- Poster presentation
- Open Access
Rotational motion of Left Ventricle by Harmonic Phase on radial tagging
© Golshani et al. 2016
- Published: 27 January 2016
- Displacement Field
- Rotational Motion
- Cardiac Mechanic
- Harmonic Phase
- Cardiac Frame
The rotational motion of the Left Ventricle (LV) is a valuable measure in assessment of cardiac mechanics and can be estimated best through radial tagging. For practical quantitative application of this newly developed sequence, corresponding analyzing tools need to be developed. In this study, we presented a processing method based on peak-combination Harmonic Phase (HARP) approach to obtain displacement fields in polar coordinate.
Short-axis human heart images with radial tagging pattern were acquired using a radial k-space sampling scheme. The data consisted of 19 cardiac frames obtained in a single breath-hold of 12 seconds duration. We exploited a Polar Fourier Transform for the reconstruction and processing of polar images. The method is based on two angularly Fourier Transforms separated by a Hankel-transform in the radial direction. Figure 1 illustrates the k-space of the first cardiac frame. After performing the first 1D-FFT of the radial tagged images, the tagging energy is concentrated in two highlighted lines which are the reminiscent of spectral peaks in the 1-1 SPAMM. Inspired by HARP, only few Fourier-coefficients around one of the highlighted lines can be isolated to form a 2D complex image of the tagging fundamental frequency. This was repeated for the conjugate highlighted line and therefore, the conjugate harmonic phase was reconstructed. Based on the peak-combination HARP, the two images were subtracted to improve the accuracy of the resulting phase map. The harmonic phase images were then unwrapped and used for calculation of the displacement field.
HARP method for 1-1 SPAMM has a dual version in the Radial tagging in which the spectral peaks are replaced with the highlighted lines after the first angular FFT. Therefore, the same advantages one may expect from HARP in conventional tagging is transferred to radial tagging, if we use PFT in image reconstruction. It should be further noted that the rotational motion can be directly measured in the polar rather than Cartesian coordinates.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.