A non-invasive clinical application of wave intensity analysis based on ultrahigh temporal resolution phase-contrast cardiovascular magnetic resonance
© Biglino et al.; licensee BioMed Central Ltd. 2012
Received: 22 March 2012
Accepted: 19 July 2012
Published: 9 August 2012
Wave intensity analysis, traditionally derived from pressure and velocity data, can be formulated using velocity and area. Flow-velocity and area can both be derived from high-resolution phase-contrast cardiovascular magnetic resonance (PC-CMR). In this study, very high temporal resolution PC-CMR data is processed using an integrated and semi-automatic technique to derive wave intensity.
Wave intensity was derived in terms of area and velocity changes. These data were directly derived from PC-CMR using a breath-hold spiral sequence accelerated with sensitivity encoding (SENSE). Image processing was integrated in a plug-in for the DICOM viewer OsiriX, including calculations of wave speed and wave intensity. Ascending and descending aortic data from 15 healthy volunteers (30 ± 6 years) data were used to test the method for feasibility, and intra- and inter-observer variability. Ascending aortic data were also compared with results from 15 patients with coronary heart disease (61 ± 13 years) to assess the clinical usefulness of the method.
Rapid image acquisition (11 s breath-hold) and image processing was feasible in all volunteers. Wave speed was physiological (5.8 ± 1.3 m/s ascending aorta, 5.0 ± 0.7 m/s descending aorta) and the wave intensity pattern was consistent with traditionally formulated wave intensity. Wave speed, peak forward compression wave in early systole and peak forward expansion wave in late systole at both locations exhibited overall good intra- and inter-observer variability. Patients with coronary heart disease had higher wave speed (p <0.0001), and lower forward compression wave (p <0.0001) and forward expansion wave (p <0.0005) peaks. This difference is likely related to the older age of the patients’ cohort, indicating stiffer aortas, as well as compromised ventricular function due to their underlying condition.
A non-invasive, semi-automated and reproducible method for performing wave intensity analysis is presented. Its application is facilitated by the use of a very high temporal resolution spiral sequence. A formulation of wave intensity based on area change has also been proposed, involving no assumptions about the cross-sectional shape of the vessel.
KeywordsWave intensity analysis Cardiovascular magnetic resonance Hemodynamics Spiral sequence
The transmission of the arterial pulse wave through the vasculature is dependent on multiple factors including: myocardial function, wall stiffness and wave reflections. Thus, analysis of wave propagation is of great clinical interest, offering insight into the integrated function of the cardiovascular system. One method by which this type of analysis can be accomplished is wave intensity analysis (WIA) .
In the traditional formulation of WIA, pulse wave velocity (c), wave reflections and energy transmission are all derived from measurements of pressure (P) and velocity (U) . Pulse wave velocity can be used to assess arterial stiffness and the wave intensity patterns can be used to assess cardiac function. For instance, the intensity of the forward compression wave (FCW) is related to myocardial contractility and the forward expansion wave (FEW) is related to diastolic relaxation . However, the requirement for invasive pressure measurements has prevented WIA from becoming a routine clinical test. Nevertheless, WIA does offer the possibility of more sophisticated analysis of the cardiovascular system and a non-invasive methodology is desirable.
Recently, a WIA formulation that uses diameter rather than P to assess wave propagation has been proposed . This method is well suited to imaging techniques that assess both vessel size and flow, and was first demonstrated using ultrasound [3, 5]. However, ultrasound is user dependent and limited by acoustic windows. A better imaging methodology for WIA may be phase contrast cardiovascular magnetic resonance (PC-CMR).
PC-CMR is a reference standard method of measuring blood velocity and has been successfully used to assess vessel distension [6, 7]. However, the calculation of c in WIA requires very high temporal resolution velocity and distension data. In addition, accurate arterial distension data can only be acquired with high spatial resolution PC-CMR. Unfortunately, high spatio-temporal resolution PC-CMR has a long acquisition time and cannot be performed in a breath hold. Thus, in previous studies free breathing PC-CMR with multiple signal averages has been used . Although this approach does provide the necessary resolution, it suffers from poor vessel wall delineation due to respiratory artefacts. An alternative approach is to use highly accelerated PC-CMR sequences. Such sequences have the capability of acquiring high-resolution data in a short breath hold. As such data are not corrupted by respiratory artefact, and vessel size should be more accurately measured. In this study, a previously validated spiral PC-CMR sequence accelerated with sensitivity encoding (SENSE)  was used to acquire high spatio-temporal resolution aortic flow data within a breath-hold. The PC-CMR data were then analysed in a semi-automated fashion deriving c, and wave intensity information.
The aims of this study were: i) to demonstrate the feasibility of performing WIA in the ascending and descending aorta of normal volunteers using spiral PC-CMR; ii) to assess intra- and inter-observer variability of the WIA processing package; iii) to use WIA to capture differences between normal volunteers and patients with coronary artery disease, thus showing clinical relevance of the integrated methodology.
Non-invasive wave intensity analysis
where ρ is the density of blood, and D is the vessel distensibility, which equals the relative change in area (dA/A) divided by the change in pressure (dP).
From equation 3, it can be seen that c is the gradient of the U-lnA loop in early systole, when no reflected waves are expected. Thus, c can be calculated in a similar way as per the gradient of the P-U loop , a method often used in traditional WIA.
A full derivation of the area formulation of wave intensity from first principles is provided in Additional file 1.
The study population consisted of 15 healthy adults (11 male, 30 ± 6 years of age) and 15 significantly (p < 0.0001) older patients with confirmed coronary artery disease referred for clinical CMR (10 male, 61 ± 13 years of age). Exclusion criteria were: i) MR incompatible implants, ii) claustrophobia, iii) pregnancy and iv) any cardiovascular disease within the volunteers cohort. The local research ethics committee approved the study.
All imaging was performed on a 1.5 T MR scanner (Avanto, Siemens Medical Solutions, Erlangen, Germany) using two spine coils and one body-matrix coil (giving a total of 12 coil elements). Phase-contrast MR was performed using a previously validated prospectively triggered, velocity encoded spiral spoiled gradient echo sequence (TE/TR: 1.9/4.8 ms, FOV: 400×400×6 mm, matrix: 192×192, VENC: 180 cm/s) . A uniform-density spiral trajectory was used, with 60 spiral interleaves, undersampled by a factor of four (so only 15 spiral interleaves were acquired for each cardiac phase). One spiral interleave was acquired per R-R interval, meaning that 15 R-R intervals were used to acquire all of the data. One additional R-R interval was required at the beginning of imaging to reach a steady state. Thus, the average breath-hold time was 11 s. The sampling pattern was rotated for each cardiac phase so that four consecutive cardiac phases comprised a fully sampled k-space with 60 spiral readouts. The resultant temporal resolution was 9.6 ms and the spatial resolution was 2.1×2.1 mm. In all subjects, PC-CMR was performed in the ascending aorta at the level of the bifurcation of the pulmonary arteries. In addition, in volunteers PC-CMR was also performed in the descending aorta at the level of the diaphragm.
All images were processed using an in-house written plug-in for the open source DICOM software OsiriX (OsiriX Foundation, Geneva, Switzerland) .Segmentation of the aorta was performed on the modulus image using a previously validated semi-automatic registration-based algorithm .In order to ensure accurate delineation of the vessel wall, the ability to alter regions of interest (ROI) was available to the operator. The final ROI was used to both calculate the aortic cross-sectional area (A) and prescribe the area in the phase image from which the mean aortic velocity (U) was calculated. While the propagation algorithm was previously validated , the possibility of performing minor adjustments to the ROI was provided to ensure the goodness of the area curve. The resultant U and A curves were interpolated to a 1 ms temporal resolution and the A curve was smoothed using a Savitzky-Golay filter  of 3rd order with 100 points window size. Then U and lnA were combined into a loop whose optimal linear portion was automatically selected (by finding the portion of the loop in early systole with the best linear fit) and c was calculated from its gradient. Combining knowledge of c with dU and dlnA data, the net, forward and backward dI A were finally plotted. These wave intensity data were post-processed in MATLAB (Mathworks Inc, Natick, MA, USA), where peaks of the FCW and FEW were measured.
Values are presented as mean ± standard deviation. Intra-observer variability was verified by the same operator repeating the analysis one week apart. Inter-observer variability was assessed by two different blinded operators performing the analysis separately. Values of c, peak FCW and peak FEW from the volunteers cohort were tested for intra- and inter-observer variability by quantifying the intraclass correlation coefficient (ICC) (two way random effects model, SPSS, IBM), both in the ascending and descending aorta. Bias and limits of agreement from Bland-Altman analysis were also assessed. Differences between the two cohorts (volunteers vs. patients) were assessed using unpaired t test, and p < 0.05 was regarded as statistically significant.
Assessment of intra- and inter-observer variability of ascending and descending aortic measurements
Ascending aorta (n = 15)
Descending aorta (n = 15)
peak FCW (×10 -5 m/s)
peak FEW (×10 -5 m/s)
peak FCW (×10 -5 m/s)
peak FEW (×10 -5 m/s)
95 % CI
Limits of agreement
Ascending aorta (n = 15)
Descending aorta (n = 15)
peak FCW (×10 -5 m/s)
peak FEW (×10 -5 m/s)
peak FCW (×10 -5 m/s)
peak FEW (×10 -5 m/s)
95 % CI
Limits of agreement
Comparison between normal volunteers and patients
Comparison of wave speed and wave intensity peaks between healthy volunteers and patients
Volunteers(n = 15)
Patients(n = 15)
5.8 ± 1.3
9.5 ± 2.4
peak FCW (×10-5 m/s)
11.5 ± 5.2
3.1 ± 2.5
peak FEW (×10-5 m/s)
1.6 ± 0.7
0.6 ± 0.4
In this study we have shown that it possible to reliably perform WIA using PC-CMR. This technique is dependent on a rapid high resolution PC-CMR sequence that was acquired in a breath hold. Thus, images were not degraded by respiratory artefact, which improved the assessment of vessel distension. We have further demonstrated that WIA measures can easily and robustly differentiate normal volunteers from older patients with coronary artery disease, detecting age-related differences in arterial wall stiffness and indicating a potential for differentiating between health and disease. This suggests that the technique does have clinical utility, although it will be important to test in other patient groups with milder disease.
Wave intensity analysis is a proven technique that allows comprehensive assessment of cardiac and vascular function. Previous invasive studies have shown that the FCW correlates with left ventricular dP/dt (myocardial contractility), while FEW correlates with left ventricular Tau (diastolic relaxation) . In addition, through the measurement of c, arterial wall stiffness can also be evaluated. Unfortunately, the traditional formulation of WIA requires invasive pressure measurements and this has limited its uptake into routine clinical practice.
Recently, a formulation of WIA has been proposed that utilizes vessel distension rather pressure. Initial phantom experiments using ultrasound to measure velocity and distension have shown excellent agreement with the traditional invasive methodology . However, ultrasound does have some disadvantages such as operator dependence and limited acoustic windows. An alternative to ultrasound is PC-CMR, which has the ability to simultaneously measure velocity and distension in the true short axis of any vessel. Unfortunately, the requirement for high spatio-temporal resolution data leads to long acquisition times and previous studies have had to use free breathing PC-CMR. This is problematic, as respiratory artefact can lead to significant edge blurring and inaccurate vessel distension measurements. In this study, we combined efficient spiral k-space trajectories with SENSE to acquire high resolution PC-CMR data in a short breath hold. This sequence has previously been validated and offers a way of acquiring velocity and distension data without respiratory contamination of the images. However, this sequence does suffer from reduced SNR due to undersampling, and edge blurring due to spiral imaging. Because of the need of making the sequence clinically useful (i.e. acquired in a short, manageable breath hold), time scan was pushed as fast as possible, requiring a SENSE factor of 4. This solution admittedly impinges on SNR, but nevertheless image quality was sufficient to semi-automatically segment the data in all cases. The other difference between this study and a previous retrospective MR study  was the use of area rather than diameter for evaluation of vessel distension. Diameter is the only measurement available in combined ultrasound imaging and Doppler assessment. However in MR, the vessel short axis is acquired and this lends itself to assessment of area. Area measurement has the benefits of not assuming circularity, as well as providing better reproducibility. Thus, it can be argued that area is the most valid way of measuring vessel distension using MR. Of course, this approach does require vessel segmentation of every frame, which is not clinically feasible when done manually. Thus, we used a semi-automatic segmentation algorithm that allows rapid processing with minimal user manipulation of the ROI’s. By combining rapid MR imaging with rapid processing we believe that this implementation of WIA can be used successfully in the clinical environment.
It should be noted that in this study a time correction  was not applied to the data as the sampling frequency was the same in all cases. However, if the sampling frequency were different in different patient groups (e.g. to reduce breath hold time) then time correction would be necessary. This would be best done by using dU/dt x dlnA/dt in the WIA formulation and could be easily implemented in our plug-in.
In volunteers, WIA patterns and values of c produced by this technique were in agreement with previous invasive and non-invasive studies [18, 19].Values of c also increased with age as observed in previous studies [20, 21], although combining healthy subjects and patients may have confounded this result. Furthermore, there were large differences in c, FCW and FEW between older patients with coronary artery disease and younger healthy volunteers. This difference is perhaps unsurprising, due to the significantly (p < 0.0001) older age of the patients’ cohort, reflecting stiffer vessels, as well as their likely reduced systolic and diastolic function. Nevertheless, it does demonstrate the clinical utility of non-invasive WIA. In addition, these measures are more load independent than other common MR measures (such as ejection fraction) and therefore do provide added value .
For a new technique to be accepted clinically it must have proven reproducibility. We have shown that in normal volunteers there is good reproducibility for c and FCW in both the ascending and descending aorta. However, in the ascending aorta there was poor reproducibility for FEW compared to the descending aorta. This is probably because of the low-end systolic signal in the ascending compared to descending aorta, due to reduced in-flow enhancement. This hampers automated segmentation and means that currently more user interaction is required, introducing greater errors in measurement. Possible solutions to this problem may lay in the development of new steady state free precession PC-CMR sequences that have high blood signal throughout the cardiac cycle. Such sequences should provide images that are much more amenable to automated segmentation with very little user interaction. Nevertheless, it should be noted that there is an almost 3-fold difference in FEW between patients and volunteers. Thus, even with poor reproducibility it is still possible to differentiate between health and disease. Of course, greater reproducibility is necessary if milder forms of disease are to be diagnosed using this technique.
R 2 coefficients for the linear part of the lnA-U loops in the volunteers’ cohort, both for ascending and descending aortic data, indicating the goodness of the linear fit
R2coefficient linear part
The main limitation of this technique is that the spiral SENSE sequence is prospectively gated and thus does not acquire data in end diastole. For WIA in central vessels this is not a major problem as there are few reflected waves in this part of the cardiac cycle. However, for other parts of the vasculature inability to acquire the whole cardiac cycle may limit this technique. Future work on retrospective gating is thus important in these situations.
A non-invasive method for calculating wave intensity based on PC-CMR data has been presented. This method depends on rapid breath-hold PC-CMR and semi-automated image processing. We have shown that this technique is reproducible for c and FCW and it is able to differentiate between patients and volunteers of different age and underlying cardiovascular condition. In the future this methodology may be useful in detection of more subtle changes in vascular and cardiac function in diseases like systemic and pulmonary hypertension.
The authors gratefully acknowledge the support of Fondation Leducq, Royal Academy of Engineering/EPSRC, National Institute of Health Research and British Heart Foundation. The authors also acknowledge Dr James Moon, Dr Daniel Sado and Dr Steve White of the Heart Hospital in London for acquiring the patients’ data.
- Parker KH, Jones CJ: Forward and backward running waves in the arteries: analysis using the method of characteristics. J Biomech Eng. 1990, 112: 322-326. 10.1115/1.2891191.View ArticlePubMedGoogle Scholar
- Parker KH: An introduction to wave intensity analysis. Med Biol Eng Comput. 2009, 47: 175-188. 10.1007/s11517-009-0439-y.View ArticlePubMedGoogle Scholar
- Ohte N, Narita H, Sugawara M, Niki K, Okada T, Harada J, Kimura G: Clinical usefulness of carotid arterial wave intensity in assessing left ventricular systolic and early diastolic performance. Hear Vessel. 2003, 18: 107-111. 10.1007/s00380-003-0700-5.View ArticleGoogle Scholar
- Feng J, Khir AW: Determination of wave speed and wave separation in the arteries using diameter and velocity. J Biomech. 2010, 43: 455-462. 10.1016/j.jbiomech.2009.09.046.View ArticlePubMedGoogle Scholar
- Borlotti A, Vermeersch S, Rietzschel E, Segers P, Khir AW: A comparison between local wave speed in the carotid and femoral arteries in healthy humans: application of a new method. Conf Proc IEEE Eng Med Biol Soc. 2010, 2010: 2857-2860.PubMedGoogle Scholar
- Chai P, Mohiaddin R: How we perform cardiovascular magnetic resonance flow assessment using phase-contrast velocity mapping. J Cardiovasc MagnReson. 2005, 7: 705-716.Google Scholar
- Beerbaum P, Korperich H, Barth P, Esdorn H, Gieseke J, Meyer H: Noninvasive quantification of left-toright shunt in pediatric patients: phase-contrast cine magnetic resonance imaging compared with invasive oximetry. Circulation. 2001, 103: 2476-2482. 10.1161/01.CIR.103.20.2476.View ArticlePubMedGoogle Scholar
- Li Y, Borlotti A, Hickson SS, McEniery CM, Wilkinson IB, Khir AW: Using magnetic resonance imaging measurements for the determination of local wave speed and arrival time of reflected waves in human ascending aorta. Conf Proc IEEE Eng Med Biol Soc. 2010, 2010: 5153-5156.PubMedGoogle Scholar
- Steeden JA, Atkinson D, Hansen MS, Taylor AM, Muthurangu V: Rapid flow assessment of congenital heart disease with high-spatiotemporal-resolution gated spiral phase-contrast MR imaging. Radiology. 2011, 260: 79-87. 10.1148/radiol.11101844.PubMed CentralView ArticlePubMedGoogle Scholar
- Tijsseling AS, Anderson A: Johannes von Kries and the history of waterhammer. J Hydr Eng. 2007, 133: 1-8. 10.1061/(ASCE)0733-9429(2007)133:1(1).View ArticleGoogle Scholar
- Khir AW, O’Brien A, Gibbs JS, Parker KH: Determination of wave speed and wave separation in the arteries. J Biomech. 2001, 34: 1145-1155. 10.1016/S0021-9290(01)00076-8.View ArticlePubMedGoogle Scholar
- Rossete A, Spadola L, Ratib O: OsiriX: an open-source software for navigating in multidimensional DICOM images. J Digit Imaging. 2004, 17: 205-216. 10.1007/s10278-004-1014-6.View ArticleGoogle Scholar
- Odille F, Steeden JA, Muthurangu V, Atkinson D: Automatic segmentation propagation of the aorta in real-time phase contrast MRI using nonrigid registration. J Magn Reson Imaging. 2011, 33: 232-238. 10.1002/jmri.22402.View ArticlePubMedGoogle Scholar
- Savitzky A, Golay MJE: Smoothing and differentiation of data by simplified least squares procedures. Anal Chem. 1964, 36: 1627-1639. 10.1021/ac60214a047.View ArticleGoogle Scholar
- Feng J, Khir AW: Determination of wave intensity in flexible tubes using measured diameter and velocity. Conf Proc IEEE Eng Med Biol Soc. 2007, 2007: 985-988.PubMedGoogle Scholar
- Li Y, Khir AW: Experimental validation of non-invasive and fluid density independent methods for the determination of local wave speed and arrival time of reflected wave. J Biomech. 2011, 44: 1393-1399. 10.1016/j.jbiomech.2010.12.019.View ArticlePubMedGoogle Scholar
- Kolyva C, Pantalos GM, Giridharan GA, Pepper JR, Khir A: Discerning aortic waves during intra-aortic balloon pumping and their relation to benefits of counterpulsation in humans. J Appl Physiol. 2009, 107: 1497-1503. 10.1152/japplphysiol.00413.2009.PubMed CentralView ArticlePubMedGoogle Scholar
- Koh TW, Pepper JR, DeSouza AC, Parker KH: Analysis of wave reflections in the arterial system using wave intensity: a novel method for predicting the timing and amplitude of reflected waves. Hear Vessel. 1998, 13: 103-113. 10.1007/BF01747827.View ArticleGoogle Scholar
- Khir AW, Parker KH: Wave intensity in the ascending aorta: effects of arterial occlusion. J Biomech. 2005, 38: 647-655. 10.1016/j.jbiomech.2004.05.039.View ArticlePubMedGoogle Scholar
- Mohiaddin RH, Firmin DN, Longmore DB: Age-related changes of human aortic flow wave velocity measured noninvasively by magnetic resonance imaging. J Appl Physiol. 1993, 74: 492-497.PubMedGoogle Scholar
- Rogers WJ, Hu YL, Coast D, Vido DA, Kramer CM, Pyeritz RE, Reichek N: Age-associated changes in regional aortic pulse wave velocity. J Am Coll Cardiol. 2001, 38: 1123-1129. 10.1016/S0735-1097(01)01504-2.View ArticlePubMedGoogle Scholar
- Kolias TJ, Aaronson KD, Armstrong WF: Doppler-derived dP/dt and -dP/dt predict survival in congestive heart failure. J Am Coll Cardiol. 2000, 36: 1594-1599. 10.1016/S0735-1097(00)00908-6.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.