- Poster presentation
- Open Access
Numbers needed to detect a significant change in myocardial strain and left ventricular twist measured by complementary spatial modulation of magnetization (CSPAMM)
© Swoboda et al; licensee BioMed Central Ltd. 2012
- Published: 1 February 2012
- Spatial Modulation
- Circumferential Strain
- Radial Strain
- Surrogate Data
- Short Axis Slice
We have calculated the inter-study reproducibility of myocardial strain and twist measured by CSPAMM and calculated the number of subjects required to detect a significant change in these parameters.
The inter-study reproducibility of parameters measured by CSPAMM such as circumferential strain, radial strain and left ventricular (LV) twist have not been described in detail and it is not known how many subjects are required to detect a significant change.
12 healthy volunteers (6 male, mean age 33±15 years) underwent CMR studies on two separate occasions (mean interval of 8±3 days) using an identical CSPAMM pulse sequence with images acquired in 3 short axis slices at the apex, mid-ventricle and base. Data were analysed using harmonic phase analysis (Tag Track, Gyrotools, Zurich, CH). Circumferential Lagrangian strain, radial Lagrangian strain and rotation were calculated for the three short axis slices. Left ventricular (LV) twist was calculated by subtracting the basal rotation from the apical rotation. The mean difference between paired measurements, the standard deviation (SD) of the differences and the coefficient of variability (CoV) were calculated.
The sample sizes required by CMR to show a clinical change with a power of 90% and an α error of 0.05 were calculated using the formula n = f (α, P) x σ 2 x 2/ δ 2 , where n is the sample size, α the significance level, P the study power required and f the value of the factor for different values of α and P (f = 10.5 for α=0.05 and p = 0.90), with σ the inter-study standard deviation of differences between visits and δ the desired difference to be detected.
Inter-study reproducibility as the percentage difference between the two scans and the equivalent Coefficient of Variation(%)
Mean diff %
Mean diff %
Mean diff %
Mean values of circumferential strain, radial strain and LV twist from 12 healthy volunteers and the sample size required to detect a 10% change in each absolute value.
LV twist (degrees)
CSPAMM tagged CMR with HARP analysis provides reproducible measurements of circumferential strain and twist with small numbers required to detect differences in serial studies. The method may provide powerful surrogate data for clinical studies.
This research has partly been supported by a grant from the British Heart Foundation. This funding body was not involved in the study design, data collection, data analysis or writing of the manuscript.
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.