- Workshop presentation
- Open Access
Improved cardiac gating at 3T with the “3D-QRS” method utilizing MRI-compatible 12-lead ECGs
Journal of Cardiovascular Magnetic Resonance volume 15, Article number: W20 (2013)
Blood is rapidly ejected from the left ventricle during early systole, travels in the aortic arch perpendicular to the MRI’s main field, thus generating a Magnetohydrodynamic (MHD) voltage which is larger than the real ECG QRS complex in high field MRI. The MHD voltage (VMHD) is severally irregular in patients with arrhythmia, since arrhythmic ECG beats are interleaved between successive sinus rhythm (SR) beats. The VMHD overlay in ECG traces can result in intermittent QRS detection, leading to blurred images and longer scan times. Since accurate gating is essential for successful cardiac imaging, we developed a “3D-QRS” method for real-time detection of the QRS complex based on 12-lead ECG traces acquired inside the MRI. We validated this method at 3T in patients with Premature Ventricular Contractions (PVCs), Atrial Fibrillation (AF) and in an exercising athlete with time-varying heart rate.
An MRI-compatible ECG system  was constructed and tested with a GE 3T MRI. It acquired 12-lead ECG traces, computed the “3D-QRS” algorithm for QRS detection, and sent out gating triggers to the scanner gating unit. The 3D-QRS was based on a new representation of the ECG channels, where ECG channels V1-V6 became the third spatial axis, adding a dimension to the conventional ECG time and voltage axes. Fig. 1(a-b) shows the 3D representation of the SR-QRS complex and VMHD. As the QRS complex and VMHD originate from the sinus node and the aortic arch respectively, the channel axis carried information on the propagation from the different cardiac sources (SR, MHD, PVC) to the surface leads (Fig. 1(d)). 3D-QRS presented a unique 3-D kernel of the QRS complex which could be clearly differentiated from VMHD even at 3T. 3D-QRS kernels were also used for separating SR from PVC beats using similar principles (Fig. 1(a,c)). A FFT 2D-cross-correlation subroutine was then used in Matlab to achieve a computational speed of <30msec for detecting QRS complexes in real-time.
Fig. 2(a-d) shows ECGs of an AF patient taken out/inside a 3T MRI: the QRS complexes in V6 at Fig.2(c) are less than 10% of the VMHD amplitude and failed to be identified with conventional 4-lead ECG gating, while the SR-QRS kernels at Fig.2(d) were distinguishable based on their unique 3-D features. Fig. 2(e-h) shows the gating results achieved in 2 AF patients and an exercising athlete subject which produced heart rate variations from 44-87bpm. Gating accuracy was compared in Fig.2(f,h) between 3D-QRS, 4-lead Vectorcardiogram (VCG) and 1-lead Pan-Tompkins algorithm based on V6. 2 set of 20-sec breath-held ECGs were considered in all cases.
In high-field MRI, the 3D-QRS method allowed accurate detection of the QRS, and beat-type separation in arrhythmia patients, which allowed for real-time scanner gating.
NIH U41-RR019703, R43 HL110427-01, AHA 10SDG261039
Tse : AHA & ISMRM. 2012
Rights and permissions
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Tse, Z., Dumoulin, C., Watkins, R. et al. Improved cardiac gating at 3T with the “3D-QRS” method utilizing MRI-compatible 12-lead ECGs. J Cardiovasc Magn Reson 15 (Suppl 1), W20 (2013). https://doi.org/10.1186/1532-429X-15-S1-W20
- Sinus Rhythm
- Atrial Fibrillation Patient
- Scanner Gating
- Athlete Subject
- Produce Heart Rate