Assessment of energy loss across aortic valves using accelerated CMR multi-point flow measurements
Journal of Cardiovascular Magnetic Resonance volume 14, Article number: W9 (2012)
A novel approach for evaluating the performance of artificial or diseased heart valves is presented and applied on in-vitro as well as in-vivo aortic valve data. The method, which is based on turbulence and flow measurements, provides a measure to assess and compare energy dissipation under varying flow conditions.
Diseased or artificial heart valves possibly lead to turbulent flow and regurgitation, both increasing the workload of the heart. Current measures for valve assessment, i.e. effective orifice area, only indirectly and partially correlate with the energy loss due to the valve . Phase-Contrast MRI makes it possible to directly quantify these energy losses, and by relating them to kinetic energy of the flow a parameter describing the hemodynamic performance of the valve can be obtained.
3D Phase-Contrast flow measurements of the aortic arch employing multiple first gradient moments provide not only information about velocities, but also allow conclusions about the turbulence intensities over a large dynamic range. Measurements with 3 different encoding steps in each direction were combined using a Bayesian analysis method adapted from  to estimate the 4D velocity vector field and turbulent kinetic energy (TKE) as proposed in . The TKE was taken and set into relation to the mean kinetic energy (MKE) of the flow. Additionally, the MKE of the forward and the regurgitant flow were compared to yield a more complete picture of possible energy losses (Fig. 1a-b).
In-vitro measurements were performed using a home-built pulsatile flow phantom to study a mechanical St. Jude Medical standard bileaflet valve (St. Jude Medical Inc., St. Paul, MN, USA), as well as a biological Transcatheter Medtronic CoreValve (Medtronic Inc., Minneapolis, MN, USA). In vivo data were acquired in 6 healthy volunteers as well as in two patients with a stenotic valve (valve area 0.9 cm2, mean gradient 34 mmHg) and a Medtronic CoreValve, respectively.
All data were acquired on a 3T Achieva system (Philips Healthcare, Best, The Netherlands) with cardiac triggering and navigator gating. The voxel size was 2 mm isotropic, and temporal resolution was 34 ms. Employing 8-fold undersampling and k-t PCA reconstruction, the nominal scan time was about 8 min without navigator efficiency taken into account.
Fig. 1c shows the relative energy losses due to turbulence and regurgitation in-vitro and in-vivo. The streamline visualization of the flow as well as an isosurface rendering of TKE values in both patients can be seen in Fig. 2. The maximum TKE values were about 150 J/m3 in volunteers, and 950 J/m3 respectively 540 J/m3 in the patients with the stenotic and the artificial valve. Patient stroke volumes were 68 ml and 80 ml, respectively.
A method for assessing valve performance independent of flow rates has been proposed. It has been demonstrated that relative energy loss differs across heart valve designs and values have been found to be 6-fold higher compared to normal subjects. The work indicates that quantification of relative energy losses may provide a potential parameter to characterize the efficiency of the cardiovascular system in general and vascular prostheses in particular.
Akins, et al: JTCS. 2008, 136-
Bretthorst : JMR. 1990, 88-
Dyverfeldt , et al: MRM. 2006, 56-
About this article
Cite this article
Binter, C., Knobloch, V., Manka, R. et al. Assessment of energy loss across aortic valves using accelerated CMR multi-point flow measurements. J Cardiovasc Magn Reson 14 (Suppl 1), W9 (2012). https://doi.org/10.1186/1532-429X-14-S1-W9