- Workshop presentation
- Open Access
A T1 and ECV phantom for global T1 mapping quality assurance: The T1 mapping and ECV standardisation in CMR (T1MES) program
Journal of Cardiovascular Magnetic Resonance volume 18, Article number: W14 (2016)
Myocardial T1 and extracellular volume (ECV) estimates have applications in a range of myocardial diseases. Factors responsible for systematic inaccuracies in T1 mapping are beginning to be known1-4 but little is known about its delivery at ‘health-care system' scale and there is no global quality assurance (QA) system. Agarose phantoms are common in MRI and nickel ions preferred for lower temperature sensitivity5. This program aims to
1 Create a partnership to design 1.5/3T phantoms for any manufacturer/sequence reflecting myocardial/blood T1 pre/post-contrast
2 Test and mass produce phantoms to regulatory standards
3 Distribute globally
4 Analyse serial scans to understand T1 mapping at scale
5 Publish recipes
6 Explore delivery of a ‘T1/ECV Standard' via local calibration
We report results of steps 1-3.
A design collaboration was created (clinicians/physicists/regulatory bodies/SME). After identifying critical design factors (Fig 1A) and discarding models with excessive B 0 /B 1 distortion, the layout in Fig 1B was adopted. 9 tubes with differently doped agarose were embedded in a gel matrix and high-density polyethylene (HDPE) macrobeads added for B 1 homogeneity. Tube diameter >20 mm was needed for regions of interest to exclude Gibbs artifacts. B0/B1 homogeneity was mapped to evaluate distortion. We hypothesised that dilution of dielectric permittivity by HDPE beads would reduce B 1 inhomogeneity. This design was compared to ones using sodium chloride (NaCl) for increased conductivity, sucrose for reduced permittivity or poly methyl-methacrylate (PMMA) microbeads. Tubes with T1 = 250-1900 ms and T2 = 45-250 ms were reproducibly manufactured and separate ranges adopted for 1.5/3T (Fig 1C). 10 Prototypes were fabricated (5 each for 1.5/3T) for gold standard measurements: T1 by inversion-recovery spin echo(IRSE, 8 inversion times, 25>3200 ms); T2 by SE(8 echo times, 10>640 ms). Prototypes were then distributed to 9 experienced/regulatory centres for further testing.
T1 maps were free from off-resonance artifacts (Fig 1D). The bottle geometry, coaxial with z and imaged transversely, showed < ± 0.3 ppm B 0 uniformity (Fig 2A). HDPE beads flattened the B 1 field at 3T (Fig 2B) especially compared to NaCl, sucrose and PMMA beads. T1 increased with temperature (0.19-1.54% change/°C) while T2 decreased(-0.93-1.45% change/°C). Comparison of gold standard values (Fig 2C,D) between prototypes confirmed reproducible manufacturing(coefficients of variation T1 0.97/1.35%, T2 1.25/2.73% for 1.5T/3T). Recipes were submitted for regulatory approval and manufacture will be complete by Sep'15.
We created a collaboration to develop CE/FDA-approved phantoms for QA of T1 and ECV protocols. 70 revised phantoms with a multi-vendor user manual are now being distributed to centres worldwide for a 1-year academic exploration of T1 mapping sequences, platform performance and stability.
Rights and permissions
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
About this article
Cite this article
Captur, G., Gatehouse, P., Kellman, P. et al. A T1 and ECV phantom for global T1 mapping quality assurance: The T1 mapping and ECV standardisation in CMR (T1MES) program. J Cardiovasc Magn Reson 18 (Suppl 1), W14 (2016). https://doi.org/10.1186/1532-429X-18-S1-W14
- Dielectric Permittivity
- Gold Standard Measurement
- Local Calibration
- Lower Temperature Sensitivity5
- PMMA Bead