- Poster presentation
- Open Access
MRI guided electrophysiological intervention with a voltage-based electro-anatomic mapping system
© Tse et al; licensee BioMed Central Ltd. 2012
- Published: 1 February 2012
- Coronary Sinus
- Scanner Room
- Image Quality Reduction
- Catheter Tracking
- Coronary Sinus Catheter
MRI visualizes luminal & vessel-wall anatomy, and identifies edema & scar tissue, contributing to improved electrophysiological (EP) ablative procedures for treatment of Ventricular Tachycardia & Atrial Fibrillation. MRI-guided EP interventions will be performed for the foreseeable future partially in & outside MRI, due to the need for X-ray/Ultrasound-compliant devices. Electromagnetically tracked catheter procedures, today’s norm for most EP procedure phases; vascular navigation, Electro-Anatomic-Mapping (EAM, the diagnostic and therapeutic phases), can only be performed outside MRI. Separate MRI tracking is required in MRI, complicating EP procedures which require moving in & out of the bore [1, 2]. Continuous catheter tracking using a single system would allow registration-free EAM in & outside MRI. The goal was developing an MR-compatible St. Jude Medical (SJM) EnSite NavX (ESN) voltage-based tracking . ESN applies 5.8/8.0 kHz voltage bursts between 3 pairs of electrodes on the chest, detecting a catheter’s position , so a challenge for intra-MRI use is MR gradient ramps which interfere with ESN operation. Minimal MR Image Quality (IQ) reduction also needs to be insured, as well as <2oC patient-skin heating due to components in MRI.
Fig.1(5) shows simultaneous ESN & MR tracking of the dual ESN&MR-tracked catheter in the phantom, verifying the lack of interference between the 2 methods. 3 EAM & MRI imaging experiments were performed in swine, also equipped with sensors to measure temperature at the surface ESN electrodes. Fig.1(6) shows EAM during imaging where catheter tracking of 2 catheters (reference catheter in the Coronary Sinus (CS) & mapping catheter in the Left Atrium) was performed simultaneously on the ESN , with <5% positional error of the CS catheter relative to its position outside MRI. IQ Reduction was <5% in both SE & GRE, with tracking updated during 60-80% of sequence duration. TR elongation was required in TR<4ms sequences. Electrode temperature rise was <1oC with 4 Watt/kg SAR sequences. Surface burns did not occur.
An MR-compatible ESN system permits registration-free, minimal-heating, EAM for EP procedure in & outside MRI, with simultaneous imaging possible.
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.