Volume 17 Supplement 1
Initial experience for first pass cardiac perfusion with iterative reconstruction in patients
© Iacuzio et al; licensee BioMed Central Ltd. 2015
Published: 3 February 2015
MR stress perfusion is a non-invasive, reliable and safe test for ischemic heart disease . Recent publications reported sensitivity and specificity of 89% and 80% respectively . Image quality improvements seem to be essential for improving the predictive value of the method. This leads to the dilemma of finding a compromise between high spatial resolution and sufficient SNR. Lately iterative reconstruction demonstrated great promise in improving SNR . The aim of the current study was to compare cardiac perfusion in 24 patients reconstructed with product and a novel prototype iterative reconstruction.
24 patients (mean age 62 ±15) were examined on a 1.5T clinical MR scanner (MAGNETOM Aera, Siemens AG, Erlangen, Germany) using a saturation prepared TFL product sequence with the following parameters: TR/TE=331/1.4ms ; Flip angle = 12°; BW = 668Hz/Px; Voxel size=0.8×0.8×10.0 mm (inplane interpolated); FOV=290mm2; Matrix = 170x192; Slice thickness=10mm; Acceleration=2; Inversion time= 180ms; Phase oversampling=60%; Motion correction.
Pharmacological stress was applied using Adenosine (Adenoscan®, Sanofi-Synthelabo). Gadolinium-based contrast agent (Magnevist, Bayer Schering Pharma) was administered and MRI was performed over 4 slices in short-axis orientation in breath-hold.
Image reconstruction was run twice online at the scanner: 1) with the product image reconstruction and 2) a regularized SENSE-type iterative reconstruction  (60 iterations, regularization 0.008). Apparent signal-to-noise ratios (aSNR)  were calculated in multiple regions of interest in the left ventricle and myocardium. In addition, normalized semi-quantitative upslope curves were qualitatively compared for both methods.
Iterative image reconstruction shows great image quality improvements over the conventional reconstruction. In the future, SNR improvements can be invested in increasing the spatial resolution which might help to better avoid e.g. the dark rim artifact.
- Nandalur KR: J Am Coll Cardiol. 2007, 50: 1343-53. 10.1016/j.jacc.2007.06.030.View ArticlePubMedGoogle Scholar
- Hamon M: J Cardiovasc Magn Reson. 2010, 12: 29-10.1186/1532-429X-12-29.PubMed CentralView ArticlePubMedGoogle Scholar
- Adluru G: MRM. 2007, 57: 1027-36. 10.1002/mrm.21248.View ArticlePubMedGoogle Scholar
- Liu J: ISMRM. 2012, 178-Google Scholar
- Chung : IEEE CIS. 2004, 365-60.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.