstrate, that the single breath-hold assessment of cardiac function using a highly accelerated 3D single breath-hold acquisition technique in combination with a low dose (0.03 mmol/kg) of an intravascular contrast agent allows for the accurate and fast assessment of LV function.
In conclusion, the use of a 3D-whole-heart cine SSFP sequence in combination with 32-channel technology and an intravascular contrast agent allows for the accurate and fast assessment of LV function within a single breath-hold.
No conclusion from our data is applicable to patients with arrhythmia. It is also important to note that the volumetric data were acquired at different time points during separate CMR examinations in a resting patient, which may cause slight changes in ventricular volumes. As this technique relies on contrast enhancement after the administration of a contrast agent, it is not applicable in patients with renal impairment. The role of intravascular contrast agents for imaging myocardial delayed enhancement is subject to on-going investigations and has not been addressed in the current study. In case information on myocardial scar imaging is relevant, gadopentetate dimeglumine may be the preferred contrast agent at the moment. However, improved cardiovascular imaging can be provided additional to faster functional assessment of the LV using intravascular contrast agents, which may be very useful in patients with congenital heart disease. Significant flow artifacts in the ventricles were not observed in our patient population during the cardiac cycle as none of our patients had abnormal inflow or outflow of the cardiac chambers.
The authors acknowledge financial support from the Department of Health via the National Institute for Health Research (NIHR) comprehensive Biomedical Research Centre award to Guy’s & St Thomas’ NHS Foundation Trust in partnership with King’s College London and King’s College Hospital NHS Foundation Trust.
- Pennell DJ, Sechtem UP, Higgins CB, Manning WJ, Pohost GM, Rademakers FE, van Rossum AC, Shaw LJ, Yucel EK: Clinical indications for cardiovascular magnetic resonance (CMR): Consensus Panel report. Eur Heart J. 2004, 25: 1940-65. 10.1016/j.ehj.2004.06.040.View ArticlePubMedGoogle Scholar
- Pattynama PM, Lamb HJ, van der Velde EA, van der Wall EE, de Roos A: Left ventricular measurements with cine and spin-echo MR imaging: a study of reproducibility with variance component analysis. Radiology. 1993, 187: 261-68.View ArticlePubMedGoogle Scholar
- Moon JC, Lorenz CH, Francis JM, Smith GC, Pennell DJ: Breath- hold FLASH and FISP cardiovascular MR imaging: left ventricular volume differences and reproducibility. Radiology. 2002, 223: 789-97. 10.1148/radiol.2233011181.View ArticlePubMedGoogle Scholar
- Muthurangu V, Noble N, Boubertakh R, et al: Single breath-hold 3D cine imaging of the Heart: a non-angulated isotrpoic acquisition using SENSE on a 32 channel system. Proc 14th Annual Meeting ISMRM, Seattle. 2006, 330: -Google Scholar
- Parish V, Hussain T, Beerbaum P, Greil G, Nagel E, Razavi R, Schaeffter T, Uribe S: Single breath-hold assessment of ventricular volumes using 32-channel coil technology and an extracellular contrast agent. J Magn Reson Imaging. 2010, 31: 838-44. 10.1002/jmri.22061.View ArticlePubMedGoogle Scholar
- Maredia N, Kozerke S, Larghat A, Abidin N, Greenwood JP, Boesiger P, Plein S: Measurement of left ventricular dimensions with contrast-enhanced three-dimensional cine imaging facilitated by k-t SENSE. Journal of cardiovascular magnetic resonance : official journal of the Society for Cardiovascular Magnetic Resonance. 2008, 10: 27-10.1186/1532-429X-10-27.View ArticleGoogle Scholar
- Jahnke C, Nagel E, Gebker R, Bornstedt A, Schnackenburg B, Kozerke S, Fleck E, Paetsch I: Four-dimensional single breathhold magnetic resonance imaging using kt-BLAST enables reliable assessment of left- and right-ventricular volumes and mass. J Magn Reson Imaging. 2007, 25: 737-42. 10.1002/jmri.20877.View ArticlePubMedGoogle Scholar
- Uribe S, Tangchaoren T, Parish V, Wolf I, Razavi R, Greil G, Schaeffter T: Volumetric cardiac quantification by using 3D dual-phase whole-heart MR imaging. Radiology. 2008, 248: 606-14. 10.1148/radiol.2482071568.View ArticlePubMedGoogle Scholar
- Ersoy H, Jacobs P, Kent CK, Prince MR: Blood pool MR angiography of aortic stent-graft endoleak. AJR Am J Roentgenol. 2004, 182: 1181-86.View ArticlePubMedGoogle Scholar
- McConnell MV, Khasgiwala VC, Savord BJ, Chen MH, Chuang ML, Edelman RR, Manning WJ: Comparison of respiratory suppression methods and navigator locations for MR coronary angiography. AJR Am J Roentgenol. 1997, 168: 1369-75.View ArticlePubMedGoogle Scholar
- Pruessmann KP, Weiger M, Scheidegger MB, Boesiger P: SENSE: sensitivity encoding for fast MRI. Magn Reson Med. 1999, 42: 952-62. 10.1002/(SICI)1522-2594(199911)42:5<952::AID-MRM16>3.0.CO;2-S.View ArticlePubMedGoogle Scholar
- Hamdan A, Kelle S, Schnackenburg B, Wellnhofer E, Fleck E, Nagel E: Single-breathhold four-dimensional assessment of left ventricular volumes and function using k-t BLAST after application of extracellular contrast agent at 3 Tesla. J Magn Reson Imaging. 2008, 27: 1028-36. 10.1002/jmri.21332.View ArticlePubMedGoogle Scholar
- Arai AE, Epstein FH, Bove KE, Wolff SD: Visualization of aortic valve leaflets using black blood MRI. J Magn Reson Imaging. 1999, 10: 771-77. 10.1002/(SICI)1522-2586(199911)10:5<771::AID-JMRI22>3.0.CO;2-Q.View ArticlePubMedGoogle Scholar
- Bland JM, Altman DG: Statistical methods for assessing agree- ment between two methods of clinical measurement. Lancet. 1986, 1: 307-11.View ArticlePubMedGoogle Scholar
- Hartmann M, Wiethoff AJ, Hentrich HR, Rohrer M: Initial imaging recommendations for Vasovist angiography. Eur Radiol. 2006, 16 (Suppl 2): B15-23.View ArticlePubMedGoogle Scholar
- Stuber M, Botnar RM, Danias PG, McConnell MV, Kissinger KV, Yucel EK, Manning WJ: Contrast agent-enhanced, free-breathing, three-dimensional coronary magnetic resonance angiography. Journal of magnetic resonance imaging : JMRI. 1999, 10: 790-99. 10.1002/(SICI)1522-2586(199911)10:5<790::AID-JMRI25>3.0.CO;2-I.View ArticlePubMedGoogle Scholar
- Taylor AM, Panting JR, Keegan J, Gatehouse PD, Jhooti P, Yang GZ, McGill S, Francis JM, Burman ED, Firmin DN, Pennell DJ: Use of the intravascular contrast agent NC100150 injection in spin-echo and gradient-echo imaging of the heart. Journal of cardiovascular magnetic resonance : official journal of the Society for Cardiovascular Magnetic Resonance. 1999, 1: 23-32. 10.3109/10976649909080830.View ArticleGoogle Scholar
- Gerretsen SC, Versluis B, Bekkers SC, Leiner T: Cardiac cine MRI: comparison of 1.5T, non-enhanced 3.0T and blood pool enhanced 3.0T imaging. Eur J Radiol. 2008, 65: 80-5. 10.1016/j.ejrad.2007.11.004.View ArticlePubMedGoogle Scholar
- Wagenseil JE, Johansson LO, Lorenz CH: Characterization of t1 relaxation and blood-myocardial contrast enhancement of NC100150 injection in cardiac MRI. Journal of magnetic resonance imaging : JMRI. 1999, 10: 784-89. 10.1002/(SICI)1522-2586(199911)10:5<784::AID-JMRI24>3.0.CO;2-A.View ArticlePubMedGoogle Scholar
- Saeed M, Wendland MF, Higgins CB: Blood pool MR contrast agents for cardiovascular imaging. Journal of magnetic resonance imaging : JMRI. 2000, 12: 890-98. 10.1002/1522-2586(200012)12:6<890::AID-JMRI12>3.0.CO;2-K.View ArticlePubMedGoogle Scholar
- Huber S, Muthupillai R, Mojibian H, Cheong B, Kouwenhoven M, Flamm SD: Rapid assessment of regional and global left ventricular function using three-dimensional k-t BLAST imaging. Magnetic resonance imaging. 2008, 26: 727-38. 10.1016/j.mri.2008.01.027.View ArticlePubMedGoogle Scholar
- Ritter CO, Wilke A, Wichmann T, Beer M, Hahn D, Kostler H: Comparison of intravascular and extracellular contrast media for absolute quantification of myocardial rest-perfusion using high-resolution MRI. Journal of magnetic resonance imaging : JMRI. 2011, 33: 1047-51. 10.1002/jmri.22557.View ArticlePubMedGoogle Scholar
- Nezafat R, Herzka D, Stehning C, Peters DC, Nehrke K, Manning WJ: Inflow quantification in three-dimensional cardiovascular MR imaging. Journal of magnetic resonance imaging : JMRI. 2008, 28: 1273-79. 10.1002/jmri.21493.View ArticlePubMedGoogle Scholar
- Makowski MR, Wiethoff AJ, Uribe S, Parish V, Botnar RM, Bell A, Kiesewetter C, Beerbaum P, Jansen CH, Razavi R, et al: Congenital Heart Disease: Cardiovascular MR Imaging by Using an Intravascular Blood Pool Contrast Agent. Radiology. 2011, 260: 680-88. 10.1148/radiol.11102327.View ArticlePubMedGoogle Scholar
- Naehle CP, Kaestner M, Muller A, Willinek WW, Gieseke J, Schild HH, Thomas D: First-pass and steady-state MR angiography of thoracic vasculature in children and adolescents. JACC Cardiovasc Imaging. 2010, 3: 504-13. 10.1016/j.jcmg.2009.12.015.View ArticlePubMedGoogle Scholar
- Jahnke C, Nagel E, Gebker R, Kokocinski T, Kelle S, Manka R, Fleck E, Paetsch I: Prognostic value of cardiac magnetic resonance stress tests: adenosine stress perfusion and dobutamine stress wall motion imaging. Circulation. 2007, 115: 1769-76. 10.1161/CIRCULATIONAHA.106.652016.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.