- Poster presentation
- Open Access
Reproducibility of peak and time to peak velocity measurements with a high resolution spiral phase velocity mapping (PVM) sequence
Journal of Cardiovascular Magnetic Resonance volume 15, Article number: E42 (2013)
Myocardial PVM may be used to assess peak velocities and the times to those peaks (TTP) in order to characterise healthy and pathological regional heart motion . However data on the reproducibility of the technique is limited. A previous breathhold (low resolution) study investigated reproducibility for peak radial and circumferential velocities ; however that of TTP measurements was not presented. A higher resolution, navigator-gated study showed that MR had better reproducibility than tissue Doppler imaging for peak and TTP velocity measurements, but only longitudinal velocities were considered . All previous studies have used prospective cardiac gating and late diastolic (atrial systole) peak and TTP velocities have not been investigated. This study presents a comprehensive assessment of the reproducibility of myocardial PVM for assessing peak and TTP velocities throughout the entire cardiac cycle.
An efficient navigator-gated spiral trajectory PVM sequence (1.4 x 1.4 mm acquired spatial resolution, 21 ms acquired temporal resolution; nominal acquisition duration 53 heartbeats) was used to scan basal, mid-ventricular and apical short-axis slices in ten healthy volunteers on two separate occasions. Retrospective cardiac gating was implemented, allowing analysis of all major velocity peaks throughout the cardiac cycle, including atrial systole. The reproducibility of peak velocities and their TTP values was assessed by Bland Altman analysis. Analysis of TTP vales was repeated after normalisation to a fixed systolic and diastolic length.
Reproducibility data are shown in Table 1. All peak velocity reproducibilities are good, including those for atrial systole which is seen in both radial and longitudinal directions. TTP values expressed as ms from the R-wave show good reproducibility for systolic peaks (-7.1+/-24.1 ms (radial)), but are less reproducible for early (26.3+/-35.7 ms) and late diastolic (25.4+/-88.2 ms) peaks due to heart rate differences on the two scanning occasions (see Figure 1). Diastolic reproducibility can be greatly improved by normalising the TTP values to a fixed systolic and diastolic length (1.3±21.3 and 3.0±10.9 ms for early diastolic and late diastolic peaks respectively), as can be seen in Table 1.
High temporal resolution myocardial PVM with retrospective cardiac gating is a highly reproducible technique for measuring all peak velocities in the cardiac cycle, including those in atrial systole. Absolute and normalised systolic TTP values have comparable reproducibility. However the reproducibility of diastolic TTP values is improved following normalisation to a fixed systolic and diastolic length. This is particularly apparent for the late diastolic peak representing atrial systole.
The authors acknowledge the support of Heart Research UK, Imperial College London and NIHR Royal Brompton Cardiovascular Biomedical Research Unit.
Foll : . Circ. 2010
Petersen : . Radiology. 2006
Delfino : . JMRI. 2006
Rights and permissions
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Simpson, R., Keegan, J. & Firmin, D.N. Reproducibility of peak and time to peak velocity measurements with a high resolution spiral phase velocity mapping (PVM) sequence. J Cardiovasc Magn Reson 15 (Suppl 1), E42 (2013). https://doi.org/10.1186/1532-429X-15-S1-E42
- Peak Velocity
- Tissue Doppler Imaging
- Entire Cardiac Cycle
- Spiral Phase
- Atrial Systole