- Meeting abstract
- Open Access
1140 A new method for measuring through-plane strain using slice following with inversion recovery (SFIR)
Journal of Cardiovascular Magnetic Resonance volume 10, Article number: A265 (2008)
MRI is a unique modality in imaging the motion of moving tissues including that of the heart. Tagging and phase contrast techniques have been used to noninvasively measure the myocardium motion. In this work, a new technique is proposed to measure the through-plane strain based on the changes in slice profile that occur due to the tissue deformation.
Slice following imaging
In slice following imaging , a thin slice is excited by slice selective inversion while a thicker slice containing the thinner one is imaged. If a deformation of tissue occurs to the excited slice, the slice thickness changes accordingly, accompanied by changes of the slice profile in the frequency domain. In case of a rectangular slice profile, changes in the slice thickness will result in changing the width of the sinc profile in the frequency domain (Fig. 1).
In order to capture the tissue deformation, complete knowledge about the slice profile in frequency domain is required. For a rectangular slice profile (s(t) = rect(t/B) and S(ωz) = Bsinc(Bωz), where B is the slice thickness), two samples at two different z-frequencies (tunings) are sufficient to fully describe the sinc function. This can be done by acquiring two images at two different points (tunings) in the kz direction by applying a suitable gradient moment in the z-direction during the refocusing lobe.
For a voxel with initial slice thickness B0, two intensities are acquired at two different tuning frequencies, I0 at ω0, and I1 at ω1. The new slice thickness at time t (Bt) can be estimated as Bt = sinc-1(I1/I0)/ω1. Then the local strain (εt) can be estimated as εt = (Bt/B0)-1.
The SFIR technique was implemented and tested on a 3 T Philips scanner. A schematic of the pulse sequence is shown in Fig. 2. Images were acquired with FOV = 300*300 mm, slice thickness = 8 mm, spatial resolution = 2 mm*2 mm, temporal resolution = 20 ms and spiral acquisition of 10 spirals with spiral window of 10 ms.
To validate the proposed technique, a phantom study was conducted. A cubic phantom of side length 10 cm was compressed with a balloon blown with air and purged in constant time intervals, in the normal direction to the image plane. R-wave is simulated exactly before the balloon start to be blown. The balloon diameter was 2 cm during its maximum blowing. During the experiment, the balloon was positioned in one corner of the phantom.
In vivo studies
Two normal volunteers were consented and scanned using the proposed sequence. For all experiments, two slice following image sets were acquired at two different tunings (0,0.6). The two sets were used to estimate the strain in each voxel using the proposed algorithm.
Results and discussion
Fig. 3 shows the phantom function images at different time frames of compression. Fig. 4 shows the calculated strain profile through time for two groups of voxels, one inside the compression region confronting the balloon and the other is far away from the balloon. The graph shows that while the second group has a nearly zero strain values, the first group exhibits a strain profile which confirms with the inflation profile of the balloon. Fig. 5 shows short-axis functional images calculated using the proposed algorithm. As expected, the myocardium shows shortening deformation through time while the cavity blood and other tissue shows nearly zero strain.
A new technique is proposed for estimating the through plane tissue deformation from the normal slice following images. No special patterns (tags) are needed, so a high spatial resolution can be obtained.
Stuber : ISMRM Proc. 418-
This work was supported in part by grants from the national institute of heart, lung and blood R01HL072704 and R01HL61912, and Donald W. Reynolds Foundation grant.
Rights and permissions
Open Access This article is published under license to BioMed Central Ltd. This is an Open Access article is distributed under the terms of the Creative Commons Attribution 2.0 International License (https://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Basha, T., Abd-Elmoniem, K.Z. & Osman, N.F. 1140 A new method for measuring through-plane strain using slice following with inversion recovery (SFIR). J Cardiovasc Magn Reson 10 (Suppl 1), A265 (2008). https://doi.org/10.1186/1532-429X-10-S1-A265
- Slice Thickness
- Tissue Deformation
- Sinc Function
- Strain Profile
- Contrast Technique