- Technical notes
- Open Access
Three-dimensional balanced steady state free precession myocardial perfusion cardiovascular magnetic resonance at 3T using dual-source parallel RF transmission: initial experience
© Jogiya et al.; licensee BioMed Central Ltd. 2014
- Received: 11 April 2014
- Accepted: 30 October 2014
- Published: 28 November 2014
The purpose of this study was to establish the feasibility of three-dimensional (3D) balanced steady-state-free-precession (bSSFP) myocardial perfusion cardiovascular magnetic resonance (CMR) at 3T using local RF shimming with dual-source RF transmission, and to compare it with spoiled gradient echo (TGRE) acquisition.
Dynamic contrast-enhanced 3D bSSFP perfusion imaging was performed on a 3T MRI scanner equipped with dual-source RF transmission technology. Images were reconstructed using k-space and time broad-use linear acquisition speed-up technique (k-t BLAST) and compartment based principle component analysis (k-t PCA).
In phantoms and volunteers, local RF shimming with dual source RF transmission significantly improved B1 field homogeneity compared with single source transmission (P = 0.01). 3D bSSFP showed improved signal-to-noise, contrast-to-noise and signal homogeneity compared with 3D TGRE (29.8 vs 26.9, P = 0.045; 23.2 vs 21.6, P = 0.049; 14.9% vs 12.4%, p = 0.002, respectively). Image quality was similar between bSSFP and TGRE but there were more dark rim artefacts with bSSFP. k-t PCA reconstruction reduced artefacts for both sequences compared with k-t BLAST. In a subset of five patients, both methods correctly identified those with coronary artery disease.
Three-dimensional bSSFP myocardial perfusion CMR using local RF shimming with dual source parallel RF transmission at 3T is feasible and improves signal characteristics compared with TGRE. Image artefact remains an important limitation of bSSFP imaging at 3T but can be reduced with k-t PCA.
- Myocardial perfusion
- Cardiovascular magnetic resonance
- Whole heart imaging
- Balanced steady state free precession
Two-dimensional (2D) myocardial perfusion cardiovascular magnetic resonance (CMR) is increasingly used as a clinical tool for the assessment of myocardial ischemia ,. In recent years, 3T MRI scanners have become more widely available and the higher signal to noise ratio (SNR) at this field strength compared with 1.5T is particularly beneficial for myocardial perfusion CMR . In parallel, more sophisticated data acceleration schemes have been introduced, such as k-space and time broad-use linear speed up technique (k-t BLAST) and k- space and time principal component analysis (k-t PCA), which in principle allow acceleration of image acquisition ≥10 fold . Together, these developments have facilitated the development of three-dimensional (3D) perfusion CMR .
Initial 3D myocardial perfusion studies have used spoiled turbo gradient echo (TGRE) acquisition. Balanced steady state free precession (bSSFP) has become popular for 2D myocardial perfusion CMR at 1.5T because it provides a gain in signal to noise ratio (SNR) of up to 85% and in contrast to noise ratio of up to 50% compared with TGRE . However, bSSFP is more susceptible to artefacts relating to B0 inhomogeneities. In addition, limits on specific absorption rates (SAR) on CMR systems are often addressed by lower B1 amplitudes and hence longer TR, which compromise optimal bSSFP pulse sequences and higher flip angles. All of these effects are more prominent at higher field strength.
A potential solution to these challenges is local RF shimming using dual-source parallel RF transmission, which has been shown to improve image homogeneity, image contrast and diagnostic confidence compared with conventional RF transmission in cardiac cine bSSFP imaging at 3T . The method reduces local SAR hot spots and optimizes the homogeneity of RF deposition. These developments hold particular promise for bSSFP 3D perfusion CMR at 3T.
It was the aim of this study to develop an accelerated bSSFP pulse sequence for 3D whole heart myocardial perfusion CMR at 3T using dual-source parallel RF transmission with local RF shimming and to compare it with a TGRE method.
Study population demographics
28.0 +/− 8.1
23.8 +/− 2.6
CMR set up
Subjects were scanned in supine position on a 3T MRI scanner (Achieva 3T-TX, Philips Healthcare, Best, Netherlands) equipped with a Quasar Dual gradient system (60 mT/m; 200 mT/m/ms) and dual-source parallel RF transmission (MultiTransmit) technology . A 6-channel cardiac phased array receiver coil, 4-lead vectorcardiogram, respiratory belt and blood pressure monitoring were used. For each perfusion scan, an intravenous bolus of 0.075 mmol/kg gadobutrol (Gadovist, Bayer, Germany) was administered followed by a 20 ml saline flush (Spectris Solaris power injector, Pittsburgh, Pennsylvania, USA).
RF Shimming and B1 maps
A cardiac-triggered B1 calibration scan was acquired as one transverse section through the middle of the left ventricle during end-diastole using a saturated double-angle method with a segmented echo-planar imaging readout in a single breath hold. This scan yields the B1 maps for each independent transmit channel (sequentially acquired in the same breathhold), and is used to obtain the phase difference and power ratio of the RF transmit channels for subsequent images with local RF shimming .
The RF shim for each scan was determined automatically by vendor-provided MRI scanner software. This software uses an algorithm to minimize the coefficient of signal variation of the B1 field based on the linear combination of phase and amplitude of the two independent RF transmit channels, based on the B1 data from the calibration scan within the user-defined local shim area. An additional B1 map with RF shimming was acquired using the optimized RF shim settings .
Pulse sequence design
In phantom experiments and in five volunteers, a bSSFP pulse sequence was developed to match a previously validated and described TGRE method  with the following pulse sequence parameters (TR/TE/flip angle 1.8 ms/0.7 ms/15° with a B1-insensitive tailored composite water suppression enhanced through T1 effects (WET) prepulse, saturation prepulse delay 150 milliseconds, acquisition timed to end systole. Using partial echo and a 75% partial Fourier sampling in the ky-kz direction and an elliptical k-space shutter, 10-fold undersampled k-t acquisition with 49 training profiles leading to a net acceleration of 7.0, k-t BLAST and k-t PCA reconstruction , reconstruction of 12 contiguous slices of 5 mm thickness, field of view 350 × 245 mm2, acquired voxel size 2.3 × 2.3 × 5 mm3, interpolated to 1.5 × 1.5 × 5 mm3). Total number of acquired profiles was 106, which results in an acquisition time per heart-beat of 191 ms [106 × 1.8 ms]. Thirty dynamics were acquired in a single inspiratory breath-hold, with every dynamic covering one heart-beat.
Based on these parameters, the bSSFP pulse sequence was optimised to maximise the acquisition flip angle while keeping the TR minimal and the acquisition shot duration to approximately 200 ms. The effect of local RF shimming using dual-source transmission was assessed by quantitative analysis of the B1 field homogeneity. B1 maps were acquired with and without RF shimming. These B1 maps are scaled as a percentage of nominal amplitude. A pixel value of 100 (%) means that the local flip angle corresponds exactly to the nominal (user defined) value.
Comparison between TGRE and bSSFP acquisition
Pulse sequence parameters
Partial Echo, Halfscan (Partial Fourier) in x-y and z planes
2.3 × 2.3 × 10 mm3
2.3 × 2.3 × 10 mm3
Homogeneity of the reconstructed perfusion images was calculated in these five subjects based upon a previously described method . Epicardial and endocardial borders (excluding papillary muscles, trabeculation and dark rim artefacts) were drawn in all ventricular slices. Myocardial mean and standard deviation of signal intensity were measured in the dynamic image with peak signal intensity. The coefficient of variation of the myocardial pixel values was measured using the formula: coefficient of variation = standard deviation/mean signal (expressed as a percentage).
Stress perfusion and detection of coronary artery disease (CAD)
In the last 5 volunteers, bSSFP stress perfusion images were acquired during intravenous adenosine-induced hyperaemia administered for 4 minutes at 140 mcg/kg/min and, following a 20 minute delay, repeated at rest. In addition, in five patients, both TGRE and bSSFP acquisitions at stress were performed in random order. Significant CAD was defined using quantitative coronary angiography (QCA) as >70% diameter stenosis of the left anterior descending, circumflex and right coronary arteries with >2 mm diameter or >50% diameter stenosis for the left main stem as described previously .
Evaluation of image quality and artefacts of MR images
Two experienced observers blinded to clinical details reviewed the CMR images (Extended Workspace, Philips Healthcare, Best, The Netherlands) in a randomised order. The whole heart images were displayed as a complete stack from base to apex and each slice was analysed to produce an overall unified score. Image quality was graded on a scale from 1 to 4 (1 = uninterpretable, 2 = poor, 3 = good, 4 = excellent) for both the standard k-t BLAST and reconstructed k-t PCA images. Image artefacts were categorised as breathing related, subendocardial dark rim artefacts  or related to the reconstruction of undersampled data.
Data were analysed using IBM SPSS, Version 19.0 (SPSS < Chicago, IL, USA). For all analyses P < 0.05 was considered significant. Continuous data were expressed as the mean ± standard deviation. Comparisons between groups were made using two-tailed paired t tests. No corrections were made for multiple comparisons. Image quality scores were compared using the Wilcoxon signed ranks test and McNemar’s test. Discrete data were expressed as percentages.
Baseline demographics of the 25 volunteers and five patients are given in Table 1. CMR examinations were completed in all subjects and no adverse effects occurred during stress acquisitions.
Pulse sequence design
Pulse sequence optimisation with dual source RF transmission in first five volunteers
Flip angle (°/range)
Maximal heart rate (bpm)
Repetition time (ms)
Echo time (ms)
Image quality and artefacts
Image quality scores for the k-t BLAST and k-t PCA images
Signal to noise and signal homogeneity
Signal-to-noise and contrast-to-noise ratio in 5 volunteers
Pulse sequence order
Stress acquisitions and patient studies
This study has shown that the use of local RF shimming with dual source RF transmission significantly improves B1 field homogeneity and facilitates bSSFP 3D whole-heart myocardial perfusion imaging at 3T. Compared with a TGRE method, 3D bSSFP demonstrated significantly improved homogeneity, SNR and CNR but similar image quality. Artefact with bSSFP was more frequent but improved with k-t PCA reconstruction.
High Field Strength
MRI at 3T has rapidly gained acceptance for many clinical applications ,. B1 inhomogeneity and restrictions related to local energy deposition pose particular challenges for cardiac imaging at 3T . Compared with other applications, myocardial perfusion CMR poses additional challenges at high field strength as susceptibility effects increase the B0 inhomogeneity and accentuate banding artefacts, requiring higher order shimming or other corrective methods .
Previous studies have reported the use of bSSFP for two-dimensional myocardial perfusion CMR at 1.5T -. bSSFP provides higher SNR, but at 3T has also been associated with increased off resonance artefact in perfusion CMR and a non-uniform flip angle distribution  causing higher local SAR (due to the higher excitation flip angle). At 3T, these issues are particularly relevant as SAR increases fourfold compared with 1.5T. A shorter TR is desirable for bSSFP but impacts gradient hardware and SAR constraint by increasing the RF duty cycle. Using longer RF pulses prolongs the TR, making off resonance artefacts more prominent.
In the present study compromises were made between the achievable flip angle and maximal heart rate of the acquisition. A flip angle of 35° represented the best compromise between shot duration and reasonable TR in both phantoms and volunteers.
Dual Source RF Transmission at 3T using bSSFP
The dielectric effect is more pronounced at 3T than at 1.5T as the RF wavelength critically approaches the size of the body, influencing standing wave formation. As a consequence, flip angle variation at 3T is high and has previously been shown to range from 32-60% in the heart . This effect contributes to reduced image homogeneity at 3T. Optimising RF shimming optimises power, amplitude, phase and waveform of each RF source to tailor the RF transmission to the patient’s anatomy. Use of two independent RF sources, as in this study, allows local RF shimming with consequent homogenisation of flip angle distribution and optimised SAR distribution within the myocardium, thus reducing local SAR peaks. The use of RF shimming has been successfully demonstrated in spine , abdomen  and breast . In the heart, B1 calibration imaging with local RF shimming and dual-source RF transmission at 3T has been described for bSSFP cine and black blood imaging with improved image quality . The technique was also shown to improve image quality and reduce the number of non-diagnostic segments for dobutamine stress CMR .
In the current study, dual RF transmission allowed shortening of the TR for bSSFP acquisition compared with single source RF transmission at the same flip angle. This permitted a relatively tight acquisition shot duration of around 200 ms for the bSSFP method, comparable to a previously described 3D TGRE myocardial perfusion method ,.
SNR, CNR and homogeneity
Homogeneity can vary in part due to the distance of tissue from the surface . Differences in signal intensity between the septum and lateral wall have previously been described in both bSSFP and TGRE imaging . Although regional differences are thought to be mainly due to B1 inhomogeneities, improved image homogeneity with bSSFP compared with TGRE observed in this study may in part be attributed to the pulse sequence characteristics including dependencies of flip angle variation.
In this study, the use of bSSFP improved SNR, CNR and homogeneity of myocardial signal, compared with an equivalent TGRE method. The increase in SNR and CNR over TGRE was consistent with previous studies at 1.5T ,. However, the magnitude in gain of SNR of bSSFP vs TGRE was not as marked as observed at 1.5T due to several factors including the differences in the coil design, relaxation times of the tissues, flip angles implemented, artefact, SAR limitations and field inhomogeneity .
k-t PCA and k-t BLAST
Although detailed study was beyond the scope of this article, differences in k-t PCA and k-t BLAST were observed, consistent with previous reports . In summary k-t PCA has a more efficient algorithm by defining spatial compartments within the 3D volume of a perfusion dataset. The number of overlapping signals is reduced and higher temporal frequency components are more accurately reconstructed. These effects are expected to improve temporal fidelity and reduce signal contamination artefact with k-t PCA and improve image quality. In this study, the use of k-t PCA reconstruction reduced the amount of artefact but this was not associated with a significant improvement of image quality.
Image quality and diagnosis of CAD
Whilst increased signal of bSSFP acquisition should have advantages for diagnosis and analysis , in this study it did not translate into improved image quality. This was largely due to the higher rate of artefacts with bSSFP. The main artefact observed was subendocaridal dark rim artefact. Although in the 5 patients studied here, dark rim artefact did not affect the correct diagnosis of CAD, the presence of this artefact may lead to false positive reporting of ischemia.
3D Myocardial Perfusion CMR
3D acquisition has several potential advantages for myocardial perfusion CMR over the conventionally used 2D methods, including better coverage, higher signal-to-noise and fewer compromises regarding the cardiac phase in which data are acquired. Developing 3D myocardial perfusion CMR has been challenging and only recent technological developments including multi-element coils and spatio-temporal undersampling methods have permitted the very fast data acquisition required for 3D myocardial perfusion CMR. The use of bSSFP at 3T, as shown in this study, is a promising further development and warrants evaluation in larger patient populations. Another recent study has reported the use of bSSFP for 3D myocardial perfusion CMR, but with a different implementation and at 1.5T . Giri et al. used no magnetization preparation but maintained the steady state by continuous application of the bSSFP kernel throughout the scan. This novel approach was feasible and demonstrated good signal characteristics. As the imaging was gated to mid-diastole, the acquisition was however limited to either single slice imaging or single slab 3D encoding and the read-out duration was longer than in our method, ranging between 300-380 ms under rest conditions. In our study it was the aim to keep the acquisition close to 200 ms comparable to previous published work, to avoid potential motion artefacts, which are more likely with a faster heart rate as seen during stress acquisitions.
Implications for patient care
Three-dimensional myocardial perfusion CMR offers better myocardial coverage than conventionally used 2D methods. 3D bSSFP myocardial perfusion CMR at 3T potentially offers further improvement of signal characteristics and may enhance the use of three-dimensional myocardial perfusion CMR for clinical application.
The flip angle of 35 degrees we chose may not be optimal for bSSFP acquisition but was the maximal achievable within the restrictions of SAR, TR, heart rate and the desired acquisition shot duration. Higher flip angles may have improved the signal and image quality further and future studies should explore the optimal compromise between flip angle, image resolution and acquisition duration further. The differences in acquisition window may also account for some of the image quality discrepancies as it was anticipated that there would be more temporal blurring in the bSSFP group. The use of k-t PCA may have mitigated some of this effect however.
The maximal heart rate of 83 bpm is a limitation, although it was not exceeded by volunteers and patients in this study even during stress studies. In many patients, however, higher heart rates will be encountered. A reduction of the read-out duration or saturation pulse delay, or acquisition at every other heartbeat can be relatively easily employed to deal with higher heart rates but the impact on diagnostic accuracy will need to be evaluated.
Lastly, we recognise the reduction in TR associated with the use of dual source parallel RF transmission in this study is the consequence of many factors not least the particular SAR model used by the vendor of the MRI scanner used here. We would not consider this “optimal” as the resulting long acquisition time led to poor image quality. A number of approaches including dieletric cushions, post processing as well as a fixed anatomy dependent mode of RF shimming could have been used to improve B1 homogeneity. Many of these are vendor specific and were not available for the current study.
Three-dimensional bSSFP myocardial perfusion CMR using local RF shimming with dual source parallel RF transmission at 3T is feasible and improves signal characteristics compared with TGRE. Image artefact remains an important limitation of bSSFP perfusion imaging at 3T.
We thank Lorna A Smith, Research Radiographer, Department of Medical Imaging, University College London, for her invaluable help with implementing the perfusion sequences.
- Hamon M, Fau G, Nee G, Ehtisham J, Morello R, Hamon M: Meta-analysis of the diagnostic performance of stress perfusion cardiovascular magnetic resonance for detection of coronary artery disease. J Cardiovasc Magn Reson. 2010, 12 (1): 29-10.1186/1532-429X-12-29.PubMed CentralView ArticlePubMedGoogle Scholar
- Greenwood JP, Maredia N, Younger JF, Brown JM, Nixon J, Everett CC, Bijsterveld P, Ridgway JP, Radjenovic A, Dickinson CJ, Ball SG, Plein S: Cardiovascular magnetic resonance and single-photon emission computed tomography for diagnosis of coronary heart disease (CE-MARC): a prospective trial. Lancet. 2012, 379 (9814): 453-60. 10.1016/S0140-6736(11)61335-4.PubMed CentralView ArticlePubMedGoogle Scholar
- Oshinski JN, Delfino JG, Sharma P, Gharib AM, Pettigrew RI: Cardiovascular magnetic resonance at 3.0 T: current state of the art. J Cardiovasc Magn Reson. 2010, 12: 55-10.1186/1532-429X-12-55.PubMed CentralView ArticlePubMedGoogle Scholar
- Vitanis V, Manka R, Giese D, Pedersen H, Plein S, Boesiger P, Kozerke S: High resolution three-dimensional cardiac perfusion imaging using compartment-based k-t principal component analysis. Magn Reson Med. 2011, 65 (2): 575-8. 10.1002/mrm.22620.View ArticlePubMedGoogle Scholar
- Jogiya R, Kozerke S, Morton G, De Silva K, Redwood S, Perera D, Nagel E, Plein S: Validation of Dynamic 3-Dimensional Whole Heart Magnetic Resonance Myocardial Perfusion Imaging Against Fractional Flow Reserve for the Detection of Significant Coronary Artery Disease. J Am Coll Cardiol. 2012, 60 (8): 756-6. 10.1016/j.jacc.2012.02.075.View ArticlePubMedGoogle Scholar
- Wang Y, Moin K, Akinboboye O, Reichek N: Myocardial first pass perfusion: steady-state free precession versus spoiled gradient echo and segmented echo planar imaging. Magn Reson Med. 2005, 5: 1123-9. 10.1002/mrm.20700.View ArticleGoogle Scholar
- Mueller A, Kouwenhoven M, Naehle CP, Gieseke J, Strach K, Willinek WA, Schild HH, Thomas D: Dual-source radiofrequency transmission with patient-adaptive local radiofrequency shimming for 3.0-T cardiac MR imaging: initial experience. Radiology. 2012, 263 (1): 77-85. 10.1148/radiol.11110347.View ArticlePubMedGoogle Scholar
- Krishnamurthy R, Pednekar A, Kouwenhoven M, Cheong B, Muthupillai R: Evaluation of a subject specific dual-transmit approach for improving B1 field homogeneity in cardiovascular magnetic resonance at 3T. J Cardiovasc Magn Reson. 2013, 15 (1): 68-10.1186/1532-429X-15-68.PubMed CentralView ArticlePubMedGoogle Scholar
- Plein S, Bloomer TN, Ridgway JP, Jones TR, Bainbridge GJ, Sivananthan MU: Steady-state free precession magnetic resonance imaging of the heart: comparison with segmented k-space gradient-echo imaging. J Magn Reson Imaging. 2001, 14 (3): 230-6. 10.1002/jmri.1178.View ArticlePubMedGoogle Scholar
- Di Bella EV, Parker DL, Sinusas AJ: On the dark rim artifact in dynamic contrast-enhanced MRI myocardial perfusion studies. Magn Reson Med. 2005, 54 (5): 1295-9. 10.1002/mrm.20666.PubMed CentralView ArticlePubMedGoogle Scholar
- Craven I, Griffiths PD, Hoggard N: Magnetic resonance imaging of epilepsy at 3 Tesla. Clin Radiol. 2011, 66 (3): 278-86. 10.1016/j.crad.2010.10.010.View ArticlePubMedGoogle Scholar
- Kim PD, Truwit CL, Hall WA: Three-tesla high-field applications. Neurosurg Clin N Am. 2009, 20 (2): 173-8. 10.1016/j.nec.2009.04.009.View ArticlePubMedGoogle Scholar
- Soher BJ, Dale BM, Merkle EM: A review of MR physics: 3T versus 1.5T. Magn Reson Imaging Clin N Am. 2007, 15 (3): 277-90. 10.1016/j.mric.2007.06.002. vView ArticlePubMedGoogle Scholar
- Wieben O, Francois C, Reeder SB: Cardiac MRI of ischemic heart disease at 3T: potential and challenges. Eur J Radiol. 2008, 65 (1): 15-28. 10.1016/j.ejrad.2007.10.022.View ArticlePubMedGoogle Scholar
- Morton G, Ishida M, Schuster A, Hussain S, Schaeffter T, Chiribiri A, Nagel E: Perfusion cardiovascular magnetic resonance: Comparison of an advanced, high-resolution and a standard sequence. J Cardiovasc Magn Reson. 2012, 14: 34-10.1186/1532-429X-14-34.PubMed CentralView ArticlePubMedGoogle Scholar
- Merkle N, Wöhrle J, Grebe O, Nusser T, Kunze M, Kestler HA, Kochs M, Hombach V: Assessment of myocardial perfusion for detection of coronary artery stenoses by steady-state, free-precession magnetic resonance first-pass imaging. Heart. 2007, 93 (11): 1381-5. 10.1136/hrt.2006.104232.PubMed CentralView ArticlePubMedGoogle Scholar
- Hunold P, Maderwald S, Eggebrecht H, Vogt FM, Barkhausen J: Steady-state free precession sequences in myocardial first-pass perfusion MR imaging: comparison with TurboFLASH imaging. Eur Radiol. 2004, 14: 409-16. 10.1007/s00330-003-2176-1.View ArticlePubMedGoogle Scholar
- Sung K, Nayak KS: Measurement and characterization of RF nonuniformity over the heart at 3T using body coil transmission. J Magn Reson Imaging. 2008, 27 (3): 643-8. 10.1002/jmri.21253.View ArticlePubMedGoogle Scholar
- Nelles M, Konig RS, Gieseke J, Guerand-van Battum MM, Kukuk GM, Schild HH, Willinek WA: Dual-source parallel RF transmission for clinical MR imaging of the spine at 3.0 T: intraindividual comparison with conventional single-source transmission. Radiology. 2010, 257 (3): 743-53. 10.1148/radiol.10092146.View ArticlePubMedGoogle Scholar
- Rao RK, Riffel P, Meyer M, Kettnaker PJ, Lemke A, Haneder S, Schoenberg SO, Michaely HJ: Implementation of dual-source RF excitation in 3 T MR-scanners allows for nearly identical ADC values compared to 1.5 T MR scanners in the abdomen. PLoS One. 2012, 7 (2): e32613-10.1371/journal.pone.0032613.PubMed CentralView ArticlePubMedGoogle Scholar
- Rahbar H, Partridge SC, Demartini WB, Gutierrez RL, Parsian S, Lehman CD: Improved B‚ÇÅ homogeneity of 3 Tesla breast MRI using dual-source parallel radiofrequency excitation. J Magn Reson Imaging. 2012, 35 (5): 1222-6. 10.1002/jmri.23571.PubMed CentralView ArticlePubMedGoogle Scholar
- Strach K, Clauberg R, Müller A, Wonneberger U, Naehle CP, Kouwenhoven M, Gieseke J, Schild HH, Thomas D: Feasibility of high-dose dobutamine stress SSFP Cine MRI at 3 Tesla with patient adaptive local RF Shimming using dual-source RF transmission: initial results. Rofo. 2013, 185 (1): 34-9.PubMedGoogle Scholar
- Constantinides CD, Westgate CR, O’Dell WG, Zerhouni EA, McVeigh ER: A phased array coil for human cardiac imaging. Magn Reson Med. 1995, 34 (1): 92-8. 10.1002/mrm.1910340114.PubMed CentralView ArticlePubMedGoogle Scholar
- Gutberlet M, Noeske R, Schwinge K, Freyhardt P, Felix R, Niendorf T: Comprehensive cardiac magnetic resonance imaging at 3.0 Tesla: feasibility and implications for clinical applications. Invest Radiol. 2006, 41 (2): 154-67. 10.1097/01.rli.0000195840.50230.10.View ArticlePubMedGoogle Scholar
- Giri S, Xue H, Maiseyeu A, Kroeker R, Rajagopalan S, White RD, Zuehlsdorff S, Raman SV, Simonetti OP. Steady-state first-pass perfusion (SSFPP): A new approach to 3D first-pass myocardial perfusion imaging.Magn Reson Med. 2013; doi:10.1002/mrm.24638.,Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.