- Technical notes
- Open Access
Rapid assessment of myocardial infarct size in rodents using multi-slice inversion recovery late gadolinium enhancement CMR at 9.4T
© Price et al; licensee BioMed Central Ltd. 2011
- Received: 12 April 2011
- Accepted: 5 September 2011
- Published: 5 September 2011
Myocardial infarction (MI) can be readily assessed using late gadolinium enhancement (LGE) cardiovascular magnetic resonance (CMR). Inversion recovery (IR) sequences provide the highest contrast between enhanced infarct areas and healthy myocardium. Applying such methods to small animals is challenging due to rapid respiratory and cardiac rates relative to T1 relaxation.
Here we present a fast and robust protocol for assessing LGE in small animals using a multi-slice IR gradient echo sequence for efficient assessment of LGE. An additional Look-Locker sequence was used to assess the optimum inversion point on an individual basis and to determine most appropriate gating points for both rat and mouse. The technique was applied to two preclinical scenarios: i) an acute (2 hour) reperfused model of MI in rats and ii) mice 2 days following non-reperfused MI.
LGE images from all animals revealed clear areas of enhancement allowing for easy volume segmentation. Typical inversion times required to null healthy myocardium in rats were between 300-450 ms equivalent to 2-3 R-waves and ~330 ms in mice, typically 3 R-waves following inversion. Data from rats was also validated against triphenyltetrazolium chloride staining and revealed close agreement for infarct size.
The LGE protocol presented provides a reliable method for acquiring images of high contrast and quality without excessive scan times, enabling higher throughput in experimental studies requiring reliable assessment of MI.
- Cardiovascular Magnetic Resonance
- Left Anterior Descend
- Late Gadolinium Enhancement
- Infarct Volume
- Late Gadolinium Enhancement Image
Cardiovascular magnetic resonance (CMR) has emerged as a valuable tool in the field of small animal cardiovascular research [1–6]. However, many of the CMR techniques that are now used as standard in the clinical setting remain a significant challenge to implement in small animals especially at high field , where relatively few optimised protocols are available as standard on experimental systems. One such technique, regarded as the gold standard for assessing myocardial infarction (MI) in patients , is late gadolinium enhancement (LGE) CMR. The extent of the LGE region is typically assessed using an inversion recovery (IR) breath-hold sequence, which can be readily applied to larger animals used in cardiovascular research [8, 9]. When it comes to assessing LGE in small animals, such as rat and mouse, these methods become difficult to implement due to the rapid heart and respiratory rates. This is confounded by the fact most CMR of small animals is typically done at high field where the inherent T1 relaxation of myocardium is substantially longer than at clinical field strengths.
For these reasons the majority of reported implementations of LGE CMR in small animals have tended to use more conventional T1-weighted sequences, namely short repetition time (TR), high flip angle, cine gradient echo [10, 11]. Nevertheless, there have been a limited number of studies recently that have used IR sequences to assess LGE at high field in both mice [12–14] and rats . In order to achieve the maximum contrast between infarcted myocardium (where extracellular contrast agents such as Gd-DTPA will accumulate to enhance MR signal) and the normal healthy myocardium, it is necessary to use an IR technique with the appropriate inversion time (TI). Finding the optimum TI, in order to sufficiently null the signal from normal myocardium, can be difficult to assess at high field, as TI is typically several times longer than the R-R wave interval, even after relatively high doses of contrast agent. Thus achieving the best contrast whilst also maintaining high quality images that have optimum gating, relative to both the respiratory and cardiac cycles, is key to accurate assessment of MI using LGE CMR in small animals.
In recent reported implementations of IR sequences to assess LGE in mice at 9.4T [12, 13], 7T  and rats at 4.7T  the approach has been to use a segmented spoiled gradient echo (GRE) or FLASH sequence, where a number of echoes or lines (with typical echo-train lengths of 4-16 being reported) are acquired following each inversion pulse. However, the alternative approach for 2D imaging is to use the relatively small gating window available to acquire a multi-slice GRE acquisition.
Here we present a robust yet fast protocol, and gating strategy, for assessing MI in rodents based on a combination of Look-Locker (LL) [17–19] and multi-slice IR-GRE sequences. We have employed a LL technique to evaluate the optimum null point to provide good tissue contrast and multi-slice IR-GRE to produce high quality images of enhanced regions following administration of Gd-DTPA. The area of infarction measured in vivo by CMR is compared to ex vivo triphenyltetrazolium chloride (TTC) staining - an established technique to assess viable myocardium - to evaluate the correlation between the two methods. The protocol was applied to two different practices used by cardiovascular MI researchers i) rats in the acute phase of a reperfused MI and ii) mice 2 days after permanent ischemic MI in order to assess a broad range of methodologies applied to small animal models of myocardial infarction.
Rat model of acute myocardial infarction
All animal procedures were approved by the University College London Ethics Committee and the Home Office (London, UK), and conducted under the Animal (Scientific Procedures) Act 1986. Wistar rats (9 males, aged 7 - 8 weeks) were anaesthetised with sodium thiopentone (i.p., 120 mg/kg, Intraval, U.K.) and underwent surgery to induce myocardial infarction. Briefly, an endotracheal tube was inserted via a tracheotomy for mechanical ventilation, after which a small thoracotomy was performed along the 4th lateral intercostal space to expose the heart. The left anterior descending (LAD) coronary artery was occluded close to its origin with a snare occluder for 30 minutes, which was then released for reperfusion. Following visual verification of successful reperfusion (i.e. disappearance of pallor and return of blood flow), the chest was closed, and spontaneous breathing was re-established. Animals were placed in the MR scanner and post-MI images were acquired at 2 hours from reperfusion.
Mouse model of myocardial infarction
B6Sv129 mice (6 males, 9-11 weeks old) were anesthetised by an intraperitoneal injection of ketamine (75 mg/kg) and medetomidine hydrochloride (1 mg/kg). The trachea was exposed through a mid-line incision and intubated through the oral pharynx for artificial respiration. Through a left anterior thoracotomy, the heart was exposed with aid of a mini-retractor. Ligation of the LAD artery was performed at a level 1 mm below the edge of the left atrium. Successful LAD occlusion was confirmed by a sudden discoloration of the anterior wall of the left ventricle. After the coronary intervention, the chest wall and skin were closed, animals were given atipamezole hydrochloride (5 mg/kg, i.p.) as a sedation reversal agent and buprenorphine (0.1 mg/kg, intramuscular) as an analgesic. After spontaneous respiration had resumed, the animal was weaned from the respirator and allowed to recover on a heating pad. Mice were imaged at 2 days post-surgery.
Anaesthetised animals were placed onto an animal cradle with a water-heating system to maintain body temperature. Oxygen and isoflurane (1-2%) was provided via a nose cone and a scavenging system was used to remove anaesthetic gases. A neonatal apnoea sensor was placed on the abdomen for respiratory gating and electrocardiogram (ECG) was obtained by needle electrodes inserted subcutaneously. Cardio-respiratory monitoring and gating were performed using an MR-compatible system (SA Instruments, NY).
Imaging was performed using a 9.4T VNMRS horizontal bore scanner (Varian Inc. Palo Alto, CA) with a shielded gradient system (400 mT/m). For rats the RF excitation was achieved using a 72 mm de-tuneable volume coil and signal received using a 4-element array cardiac coil (Rapid Biomedical GmbH, Germany). Mice were imaged using a 33 mm quadrature volume coil (Rapid Biomedical GmbH, Germany) and 1000 mT/m gradient insert. Cardiac images of conventional orientations (2-chamber, 4-chamber and short-axis) were obtained as described below. Initially, axial and longitudinal scout images were acquired to plan the 2-chamber long-axis view. A 4-chamber long-axis view was then acquired perpendicular to the 2-chamber orientation, after which a series of short-axis images was obtained perpendicular to the LV long-axes from the 2- and 4-chamber images - in a similar fashion described in . In order to cover the whole LV from apex to base, typically 15 short-axis image slices were acquired for rats and 10 for mice.
A double (respiratory + cardiac) gated spoiled gradient echo sequence was used to acquire cine cardiac images, and used to calculate left ventricular mass (LVM). Imaging parameters for rats were as follows; echo time (TE) = 1.7 ms, repetition time (TR) = ~7.5 ms, flip angle = 15°, slice thickness = 1 mm, field of view (FOV) = 40 × 40 mm, matrix size = 192 × 192, number of signal averages = 1. Twenty time-frames were recorded for every cardiac cycle, and phase encoding was incremented linearly following each cycle. A single short-axis slice was obtained in approximately 45 seconds, leading to a total scan time for cine data of around 10 to 15 minutes. In mice the following parameters were adjusted: TE = 1.1 ms, TR = 4-5 ms, FOV = 25.6 × 25.6 mm, matrix size = 128 × 128, with 10 slices acquired taking ~5 minutes.
For delayed contrast enhancement 0.6 mmol/kg Gd-DTPA (Magnevist, Schering AG, Germany) was injected, via an intra-venous route for rats (tail vein), following initial baseline scanning. In mice contrast agent was administered via an intra-peritoneal route, due to the difficulty and unreliability in cannulating the mouse tail vein.
Look-Locker (LL) TI evaluation
The LL scan was followed by a high SNR multi-slice IR-GRE sequence with single TI point and flip angle of 90°; maintaining other parameters was important to achieve a similar effective nulling of healthy myocardium compared to LL acquisition. TI is selected from the initial multiple frames of the LL scan to be the closest gated TI point which most effectively nulls healthy myocardium and consequently provides the best contrast whilst also maintaining image quality from the optimum gating strategy. The repetition time between each slice pulse (TRs) was 3.6 ms, with a sequential acquisition order starting at the apex; typically 9 slices were needed to cover the area of infarction in the rat. Mice were also imaged using the same LGE protocol as described above except again for the following parameters; TE = 1.1 ms, TRs = 3.1 ms, FOV = 25.6 ×25.6 mm, matrix size = 128 × 128, and 7 slices typically needed to cover area of infarction.
Ex vivo infarct measurement by TTC staining
After all imaging was completed, rat hearts were extracted for infarct quantification by triphenyltetrazolium chloride (TTC) staining and planimetry. Hearts were extracted and frozen at -20°C for 2 hours and cut into 2 mm-thick slices. To demarcate infarcted myocardium from viable myocardium, heart slices were put into 1% TTC solution at 37°C for 15 minutes. Heart slices were then fixed in 4% PFA overnight, and placed between two Perspex sheets to be imaged using a colour flat-bed scanner at 600 dpi resolution. Images were subsequently analysed using a semi-automatic, colour-based segmentation technique in ImageJ software (NIH).
Obtaining meaningful estimates of background noise in images acquired using receive array coils is difficult to achieve without having acquired additional noise reference scans or repeated acquisitions . For this reason we decided not to quote SNR/CNR values and instead opted for the contrast resolution to give a better comparative estimate of contrast between the relevant tissues. Contrast resolution is defined as CR = (SA-SB)/(SA+SB). Signal intensities in the LV blood pool, healthy and infarcted myocardium were measured from 3 consecutive slices containing both infarcted and healthy myocardium using ImageJ. For each slice, an elliptical region of interest (ROI) was selected to include the largest continuous area of enhanced myocardium possible; a second ROI of the same size was then placed diagonally opposite for measurement of remote healthy myocardium, typically within the septum.
Left ventricular mass was assessed from the full stack of short-axis cine images using freely available software Segment http://segment.heiberg.se/ by means of the semi-automated LV tracking tool and valve plane tracked on the 2 and 4-chamber views as described in . Infarct volume was calculated from the late gadolinium images in Segment using the auto delineate (weighted method) function which is constrained by the endo and epi-cardial borders defined in the prior segmentation of LVM. Infarct volume is expressed as percentage of left ventricular mass for TTC comparison. Infarct volumes by ex vivo TTC and in vivo CMR were blinded and anonymised prior to analysis. Bland-Altman and linear regression analysis was applied to the LGE-CMR and TTC staining data. All statistical analysis was performed in GraphPad Prism (v4.01 for Windows).
The LGE slices are acquired sequentially up from the apex, with a TRs = 3.1 ms and 7 slices typically needed, the mouse LGE region can be covered within the time frame of 22 ms frame the R wave trigger. Therefore the majority of the LGE region is acquired close enough to the end diastolic cine frame, with just the upper slices starting to edge into systole. Furthermore, as it is the endo and epi-cardial borders surrounding the LGE regions that are being utilised, these are inevitably akinetic and adjusting these boundaries or taking later cine frame segmentations to fit the later LGE slices is rarely necessary.
Contrast resolution between infarcted and remote myocardium was measured in LGE images as 82.6 ± 2.6% in rats and 73.9 ± 13.1% in mice; sufficient for the purpose of infarct volume segmentation and measurement. Contrast resolution between infarct and blood pool was 27.1 ± 3.8% in rats and -1.4 ± 7.8% in mice, which illustrates why LV segmentation from cine data was particularly useful in mice.
CMR provides a remarkably clear and non-invasive way to assess the extent of myocardial infarction with the use of extracellular contrast agents. It is undoubtedly the gold standard way to assess the extent and location of infarcted tissue in patients, with a typical LGE imaging protocol based around an inversion recovery sequence. However, when it comes to imaging small animals at high field strengths there are varying opinions on the best methods to use. In all the recent implementations of IR sequences to assess LGE in mice at 9.4T [12, 13], 7T  and rats at 4.7T  the approach has been to use a segmented FLASH sequence, where a number of echoes/lines (ETL = 4-16) are acquired following each inversion pulse, with some variations in gating strategy. In the work described by Chapon et al. , LGE was assessed in a permanent occluded mouse model of MI using segmented IR-FLASH. ECG gating was applied to the inversion pulse and data acquired at a set TI, and subsequently repeated for many TI values to evaluate the best value for optimum contrast. Bohl et al.  used a segmented 3D IR-GRE sequence to achieve high resolution scanning of MI in mice at 9.4T. This method is likely to be the best approach to use when high resolution images are necessary, particularly as achieving high through plane resolution would generally limit the use of conventional 2D acquisition approaches. The later study also employed IR-(snapshot)FLASH to map T1 of infarct and normal myocardium temporally following both intravenous and intraperitoneal administration of Gd-DTPA(-BMA).
In a study by Thomas et al.  IR sequences were compared directly to T 1 weighted cine in rats at 4.7T and concluded that T1-weighted cine was preferred due to the speed advantage of acquiring both global cardiac function and LGE data in one. This study utilised a necrotic-specific contrast agent removing the problem of dynamic changes in contrast that are exhibited by standard Gd-chelated agents. In a similar study by Protti et al.  a Gd-DTPA bolus was given i.p. in mice at 7T and cine-FLASH images acquired in the 20-30 minute window compared to IR-FLASH performed in the 30-50 minute period following injection. Cine-FLASH was also preferred here to a segmented IR for being less user dependent while providing similar accuracy. A likewise comparison of T 1 weighted cine and IR-FLASH at 9.4T, with an additional comparison to a multi-slice IR method as proposed in this study which should offer higher SNR efficiency would also be desirable. The main disadvantage with opting for a single cine scan to assess both LGE and functional parameters is that signal from healthy myocardium needs to be further saturated in order to boost contrast relative to infarcted tissue, which could compromise functional assessment particularly when attempting to delineate the epi-cardial border for measurement of LV mass.
In this study we present an IR-based method for assessing LGE where the acquisition of multiple slices, rather than multiple echoes or phase-encode lines, also yields encouraging results. Although the amount of data acquired at each time point is similar in both methods, acquiring all slices rather than multiple k-space lines allows for the use of higher flip angles and thus higher SNR efficiency can be achieved over the segmented FLASH technique. Acquiring one line of k-space from each slice per inversion allows for the full recovered longitudinal magnetisation to be sampled, whereas segmented FLASH requires lower flip angles to limit the saturation effects through the echo train.
In the presented method here both LL and IR-GRE scans are performed without allowing for complete T1 recovery, where the overall TR is determined by the respiratory rate. The effect of not allowing for full recovery of longitudinal magnetisation is to reduce the effective null points and signal intensity, which in turn reduces CNR slightly . However, it has been shown that sufficient SNR and contrast can be easily achieved with only single averaged scans employing TRir ~ 1s and subsequently TAs ~ 3 minutes.
In this implementation a double gated inversion is used followed by a delay of an integer number of R-wave intervals to ensure all data contributing to an image is acquired at exactly the same point in the cardiac cycle. This approach means that TI is set to a multiple of the R-R wave interval possibly leading to a small deviation from the maximum obtainable contrast, although adding a trigger delay to move to end-systole for example could improve this if desired. The benefits of a more precise and consistent gated acquisition is evident from the clarity of the images, whilst still maintaining sufficiently high levels of contrast between infarcted and viable myocardium. Another advantage of using a multi-slice over segmented FLASH is that each image consists of data acquired from a narrower window in the cardiac cycle. As data contributing to each slice is acquired from a 1 × TR window rather than ETLxTR cardiac motion is reduced.
The speed of acquisition, in addition to the time after injection, is an important factor for LGE using Gd-DTPA, as previous studies have reported relatively rapid pharmacokinetic changes in contrast at the periphery of infarcts [25, 26]. The delay of ~10 minutes following injection to LGE imaging was primarily chosen to allow for development of enhancement in the infarcted tissue, but also it has been shown that imaging too soon after administration can lead to an over-estimation of infarct size . However, this delay could be further increased to improve contrast between infarct and remote myocardium when contrast is being administered via the i.p. route. Bohls et al.  recently demonstrated maximum T 1 contrast is achieved between 13.5-22 minutes after i.v. injection and 30-40 mins after i.p. injection in mice using a similar concentration and agent (0.5 mmol/Kg Gd-DTPA-BMA). Therefore, although our time-point for rats i.v. is likely close to optimum for maximum contrast, allowing a longer delay for mice after i.p. injections should yield further increases in contrast if scan time constraints allowed. A further potential benefit could be that the contrast between infarct and the blood pool may increase sufficiently to potentially allow for segmentation of infarct area without using the boundaries found from the LV cine segmentation, as was the case for the rat LGE images in this study. However, the recovery of blood T 1 , which would improve contrast between blood and infarcted tissue, is confounded by the fact the blood pool signal level is also influenced by inflow enhancement. Therefore, it is hard to suppress the blood signal without also introducing more complicated preparation pulses and timings.
A possible limitation to the study protocol applied here to mice is that it could be argued that resolution should be increased relative to the size of the animal. The biggest benefit would likely come from an increase in the through plane resolution, by decreasing slice thickness and increasing the number of slices. Reducing slice thickness to 0.5 mm for example would be achievable with the gradient capability of the system mentioned, but the resultant slice pack would extend to approximately 1/3 rd of the cardiac cycle, unless the acquisition were split into two interleaved acquisitions. Subsequently this would double scan time and reduce SNR by 50%, however, with this protocol there should still be sufficient levels of signal and contrast to segment LGE area without needing additional signal averages.
The additional step presented in this protocol to optimise the best gated TI point on an individual animal basis using a Look-Locker scan, is useful to confirm arrival and degree of local tissue contrast. However, it is not essential as it has been shown that with the given concentrations of contrast agent used here and at the LGE imaging window following injection we opted for the TI is consistently in the range 300-450 ms, leading to a R-R gate of 2-3 in rats (i.v.) and predominately 3 (occasionally 4) R-waves in mice (i.p.). Both LL and LGE scans are readily acquired in ~3 minutes each thus allowing for a rapid assessment of LGE whilst minimal contrast agent washout occurs. The functional cine assessment, typically desired for most CMR studies of MI, could easily be performed in the time between injection of contrast and LGE imaging to further improve throughput with an additional SNR boost to the cine data that could aid segmentation of LVM. Therefore, a complete functional and LGE assessment could be achieved in around 10-15 minutes from injection of contrast with the 200 μm in-plane resolution used here.
This study presents a fast and robust protocol for the reliable assessment of myocardial infarction in small animals, using a multi-slice IR sequence at 9.4T. By employing the use of a Look-Locker sequence the optimum inversion time can be checked on an individual animal basis, while the approach used for gating and acquisition provides LGE images of ample contrast and image quality but with scan times of less than 5 minutes. The method has been validated against tissue viability staining in rats following acute reperfused MI, and also shown to work well in mice following non-reperfused MI post 48 hours.
We believe this approach to assessing LGE in small animals offers an accurate and reproducible method to determine infarct size without the need for lengthy scans.
This work was supported by the British Heart Foundation, the Biotechnology and Biological Sciences Research Council and the Engineering and Physical Sciences Research Council.
- Epstein FH: MR in mouse models of cardiac disease. Nmr in Biomedicine. 2007, 20: 238-255. 10.1002/nbm.1152.View ArticlePubMedGoogle Scholar
- Hoit BD: New approaches to phenotypic analysis in adult mice. Journal of Molecular and Cellular Cardiology. 2001, 33: 27-35. 10.1006/jmcc.2000.1294.View ArticlePubMedGoogle Scholar
- Price AN, Cheung KK, Cleary JO, Campbell AE, Riegler J, Lythgoe MF: Cardiovascular magnetic resonance imaging in experimental models. Open Cardiovasc Med J. 2010, 4: 278-292. 10.2174/1874192401004010278.PubMed CentralView ArticlePubMedGoogle Scholar
- Schneider JE, Cassidy PJ, Lygate C, Tyler DJ, Wiesmann F, Grieve SM, et al: Fast, high-resolution in vivo cine magnetic resonance imaging in normal and failing mouse hearts on a vertical 11.7 T system. Journal of Magnetic Resonance Imaging. 2003, 18: 691-701. 10.1002/jmri.10411.View ArticlePubMedGoogle Scholar
- Schneider JE, Lanz T, Barnes H, Medway D, Stork LA, Lygate CA, et al: Ultra-fast and accurate assessment of cardiac function in rats using accelerated MRI at 9.4 Tesla. Magnetic Resonance in Medicine. 2008, 59: 636-641. 10.1002/mrm.21491.View ArticlePubMedGoogle Scholar
- Vallee JP, Ivancevic MK, Nguyen D, Morel DR, Jaconi M: Current status of cardiac MRI in small animals. Magnetic Resonance Materials in Physics Biology and Medicine. 2004, 17: 149-156. 10.1007/s10334-004-0066-4.View ArticleGoogle Scholar
- Kim RJ, Wu E, Rafael A, Chen EL, Parker MA, Simonetti O, et al: The use of contrast-enhanced magnetic resonance imaging to identify reversible myocardial dysfunction. N Engl J Med. 2000, 343: 1445-1453. 10.1056/NEJM200011163432003.View ArticlePubMedGoogle Scholar
- Kim RJ, Fieno DS, Parrish TB, Harris K, Chen EL, Simonetti O, et al: Relationship of MRI delayed contrast enhancement to irreversible injury, infarct age, and contractile function. Circulation. 1999, 100: 1992-2002.View ArticlePubMedGoogle Scholar
- Simonetti OP, Kim RJ, Fieno DS, Hillenbrand HB, Wu E, Bundy JM, et al: An improved MR imaging technique for the visualization of myocardial infarction. Radiology. 2001, 218: 215-223.View ArticlePubMedGoogle Scholar
- Yang Z, Berr SS, Gilson WD, Toufektsian MC, French BA: Simultaneous evaluation of infarct size and cardiac function in intact mice by contrast-enhanced cardiac magnetic resonance imaging reveals contractile dysfunction in noninfarcted regions early after myocardial infarction. Circulation. 2004, 109: 1161-1167. 10.1161/01.CIR.0000118495.88442.32.View ArticlePubMedGoogle Scholar
- Epstein FH, Yang ZQ, Gilson WD, Berr SS, Kramer CM, French BA: MR tagging early after myocardial infarction in mice demonstrates contractile dysfunction in adjacent and remote regions. Magnetic Resonance in Medicine. 2002, 48: 399-403. 10.1002/mrm.10210.View ArticlePubMedGoogle Scholar
- Bohl S, Lygate CA, Barnes H, Medway D, Stork LA, Schulz-Menger J, et al: Advanced methods for quantification of infarct size in mice using three-dimensional high-field late gadolinium enhancement MRI. Am J Physiol Heart Circ Physiol. 2009, 296: H1200-H1208. 10.1152/ajpheart.01294.2008.PubMed CentralView ArticlePubMedGoogle Scholar
- Chapon C, Herlihy AH, Bhakoo KK: Assessment of myocardial infarction in mice by late gadolinium enhancement MR imaging using an inversion recovery pulse sequence at 9.4T. J Cardiovasc Magn Reson. 2008, 10: 6-10.1186/1532-429X-10-6.PubMed CentralView ArticlePubMedGoogle Scholar
- Ojha N, Roy S, Radtke J, Simonetti O, Gnyawali S, Zweier JL, et al: Characterization of the structural and functional changes in the myocardium following focal ischemia-reperfusion injury. Am J Physiol Heart Circ Physiol. 2008, 294: H2435-H2443. 10.1152/ajpheart.01190.2007.View ArticlePubMedGoogle Scholar
- Thomas D, Dumont C, Pickup S, Misselwitz B, Zhou R, Horowitz J, et al: T1-weighted cine FLASH is superior to IR imaging of post-infarction myocardial viability at 4.7T. J Cardiovasc Magn Reson. 2006, 8: 345-352. 10.1080/10976640500451986.PubMed CentralView ArticlePubMedGoogle Scholar
- Protti A, Sirker A, Shah AM, Botnar R: Late gadolinium enhancement of acute myocardial infarction in mice at 7T: cine-FLASH versus inversion recovery. J Magn Reson Imaging. 2010, 32: 878-886. 10.1002/jmri.22325.View ArticlePubMedGoogle Scholar
- Graumann R, Deimling M, Heilmann T, Oppelt A: A new method for fast and precise T1 determination. Proceedings of the Society of Magnetic Resonance in Medicine. 1986, 1986: 922-923.Google Scholar
- Look DC, Locker DR: Time saving in measurement of NMR and EPR relaxation times. Review of Scientific Instruments. 1970, 41: 250-251. 10.1063/1.1684482.View ArticleGoogle Scholar
- Brix G, Schad LR, Deimling M, Lorenz WJ: Fast and Precise T1 Imaging Using A Tomrop Sequence. Magnetic Resonance Imaging. 1990, 8: 351-356. 10.1016/0730-725X(90)90041-Y.View ArticlePubMedGoogle Scholar
- Schneider JE, Wiesmann F, Lygate CA, Neubauer S: How to perform an accurate assessment of cardiac function in mice using high-resolution magnetic resonance imaging. Journal of Cardiovascular Magnetic Resonance. 2006, 8: 693-701. 10.1080/10976640600723664.View ArticlePubMedGoogle Scholar
- Kellman P, McVeigh ER: Image reconstruction in SNR units: a general method for SNR measurement. Magn Reson Med. 2005, 54: 1439-1447. 10.1002/mrm.20713.PubMed CentralView ArticlePubMedGoogle Scholar
- Heiberg E, Wigström L, Carlsson M, Bolger AF, Karlsson M: Time Resolved Three-dimensional Automated Segmentation of the Left Ventricle. Proceedings of IEEE Computers in Cardiology. 2005, 32: 599-602.Google Scholar
- Riegler J, Cheung KK, Man YF, Cleary JO, Price AN, Lythgoe MF: Comparison of segmentation methods for MRI measurement of cardiac function in rats. J Magn Reson Imaging. 2010, 32: 869-877. 10.1002/jmri.22305.View ArticlePubMedGoogle Scholar
- Jivan A, Horsfield MA, Moody AR, Cherryman GR: Dynamic T-1 measurement using snapshot-FLASH MRI. Journal of Magnetic Resonance. 1997, 127: 65-72. 10.1006/jmre.1997.1177.View ArticlePubMedGoogle Scholar
- Oshinski JN, Yang ZQ, Jones JR, Mata JF, French BA: Imaging time after Gd-DTPA injection is critical in using delayed enhancement to determine infarct size accurately with magnetic resonance imaging. Circulation. 2001, 104: 2838-2842. 10.1161/hc4801.100351.View ArticlePubMedGoogle Scholar
- Judd RM, Kim RJ: Imaging time after Gd-DTPA injection is critical in using delayed enhancement to determine infarct size accurately with magnetic resonance imaging. Circulation. 2002, 106: E6-10.1161/01.CIR.0000019903.37922.9C.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.