4D tissue phase mapping: clinically viable acquisition protocol and new method of visualisation
Journal of Cardiovascular Magnetic Resonance volume 17, Article number: O46 (2015)
Tissue phase mapping (TPM) has been shown to be capable of providing insights into healthy and diseased motion . Typically 3 short-axis slices are acquired, providing full 3-directional velocity vectors but only in 2D slices, preventing the calculation of potentially important parameters such as through-plane strain rate. More recently, several studies have presented 4DTPM [2, 3], however acquisition times can be unfeasibly long, image quality, temporal and spatial resolution is generally poor, and visualising the data is a challenge. This abstract presents initial application of EnSight (CEI, USA) visualisation to 4DTPM data, acquired with a duration, quality and resolution which do not prevent clinical translation.
A navigator-gated black-blood TPM sequence  was adapted to allow 3D acquisitions. A 3D slab with matrix size 120x160x8 (75% slice oversampling) was acquired with spatial resolution of 2x2x4mm and temporal resolution of 24.4ms (40 phases) in a healthy volunteer. PEAK-GRAPPA  factor 5 led to acquisition time of 16m45s. Images were manually segmented using MATLAB (The Mathworks, MA) before importing into EnSight. Vector arrows were used at each pixel to represent speed at each position.
Image quality in all slices can be seen in Figure 1 where magnitude and through-plane phase images during diastole are shown. Figure 2 shows EnSight visualisations of the 3D volume from two views at peak systole and peak diastole, along with velocity-time curves for all slices. EnSight visualisations clearly show regional variation of velocities, for example lowered septal peak diastolic velocities. The graphs show expected motion features, including the notch in radial and circumferential velocities due to isovolumic relaxation. A clear decrease in peak velocities (eg peak diastolic longitudinal velocity) from base to apex can also be seen.
By adapting a 2DTPM sequence, high quality 4DTPM data can be acquired in a scan time that does not prevent clinical use. In-plane and temporal resolution are high, while through plane resolution was sacrificed in favour of acquisition time (it is not expected that this will greatly affect derived parameters). A limitation of the acquisition is that the entire ventricle is not covered. The protocol must therefore be adapted to acquire the pertinent data in future studies. EnSight allows visualisation of 4D data which can provide intuitive understanding of the motion occurring throughout the cardiac cycle. While only two time frames and two views are shown here, in reality the motion can be viewed as a 3D cine which can be interactively rotated in all dimensions. Future acquisitions in patients will highlight the use of such visualisation and may provide a better understanding of the complex cardiac motion.
Simpson : 10.1002/jmri.23756
Lutz : 10.1186/1532-429X-14-74
Steeden : (abstract) 10.1186/1532-429X-16-S1-W30
Menza : ISMRM. 2013, (abstract 4483)
Jung : 10.1002/jmri.21561
About this article
Cite this article
Simpson, R., Menza, M. & Hennig, J. 4D tissue phase mapping: clinically viable acquisition protocol and new method of visualisation. J Cardiovasc Magn Reson 17 (Suppl 1), O46 (2015). https://doi.org/10.1186/1532-429X-17-S1-O46
- Circumferential Velocity
- Plane Resolution
- Peak Diastole
- Tissue Phase Mapping
- Peak Diastolic Velocity