- Workshop presentation
- Open Access
Investigation of stenotic jets using 3D-PC-UTE
Journal of Cardiovascular Magnetic Resonance volume 14, Article number: W55 (2012)
Conventional phase contrast (PC) velocity measurements fail in situation of high velocities and complex flow, as after a stenosis. Specifically, volume flow and peak velocities are underestimated, and velocity-to-noise ratio (VNR) is decreased due to flow voids. These effects are much dependent on the echo time (TE). Reducing TE will drastically improve the results, both concerning VNR and accuracy.
The method of ultra-short TE (UTE) allows for sub-millisecond TE, and has recently been implemented for 2D through-plane PC(2D PC-UTE). This work aims to investigate peak velocities and visualize stenotic jets in a 3D volume using PC-UTE.
Two flow phantoms, PH1 and PH2, were used in two separate scanning sessions. PH1 consisted of plastic tubes, d=20 mm, in which were placed discs with central, circular holes of d=3 and 9 mm (s1 and s2). PH2 is a commercially available phantom mimicking a carotid bifurcation with a 60% stenosis in the left branch. A steady flow of water doped with Gadovist was pumped through the phantoms. In PH1, the 3D volume covered both s1 and s2. In PH2, the volume covered one slice before the stenosis. Two 3D PC sequences were used: a conventional, cartesian 3D PC (3D PC), and a radial 3D PC-UTE. Both had 12 slices, 14 time frames, TR=6 resp. 8 ms, ETL=5, voxel size 1.5x1.5x3 mm3 and through-plane velocity encoding, venc=700 cm/s. The important difference is TE, which is 3.7 ms for the 3D PC and 0.76 ms for 3D PC-UTE.
For analysis, ROI’s were drawn over the vessel of interest. In PH1, the ROI was chosen with a diameter of 20 mm, and identical ROIs were used for all slices. For PH2, the ROI’s were drawn manually on the velocity images. In both phantoms, identical ROIs were used in the analysis of both datasets.
A quantitative analysis of the peak velocities was made by extracting the peak velocity in each ROI, and averaging that over the time frames for each slice. The standard deviation of the peak velocity was also calculated using data from the different time frames.
In Figure 1, artefacts are evident in the 3D PC contour plots, while the 3D PC-UTE shows a more concentric jet. Figure 2 shows the peak velocities for (a) s1 in PH1 and (b) PH2. In the 3D PC-UTE data, the peak velocity occurs in the slice closest to the stenosis and then decreases in slices more distal to the obstruction. In 3D PC data sets, peak velocity has a much larger variation, and the absolute peak occurs in more distal slices. A limitation of the study is that true peak velocity is now know, but estimations based on the known flow rate and measured jet areas indicate that the true value is at least as high as the 3D PC-UTE estimate.
3D PC-UTE is shown to improve the accuracy and reduce variance in measurements of peak velocity in stenotic jets. The 3D implementation allows for finding the peak velocity even if it is unknown a priori where in the jet it occurs or if there is through-plane motion during the cardiac cycle.
The Swedish Strategic Foundation.
About this article
Cite this article
Bloch, K.M., Nilsson, A. & Ståhlberg, F. Investigation of stenotic jets using 3D-PC-UTE. J Cardiovasc Magn Reson 14, W55 (2012). https://doi.org/10.1186/1532-429X-14-S1-W55
- Peak Velocity
- Phase Contrast Sequence
- Strategic Foundation
- Flow Phantom
- Phase Contrast Data