- Poster presentation
- Open Access
Increased maximum gradient amplitude improves robustness of spin-echo cardiac diffusion-weighted MRI
© Aliotta et al; licensee BioMed Central Ltd. 2015
- Published: 3 February 2015
- Apparent Diffusion Coefficient
- Maximum Gradient
- Diffusion Weighting
- Bulk Motion
- Fast Heart Rate
Cardiac motion presents a major challenge in diffusion weighted MRI (DWI), often leading to large signal dropouts that necessitate repeated measurements (Pai, V.M., MRM 2011). While cardiac DWI is generally ECG gated to apply diffusion weighting during peak-systole or end-diastole, these intervals can be short and difficult to pinpoint, resulting in poor sequence reproducibility.
Recent improvements in gradient hardware provide larger maximum gradients than current systems (Gmax=80mT/m), which can substantially reduce the temporal footprint of diffusion preparation and make cardiac DWI more robust to bulk motion.
A left ventricular (LV) motion model simulated motion of the healthy heart with 30-70ms quiescent intervals (tQ). Monopolar encoded SE-DWI (b=500 s/mm2, 3 directions) was simulated using: Gmax=40 and 80mT/m with diffusion gradients centered at mid-quiescence and with a range temporal offsets (ΔT=±20ms). Complex Gaussian noise was added such that SNR=50 for b=0 images. Bulk motion induced error was measured by the bias in apparent diffusion coefficient (ADC) recovery from the programmed value (ADC=1x10-3 mm2/s). Sequences that recovered ADC with bias<10% for ΔT=±10ms were deemed robust to motion.
Three healthy volunteers were scanned in a 3.0 T Siemens Prisma (Gmax=80mT/m) scanner using breath hold cardiac DWI (12 directions, monopolar encoding, single-shot SE EPI readout). Five repetitions of each sequence were acquired: 1) G40-b300: Gmax=40mT/m, b=300 mm2/s, TE=44ms; 2) G40-b100: Gmax=40mT/m, b=100mm2/s, TE=36ms; and 3) G80-b300: Gmax=80mT/m, b=300 mm2/s, TE=36ms. Imaging was timed to the diastolic quiescent interval, which was determined visually from CINE images.
Two observers evaluated the quality of all images as: "acceptable"-no significant signal dropouts in myocardium or "unacceptable"-significant signal dropouts, and directly compared G40-b300 to G80-b300 (fixed b-value).
Simulations show that G80 recovered ADC more accurately than G40 for all tQ and ΔT and was robust to motion for tQ≥30ms. This is likely due to the shorter diffusion preparation (G40 tprep=39ms, G80 tprep=28ms) and indicates that G80 will perform more consistently for short tQ (fast heart rates, systolic imaging) or changes in heart rhythm.
With fixed b-value=300mm2/s in vivo, G80 had consistently better image quality than G40. In agreement with simulation, this indicates that G80 improves the robustness of cardiac DWI for the same b-value. With fixed TE=36ms, G40-b100 was acceptable more frequently than G80-b300, but with insufficient diffusion weighting. Increased Gmax can thus improve diffusion sensitivity with less loss of robustness.
This research was supported by Siemens Medical Solutions and the Department of Radiological Sciences at UCLA.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.