- Oral presentation
- Open Access
An MRI-compatible defibrillator: initial testing in volunteers and swine
© Schmidt et al. 2016
- Published: 27 January 2016
- SSFP Imaging
- Power Outlet
- Defibrillation System
- Cardiac Array
- Monitoring Utility
Performing MRI-guided interventions, such as for the treatment of ventricular tachycardia, requires the ability to rapidly detect and treat acute cardiac conditions, should they occur in the MRI. We developed MRI-compatible 12-lead ECG systems, and demonstrated detection of acute ischemia during imaging [Tse, MRM. '13,Zhang, MRM. '15 ], providing a tool to non-invasively detect cardiac events in the MRI. It is presently necessary to remove patients from the MRI to treat cardiac events. This increases risk, consequently restricting MRI imaging, as well as MRI-guided intervention, to lower-risk patients and lower-risk procedures. Defibrillation in the bore could, therefore, accelerate the growth of diagnostic (such as exercise stress-perfusion and acute post-infarct studies) and therapeutic MRI studies. The study objective is to develop a defibrillation system that can safely be used on patients inside a 1.5 Tesla MRI. In addition, this system should not compromise MRI image quality.
A slight 2 and 1 dB reduction in image SNR was observed (Figure 1C) with the system running on A/C power, and running on battery, respectively, relative to the system OFF. The system OFF state displayed no discernable SNR penalty relative to removal of the system from the room. Image artifacts from the defibrillator pads extended only ~6 mm below the skin. Defibrillator-pad heating did not occur. The swine defibrillations were completed without any complications.
Defibrillation inside the MRI may be performed with a modified commercial defibrillator system, enabling the performance of higher-risk imaging and MRI-guided interventions.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.