- Poster presentation
- Open Access
High resolution high density CMR circular tagging
© Aghaeifar et al; licensee BioMed Central Ltd. 2013
- Published: 30 January 2013
- Directional Filter
- Angle Image
- 164x192 Matrix
- Siemens Scanner
- Wide Clinical Application
Wall thickening is the dominant factor for the effective ejection of the left ventricle (LV). Circular tagging is the suitable pattern for studying the radial strain which quantifies the wall thickening. High density tagging facilitates the post processing required for this quantification . Following the recent sequence development for circular tagging, in this work we propose a k-spaced based approach for increasing the resolution and density of the circular tags during the reconstruction.
In this study the polar HARP concept, which encodes the position of polar tags in the reconstructed phase, was extended using successive directional filters to decrease distortions that may appear in angle image. The density of taglines was then increased by multiplication of the phase of the angle images. The image resolution was also enhanced to accommodate the high density taglines. The following processes were all embedded in the online reconstruction of the images on the scanner: phase image extraction by polar harp, increasing the tag density through phase multiplication, zero-padding for increasing the spatial resolution, and finally masking the phase image based on the reconstructed magnitude image.
This study was conducted using ICE (Image Calculation Environment), which is the image reconstruction environment for Siemens scanners. ICE is built upon the C++ high-level programming language. The default reconstruction chain was modified and the entire process was embedded in a single functor in a separate pipeline. Total reconstruction time increased less than a fraction of a second for each frame.
The reconstruction chain was then examined on clinical Siemens scanners at 1.5T & 3.0T. Common MR parameters were set as following: 280 mm FOV, 6mm slice thickness, TE/TR= 2.89/52.24 ms, 250 Hz/pixel bandwidth, 15° flip angle, and 164x192 matrix size.
This work demonstrates the possibility of making high resolution high density CMR circular tagging on the myocardium. It further shows reconstruction pipeline has great potential for embedding offline processes. Since frequency and image domain both are accessible simultaneously in reconstruction chain, no further transform required and desired processes execute in almost real-time, which has wide clinical applications due to instant display of the images after acquisition.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.