- Poster presentation
- Open Access
Single breathhold 3D balanced SSFP coronary MRA at 3.0T: a reproducibility study
© Soleimanifard et al; licensee BioMed Central Ltd. 2013
- Published: 30 January 2013
- Vessel Length
- Balance SSFP
- Reconstructed Voxel Size
- bSSFP Sequence
- Oversampling Factor
Balanced steady-state free precession (bSSFP) imaging  has developed as the method of choice for coronary MRA at 1.5T. The bSSFP acquisition, however, is especially susceptible to static magnetic field (B0) inhomogeneities and radio frequency transmit field (B1+) distortions, both of which are more pronounced at higher field strengths, resulting in degraded and variable image quality at 3.0T compared to 1.5T. Although modified bSSFP sequences have been recently developed , these modified implementations too come with notable drawbacks such as longer scan time and lower SNR compared with conventional bSSFP. Therefore, the adoption of bSSFP at 3.0T is still quite limited and it has led to reutilization of spoiled gradient echo techniques at 3.0T, sometimes at the expense of using contrast agents . Recent advances in hardware and software, such as 32-channel receive coils, multi-transmit systems, and localized shimming, may however improve the performance of bSSFP at 3.0T. The purpose of this work was to develop an accelerated single breathhold bSSFP sequence at 3.0T taking advantage of these advances, and to test reproducibility of the implemented sequence.
A three-dimensional, volume-targeted bSSFP sequence was implemented on a 3.0T MR scanner equipped with multi-transmit system and a 32-channel cardiac phased-array coil. Image-based shimming was performed using B0-map  and B1+-map [5, 6] images to minimize field inhomogeneities. Parallel imaging (2.5 SENSE factor with 1.3 oversampling factor) was used to accelerate the acquisition. A half-alpha TR-half preparation pulse with 10 startup RF pulses was used to accelerate the transition to steady-state. Fat suppression was achieved using a spectrally selective saturation pulse. Scan parameters were the following: TR/TE 3.9/1.9 ms, RF excitation angle 50°, field-of-view 300×300×20 mm3, reconstructed voxel size 0.8×0.8×1.0mm3,, acquisition window 105ms, scan time 20.5±2.0s. 15 healthy volunteers and 3 patients with coronary artery disease (CAD) were included (age 38±18 (21-75) years; 8 female), and the acquisition was repeated in 9 subjects. The vessel length, diameter, and sharpness were analyzed using the Soap-Bubble reformatting tool . The intra-observer, inter-observer, and inter-scan reproducibilities were quantified using regression analysis and intra-class correlation coefficient, in a blinded manner.
Length, diameter, and sharpness of coronary arteries in 18 subjects (RCA = 15, LAD = 5)
Reference analysis (subset)
Repeat scan analysis
Vessel Sharpness [%]
The refined 3D bSSFP acquisition, using a state-of-the-art MR scanner, allows accelerated, reproducible imaging of major coronary arteries during a single breathhold in healthy adult subjects and patients with CAD.
NIH/NHLBI (ROIHL084186, ARRA 3R01Hl084186-04S1) and American Heart Association (12PRE11510006).
- Electromedica. 1986, 54: 15-18.Google Scholar
- Lee HL, Shankaranarayanan A, Pohost GM, Nayak KS: Improved coronary MR angiography using wideband steady state free precession at 3 tesla with sub-millimeter resolution. JMRI. 2010, 31: 1224-29. 10.1002/jmri.22150.PubMed CentralView ArticlePubMedGoogle Scholar
- JACC. 2009, 54: 69-76.Google Scholar
- MRM. 2004, 51: 799-806.Google Scholar
- MRM. 2006, 55: 1326-33.Google Scholar
- Muella A, Kouwenhoven M, Naehle CP, Gieseke J, Strach K, Willinek WA, Schild HH, Thomas D: Dual-source radiofrequency transmission with patient-adaptive local radiofrequency shimming for 3.0-T cardiac MR imaging: initial experience. Radiology. 2012, 263: 77-85. 10.1148/radiol.11110347.View ArticleGoogle Scholar
- MRM. 2002, 48: 658-66.Google Scholar
- Kaul MG, Stork A, Bansmann PM, Nolte-Ernsting C, Lund GK, Weber C, Adam G: Evaluation of balanced steady-state free precession (TrueFISP) and K-space segmented gradient echo sequences for 3D coronary MR angiography with navigator gating at 3 Tesla. RoFo. 2004, 176: 1560-65. 10.1055/s-2004-813629.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.