- Technical notes
- Open Access
T2STIR preparation for single-shot cardiovascular magnetic resonance myocardial edema imaging
Journal of Cardiovascular Magnetic Resonance volume 21, Article number: 72 (2019)
Myocardial edema in acute myocardial infarction (AMI) is commonly imaged using dark-blood short tau inversion recovery turbo spin echo (STIR-TSE) cardiovascular magnetic resonance (CMR). The technique is sensitive to cardiac motion and coil sensitivity variation, leading to myocardial signal nonuniformity and impeding reliable depiction of edematous tissues. T2-prepared balanced steady state free precession (T2p-bSSFP) imaging has been proposed, but its contrast is low, and averaging is commonly needed. T2 mapping is useful but requires a long scan time and breathholding. We propose here a single-shot magnetization prepared sequence that increases the contrast between edema and normal myocardium and apply it to myocardial edema imaging.
A magnetization preparation module (T2STIR) is designed to exploit the simultaneous elevation of T1 and T2 in edema to improve the depiction of edematous myocardium. The module tips magnetization down to the –z axis after T2 preparation. Transverse magnetization is sampled at the fat null point using bSSFP readout and allows for single-shot myocardial edema imaging. The sequence (T2STIR-bSSFP) was studied for its contrast behavior using simulation and phantoms. It was then evaluated on 7 healthy subjects and 7 AMI patients by comparing it to T2p-bSSFP and T2 mapping using the contrast-to-noise ratio (CNR) and the contrast ratio as performance indices.
In simulation and phantom studies, T2STIR-bSSFP had improved contrast between edema and normal myocardium compared with the other two edema imaging techniques. In patients, the CNR of T2STIR-bSSFP was higher than T2p-bSSFP (5.9 ± 2.6 vs. 2.8 ± 2.0, P < 0.05) but had no significant difference compared with that of the T2 map (T2 map: 6.6 ± 3.3 vs. 5.9 ± 2.6, P = 0.62). The contrast ratio of T2STIR-bSSFP (2.4 ± 0.8) was higher than that of the T2 map (1.3 ± 0.1, P < 0.01) and T2p-bSSFP (1.4 ± 0.5, P < 0.05).
T2STIR-bSSFP has improved contrast between edematous and normal myocardium compared with commonly used bSSFP-based edema imaging techniques. T2STIR-bSSFP also differentiates between fat that was robustly suppressed and fluids around the heart. The technique is useful for single-shot edema imaging in AMI patients.
As myocardial edema has long T2, T2-weighted cardiovascular magnetic resonance (CMR) imaging is usually used in the imaging of acute myocardial infarction (AMI) [1,2,3], where edema commonly occurs. T2-weighted turbo spin echo (TSE) combined with short tau inversion recovery (STIR) and dark-blood preparation (STIR-TSE) has been widely used clinically for this purpose . However, the technique is sensitive to cardiac motion, leading to signal loss and hence myocardial signal inhomogeneity. Coil sensitivity variation increases variation of the myocardial signal. Additionally, the stagnant blood at the subendocardial rim would sometimes mimic edematous tissues. These issues of STIR-TSE impede reliable depiction of edematous tissues [5, 6]. Moreover, STIR-TSE is sensitive to arrhythmia and respiratory motion.
Several CMR techniques have been proposed to address these issues of STIR-TSE for myocardial edema imaging. The STIR pulse in STIR-TSE is sometimes replaced by the spectral attenuated inversion recovery (SPAIR) pulse for fat suppression  to reduce motion sensitivity. The method is sensitive to main field inhomogeneity. It has also been shown that the technique depends on the proper choice of image readout time . Single-shot T2-prepared balanced steady state free precession (T2p-bSSFP)  shows higher diagnostic accuracy than STIR-TSE for edema imaging [9, 10]. It is robust to cardiac motion and arrhythmia and avoids the bright subendocardial rims caused by stagnant blood that mimic myocardial edema as in STIR-TSE. However, it is sensitive to coil sensitivity variation, and multiple acquisitions are commonly needed to improve image SNR. Another method, ACUT2E , improves T2 weighting intrinsic in bSSFP by using 180° excitation pulses. Yet, high flip angles for the excitation pulses might not be achieved due to the specific absorption rate (SAR) limit and the technique’s sensitivity to transmit (B1+) field inhomogeneity, especially at a high field. Similar to STIR-TSE, ACUT2E is a segmented sequence and is sensitive to arrhythmia and respiratory motion as well.
T2 mapping [12,13,14,15] CMR was proposed for myocardial edema imaging . T2 quantification can help objectively distinguish between normal and edematous myocardium. It can also detect myocarditis with systematic T2 elevation in myocardium. In this method, several T2p-bSSFP images with different echo times are acquired over multiple heart beats. The T2 map is then obtained through pixelwise curve fitting of these T2-weighted images upon image registration. The technique is sensitive to arrhythmia and requires breathholding. Moreover, the accuracy of T2 maps depends on multiple factors, e.g., heart rate, fitting model, the patient’s breathholding ability, off-resonance effect, etc. [13, 17,18,19].
Recent parametric mapping studies on myocardial edema reaffirmed earlier findings  that both T1 and T2 values are increased in edematous tissues [21, 22]. Although the native T1 of myocardium also increases in the presence of fibrosis, there is no measurable fibrosis expected at 2 to 5 days post-reperfusion. Therefore, any increase in myocardial T1 most likely comes from edema. Thus, a CMR sequence that exploits the elevation of both T1 and T2 values may yield an edema detection method at a higher specificity. bSSFP cine, whose signal is proportional to √(T2/T1), has been proposed to depict edematous tissue . However, the signal equation shows that the signal increase due to T2 elevation is attenuated by the concomitant increase in T1. The resulting signal change is small. The method is therefore sensitive to coil sensitivity variation. Postcontrast bSSFP cine shortens myocardial T1 values and improves the contrast between edematous and normal myocardium . The use of contrast agents increases patient risk. Also, the image contrast depends on contrast dosage and the delay time after contrast injection.
This study proposes a novel single-shot CMR technique for edema imaging. The new technique uses a magnetization preparation module called T2STIR (T2 prepared inversion with STIR) to exploit the simultaneous elevation of T1 and T2 in edematous tissues for improved differentiation of edematous and normal myocardium while suppressing fat. The module is combined with single-shot bSSFP, making it insensitive to arrhythmia and breathing motion. The performance of the resulting single-shot sequence is assessed by simulation, a phantom study, and in vivo studies in healthy subjects. The technique is finally applied to a small number of patients with known myocardial edema to demonstrate its clinical feasibility. The preliminary work of this study was first reported in .
The T2STIR preparation module
Figure 1 (a) shows the signal evolutions of edema and normal myocardium with the T2STIR module. Contrast of the T2STIR module between edema and normal myocardium is generated by T2 preparation together with spin inversion. Upon inversion of the T2 prepared magnetization, the longitudinal magnetization of a tissue is given by:
Here, TEprep is the echo time of the T2 preparation module, and T2 is the spin-spin relaxation time of the tissue. The resulting magnetization is then sampled at TIfat, the fat null point, which is given by:
where T1, fat and T2, fat are the T1 and T2 relaxation times for fat.
After spin inversion, normal myocardium recovers faster than edematous myocardium (see Fig. 1(a)). With magnitude reconstruction, inverted magnetization is “reverted”: edematous tissues now have higher signal intensity than normal myocardium. The inversion and the partial recovery over TI accentuate the signal difference between edema and normal myocardium generated by T2 preparation alone. Here, the choice of TI to null fat is similar to the STIR pulse. This new magnetization preparation module is hereby referred to as T2STIR.
Equation (2) shows that TIfat is determined by T1 and T2 of fat (which are 449 ms and 53 ms at 3 T, respectively ) and the TEprep used in the T2 preparation part of the pulse. For instance, at TEprep = 60 ms, TIfat would be 125 ms.
T2STIR-prepared single-shot edema imaging
Figure 1(b) shows the implementation of the T2STIR module designed based on the modified B1-insensitive rotation pulse (mBIR-4) . It has three components: a reverse adiabatic half passage (rAHP) pulse, followed by an adiabatic full passage (AFP) pulse and finally an adiabatic half passage (AHP) pulse. The AFP pulse is used twice to reduce the module’s sensitivity to B0 and B1+ field inhomogeneity . The phase shift and delay times among the four pulses are designed to flip the transverse magnetization down to the –Mz direction after T2 preparation. The rAHP pulse, AFP pulse and AHP pulse are 2.56 ms, 5.12 ms and 2.56 ms long, respectively, giving a minimum TEprep of 15.36 ms. The frequency sweep is 4.88 kHz. A spoiler gradient at the end of the module destroys any coherent transverse magnetization after the preparation module. Figure 1(b) shows the amplitude and frequency modulations of this adiabatic T2STIR preparation pulse based on the tan/tanh modulation function.
Imaging is performed using the single-shot bSSFP readout, similar to that of T2p-bSSFP. Linear flip angle (LFA) pulses for catalysis  and linear reordering are used to minimize transient oscillations in bSSFP at the start of acquisition . The acquisition is timed to mid-diastole. The sequence is hereafter referred to as T2STIR-bSSFP.
In T2STIR-bSSFP, TIfat is the time from the end of the T2STIR module to the k-space center of the bSSFP readout. Let TIfill be the fill time between the end of the T2STIR module and the start of bSSFP readout. Then,
where R is the number of LFA pulses, and N is the number of lines acquired between the end of the LFA pulse train and the k-space centerline. As TIfill ≥ 0, the choice of R and N together set the lower limit of TIfat, and hence the maximum TEprep allowed. Experience from T2p-bSSFP shows that a TEprep of 60 ms is commonly used to generate T2-induced signal differences for the differentiation of edema and normal myocardium . If this same TEprep is used in the T2STIR module, it corresponds to a TIfat of approximately 125 ms and would satisfy Eq. (3).
T2STIR-bSSFP was compared with T2p-bSSFP with TEprep varying from 40 ms to 70 ms through Bloch equation simulation using MATLAB (version 2017a, The MathWorks Inc., Natick, Massachusetts, USA). The parameters for bSSFP readout used in the simulation were as follows: 10 LFA pulses, 22 phase encoding lines before the k-space center, flip angle = 60°, TE/TR = 1.3/2.6 ms. The T1 and T2 values were 1139 ms and 52 ms for normal myocardium and 1434 ms and 75 ms for edematous tissue. They were the relaxation parameters of the compartments in the phantom study below.
A phantom experiment was performed to evaluate the performance of T2STIR-bSSFP and compare it with T2p-bSSFP and T2 mapping . The sequence was implemented on a 3 T clinical CMR system (TIM TRIO, Siemens Healthineers, Erlangen, Germany) for all experiments. A two-compartment phantom, one for normal myocardium and the other for edema, was built with agar doped with NiCl2. The T1 values of the compartments were measured using an inversion recovery spin echo sequence (TR = 10 s, TI changing from 100 ms to 3600 ms). The T2 values were measured using a spin echo sequence (TR = 10 s, TE varying from 10 ms to 150 ms). The T1 and T2 values were 1139 ms and 52 ms for normal myocardium compartment and 1434 ms and 75 ms for edema compartment, respectively.
The phantom was imaged with T2STIR-bSSFP, T2p-bSSFP, and T2 mapping . Simulated electrocardiogram (ECG) with an RR interval of 800 ms was used. The value of TEprep in the T2STIR and T2p modules was varied from 40 ms to 70 ms in 5 ms increments. Imaging parameters identical to all sequences were as follows: FOV = 240 × 83 mm2, matrix size = 128 × 44, pixel size = 1.9 × 1.9 mm2, slice thickness = 8 mm, TE/TR = 1.3/2.6 ms, bandwidth = 1447 Hz/pixel, flip angle = 60°. Parallel imaging was not used, allowing accurate estimation of signal and noise in the images. The experiments were repeated four times. Images were then averaged for analysis.
Healthy subject study
A protocol for in vivo imaging was developed and evaluated in healthy subjects. Blood was used as a surrogate for edema because its T1 and T2 values were both longer than normal myocardium. The study was approved by the institutional review board (IRB) at Shenzhen Institutes of Advanced Technology. Seven healthy subjects (5 males, 26 ± 3 years) were recruited, and informed consent was obtained. For each subject, three short axis slices (basal, mid-ventricular and apical) and one horizontal long axis of the heart were acquired using T2STIR-bSSFP and T2p-bSSFP. Imaging parameters for both single-shot sequences were identical: FOV = 360 × 260 mm2, matrix size = 192 × 138, pixel size = 1.9 × 1.9 mm2, slice thickness = 8 mm, TE/TR = 1.3/2.6 ms, flip angle = 60°, bandwidth = 1447 Hz/pixel, phase resolution = 75%, TEprep = 60 ms, and TIfill was 20 ms (TIfat = 125 ms) in T2STIR-bSSFP. GRAPPA rate 2 with 24 auto-calibration lines was used to reduce the image acquisition window.
The ability of the three imaging techniques (T2STIR-bSSFP, T2p-bSSFP and T2 mapping) in detecting edema was evaluated in AMI patients. The study was approved by the IRB of West China Hospital, where patient recruitment and scanning were carried out. Seven patients (5 males, 60 ± 10 years) with ST-segment elevation myocardial infarction (STEMI), identified by clinical presentation, ECG and coronary angiograms, were recruited after reperfusion. Informed consent was obtained from each patient. The patients were examined on days 2 to 5 post-reperfusion using the institution’s standard CMR protocols, including STIR-TSE, T1 maps  and T2 maps  on a 3 T CMR scanner (TIM TRIO, Siemens Healthineers). The patients were then imaged with T2STIR-bSSFP and T2p-bSSFP (using the protocols in the healthy subject study). Late gadolinium enhancement (LGE) images were then acquired for each patient approximately 10 min after the injection of gadolinium-based contrast agent (0.15 mmol/kg, Magnevist, Bayer, Whippany, New Jersey, USA). Imaging parameters of the relevant sequences are listed in Table 1.
The performance of T2STIR-bSSFP was assessed by (1) the signal difference in the simulation or the contrast-to-noise ratio (CNR) in images and (2) the contrast ratio between edema and normal myocardium . In the images, signal intensity (SI) was given by the mean signal of the region of interest (ROI). Noise was given by the standard deviation (SD) of normal myocardium . The equations to find the signal difference, signal to noise ratio (SNR), CNR, and contrast ratio for T2p-bSSFP and T2STIR-bSSFP, as well as the calculation of SNR, CNR and contrast ratio for the T2 maps, are defined in the Additional file 1.
In the phantom study, ROIs were manually drawn for each compartment. The signal difference and the contrast ratio between the normal and edematous myocardium compartments of the phantom for T2STIR-bSSFP and T2p-bSSFP at each TEprep were found. The correlation between the simulation and the phantom study was assessed by linear regression. The contrast ratio of the T2 value between the normal and edematous myocardium compartments in the T2 map was also calculated.
For healthy subjects, the ROIs of myocardium and blood pool were manually drawn for each subject. The myocardial signal was measured at the septum. Blood signals were measured at the center of the blood pool in the left ventricle. The CNRs and contrast ratios between the myocardium and the blood (surrogate for edema) of T2STIR-bSSFP and T2p-bSSFP were calculated for each slice. The average CNRs and contrast ratios over all the slices were recorded for each sequence.
For AMI patients, the ROI of remote myocardium was manually drawn on the LGE images by an experienced reader (5-year CMR experience). This ROI was copied and pasted on T1 map and T2 map to keep the ROI position the same as on the LGE image. The ROIs on T1 and T2 maps were slightly adjusted to ensure the entire ROI was inside the myocardium if the myocardium on T1 or T2 map was not aligned with the myocardium on LGE image due to motion. The mean and SD values of these regions were calculated for the T1 and T2 maps. Myocardial regions with T1 and T2 values larger than the mean + 2 SD of the remote myocardium were identified as edematous regions. At the regions where both T1 and T2 were elevated (the two parametric maps may have slightly different shapes), contours for signal measurement were manually drawn inside these regions. Regions of microvascular obstructions (identified on LGE images) and artifacts (i.e., motion artifact, banding artifact), if present, were excluded from the ROIs. The CNR and contrast ratio between the myocardium and edema in each image were calculated for T2STIR-bSSFP, T2p-bSSFP, and T2 mapping. The two performance indices were then averaged among the patients for each imaging method compared in this study. The CNR and contrast ratio differences (given as the mean ± SD) among the three sequences compared were tested for statistical significance. The data were tested for a normal distribution using the Kolmogorov-Smirnov Test. The CNR and the contrast ratio among the different methods were compared using the Mann-Whitney U-test. Statistical significance was set at P < 0.05. The extents of edema determined by the different methods are shown in Additional file 1. Note that the STIR-TSE was not included in the sequence comparison in this study since previous clinical studies already showed that T2p-bSSFP has better diagnostic accuracy than STIR-TSE [8, 10].
The variation in signal difference and contrast ratio between normal myocardium and edema with TEprep in T2STIR-bSSFP and T2p-bSSFP are shown in Fig. 2. The contrast ratio of the T2 values between the two compartments is also shown in Fig. 2(b) for comparison. The signal differences in T2STIR-bSSFP are higher than those of T2p-bSSFP and are insensitive to TEprep in both cases. Meanwhile, the contrast ratios of the two sequences increase with TEprep. The contrast ratio increases faster with TEprep for T2STIR-bSSFP than for T2p-bSSFP. T2STIR-bSSFP has the highest contrast ratio among the three edema imaging sequences compared. Note that the contrast ratios from T2p-bSSFP and the T2 map are similar, as both were T2 prepared in the same way.
At TEprep = 60 ms, the signal difference between edema and normal myocardium in T2STIR-bSSFP is 15% higher than that of T2p-bSSFP. At the same TEprep, the contrast ratios between edema and normal myocardium for T2p-bSSFP and T2 map are approximately the same and are approximately 30% lower than that of T2STIR-bSSFP.
The signal differences of the simulation and the phantom study showed strong correlations for T2p-bSSFP (R2 = 0.90) and T2STIR-bSSFP (R2 = 0.89) (Fig. 3(a)). Excellent correlations between the contrast ratios of the simulation and the phantom study were obtained for T2p-bSSFP (R2 = 0.99, slop = 0.99) and T2STIR-bSSFP (R2 = 0.98, slop = 0.98) (Fig. 3(b)). The T2 values measured by T2 mapping were 74 ms and 55 ms for the two phantom compartments representing edema and normal myocardium, respectively. The corresponding contrast ratio for the T2 map (1.36) is very close to the expected value (1.44).
In vivo experiments
Healthy subject study
The average CNR between myocardium and blood (surrogate for edema) of T2STIR-bSSFP was higher than that for T2p-bSSFP (29.6 ± 8.3 vs. 19.0 ± 5.7, P < 0.001). The average contrast ratio of T2STIR-bSSFP was also higher than that of T2p-bSSFP (6.1 ± 0.9 vs. 2.9 ± 0.5, P < 0.001).
Among the seven AMI patients, four had microvascular obstruction. They were identified and excluded from the corresponding ROIs. Table 2 summarizes the mean and SD of SI and SNR for normal and edematous myocardium for all the edema imaging techniques compared. Among the three methods, T2 map has the smallest coefficient of variation (which is the reciprocal of SNR). Figure 4 shows the CNRs and the contrast ratios of T2 map, T2p-bSSFP, and T2STIR-bSSFP in bar graphs. The CNR of T2STIR-bSSFP was higher than that of T2p-bSSFP (5.9 ± 2.6 vs. 2.8 ± 2.0, P < 0.05). There was no significant difference between the CNRs of the T2 map and T2STIR-bSSFP (T2 map: 6.6 ± 3.3 vs. 5.9 ± 2.6, P = 0.62). However, the contrast ratio of T2STIR-bSSFP (2.4 ± 0.8) was higher than that of the T2 map (1.3 ± 0.1, P < 0.01) and T2p-bSSFP (1.4 ± 0.5, P < 0.05).
Figure 5 shows the images from one AMI patient. The infarct region in the LGE image matched the location where both T1 and T2 were elevated in the two parametric maps. The dark core inside the edema/infarct in LGE (where T1 and T2 in the parametric maps were both low) most likely corresponded to microvascular obstruction. The signal from the normal myocardium in T2STIR-bSSFP was lower than that of T2p-bSSFP, making the edematous region in T2STIR-bSSFP obvious. Note the bright signal surrounding the heart (yellow arrow) in both single-shot bSSFP images. The fat-suppressed T2STIR-bSSFP image suggested that it was likely pericardial fluid. In the T2p-bSSFP image, both epicardial fat and pericardial fluid were bright. Edema was also depicted in STIR-TSE (Fig. 5(d)).
Figure 6 shows the images from another AMI patient. T1 and T2 elevation from the parametric maps (Fig. 6(c) and (f)) at the septal wall suggested the presence of edema. Enhancement in the LGE image (Fig. 6(e)) matched the position of edema in the parametric maps. Again, the signal for remote myocardium in the T2STIR-bSSFP image was lower than that in the T2p-bSSFP image, while the reverse was true for edema. The contrast improvement in T2STIR-bSSFP compared with T2p-bSSFP was obvious. The edematous region also appeared in STIR-TSE (Fig. 6(d)). However, the signal of the normal myocardium is highly non-uniform and might be affected by coil sensitivity variation. The T2STIR-bSSFP images from the remaining 5 patients are shown in Additional file 1: Figure S2.
This study proposes a novel magnetization preparation module, T2STIR, for single-shot myocardial edema imaging and demonstrates its relevance to myocardial edema imaging. Compared with the T2p module, the T2STIR module increases the contrast between edematous tissues and normal myocardium without increasing SAR. The simulation and phantom experimental results agreed well with each other. Both showed improved contrast of T2STIR-bSSFP compared with T2p-bSSFP and T2 mapping. In vivo experiments showed that T2STIR-bSSFP outperformed the other two common edema imaging methods: in healthy subjects, the technique showed an improved CNR and contrast ratio between blood and myocardium compared with T2p-bSSFP. In patients, T2STIR-bSSFP had higher CNR than T2p-bSSFP and the highest contrast ratio compared to T2p-bSSFP and T2 mapping.
A T2 preparation pulse combined with inversion recovery has been proposed in previous studies [32,33,34,35,36]. However, the inversion times (TI) in these applications were chosen to null tissues other than fat (normal myocardium, blood, etc.). The T2STIR module proposed here is novel in that (1) TI is selected to suppress the tissue with the shortest T1 (fat in this case) and (2) the inversion pulse is integrated with the T2 preparation module and reduces the SAR. Although blood is not nulled in T2STIR-bSSFP, no flow artifact was observed in the imaging experiments, especially in the long axis view. The reason is that the T2STIR pulse is nonselective. Additionally, image acquisition is performed in mid-diastole when blood flow is minimal. Flow artifacts are therefore much less likely.
A key feature of T2STIR-bSSFP compared with other gradient echo-based edema imaging sequences is its robust fat suppression. It removes the dark rim at the fat-water interface and improves tissue visualization (Fig. 5(a)). Fat suppression in T2STIR-bSSFP also helps differentiate pericardial fluid (common in AMI patients) from epicardial fat, both bright in bSSFP images (Fig. 5(b)). While fat suppression is possible with spectrally selective pulses in T2p-bSSFP, the technique is sensitive to main field inhomogeneity and increases SAR, especially at a high field. Theoretically, the performance of fat suppression in T2STIR-bSSFP would be affected by the T1 and T2 values of fat according to Eq. . Incomplete inversion and the excitation pulses from the bSSFP readout before the k-space center would also affect fat suppression. Our experience with the single-shot T2STIR-bSSFP from the in vivo imaging experiments using TEprep = 60 ms revealed that a TIfat ranging from 110 ms to 130 ms can effectively suppress fat. The choice of TIfat is therefore quite forgiving in practice.
The proposed T2STIR magnetization preparation module would be attractive at a main magnetic field beyond 3 T for two reasons. First, the T2STIR module will have improved contrast. As T1 increases with the magnetic field, the T1 difference between edematous and normal myocardium is increased. The contrast between the two tissues would therefore increase at a higher magnetic field with the T2STIR module. Additionally, the longer T1 of fat at higher field strength allows the use of longer TEprep (while TIfat satisfies Eq. ), further improving the edema contrast of this technique. Second, T2STIR-bSSFP is generally more favorable to other multishot edema imaging techniques in terms of SAR. For instance, single-shot T2STIR imaging is only 33% that of T2 mapping. However, the bSSFP readout is prone to banding artifact caused by field inhomogeneity, especially at 3 T and beyond. The banding artifact may be avoided or moved out of the ROI by using local shimming and frequency scout . Alternatively, spoiled gradient readout can be used instead of bSSFP readout in single-shot T2STIR imaging if banding artifacts still persists.
While T2STIR-bSSFP shows the highest contrast ratio among the three edema imaging sequences compared and holds great promise for edema imaging in STEMI patients, it is not meant to replace T2 mapping. The latter method can identify global changes in T2 values, such as diffuse myocarditis and Takotsubo cardiomyopathy , for which T2STIR-bSSFP is not applicable. In fact, the two techniques may be used together: the single-shot T2STIR-bSSFP technique can identify edematous regions of the whole heart without requiring breath-holding. The T2 mapping technique can then be performed at the selected location where edema is found. Taking into account the time needed for full T1 recovery between consecutive slices, T2STIR-bSSFP can cover the whole heart (8 slices) in approximately 40 s with no need for breath-hold.
STIR-TSE images are susceptible to coil sensitivity variation. The same is true for T2p-bSSFP, in which the SI of edema in the images is only 25–50%  higher than that of normal myocardium. In T2STIR-bSSFP, the SI of edema is 60–220% higher than of normal myocardium in both phantom and patients without coil sensitivity correction. Coil sensitivity correction is therefore optional, although it helps to further improve image contrast. In T2 mapping, curve fitting removes the effect of coil sensitivity on T2 values, but respiratory motion or imperfect breath-holding may increase the variability of T2 values. Also, the estimated T2 value in T2 mapping may vary with arrhythmia, especially for long T1 tissues (Additional file 1: Table S1).
In this study, the performance of three bSSFP-based sequences was evaluated using the CNR and the contrast ratio. SNR measurement in magnitude images is not accurate due to a few reasons. The use of phased array coils results in spatially non-uniform signals in magnitude images but does not affect the parametric maps. In vivo noise measurement is not easy either. The use of parallel imaging results in spatially varying noise and makes noise measurement nontrivial . Image noise also depends on imaging parameters such as voxel size and readout bandwidth. In this study, noise was measured as the SD of normal myocardium  and was applied to both qualitative images and parametric maps. Because it is difficult to accurately measure SNR in vivo, the contrast ratio is used to complement the CNR in quantifying image contrast. The contrast ratio is independent of noise and imaging parameters used. This metric has been shown elsewhere [40, 41] to be a better indicator than CNR in describing the ability of an imaging technique to differentiate two types of tissues. Additionally, the metric is applicable to both qualitative images and parametric maps.
In addition to the SNR measurement method, the study has other limitations. Though T2STIR-bSSFP images show edema and its positions correctly in all patient cases, its use for quantifying the extent of edema may need further study. Additional file 1 show that when the mean + 2SD was used empirically as the criterion for edema area quantification, errors exist in some cases. While the new technique has better image contrast than other bSSFP techniques compared, the spatial variation of coil sensitivity over the T2STIR-bSSFP images may affect accurate delineation of edematous areas to some extent. In addition, the low contrast between the edema region and the adjacent bright blood pool may make the detection of sub-endocardial edema using T2STIR-bSSFP difficult. A single-shot dark-blood T2STIR-bSSFP may be a good way to address this issue and will be investigated in our future work. Nevertheless, this study is a proof of concept, designed to demonstrate the feasibility and potential of T2STIR-bSSFP. It was not a clinical study. Our future work will evaluate the strength, limitations and the optimal threshold value for the segmentation of edema in T2STIR-bSSFP images, and establish the clinical relevance of T2STIR-bSSFP in a clinical setting through a large patient cohort.
This study proposes T2STIR-bSSFP that exploits the elevation of both T1 and T2 values of edema to increase the contrast between edematous and normal myocardium. The single-shot technique provides a fast and robust method for myocardial edema imaging with improved contrast compared with several other edema imaging techniques at 3 T.
Availability of data and materials
The datasets are available from the corresponding author upon reasonable request.
Adiabatic full passage
Adiabatic half passage
Acute myocardial infarction
Balanced steady state free precession
Cardiovascular magnetic resonance
Institutional review board
Linear flip angle pulses
Late gadolinium enhancement
Modified B1-insensitive rotation pulse
Reverse adiabatic half passage
Region of interest
Specific absorption rate
Spectral attenuated inversion recovery
ST elevation myocardial infarction
Short tau inversion recovery
T2-prepared balanced steady state free precession
T2 prepared inversion with short tau inversion recovery balanced steady state free precession
Turbo spin echo
Eitel I, Friedrich MG. T2-weighted cardiovascular magnetic resonance in acute cardiac disease. J Cardiovasc Magn Reson. 2011;13:13.
Friedrich MG. Myocardial edema--a new clinical entity? Nat Rev Cardiol. 2010;7:292–6.
Abdel-Aty H. Myocardial edema imaging of the area at risk in acute myocardial infarction seeing through water. JACC Cardiovasc Imaging. 2009;2:832–4.
Simonetti OP, Finn JP, White RD, Laub G, Henry DA. "black blood" T2-weighted inversion-recovery MR imaging of the heart. Radiology. 1996;199:49–57.
Abdel-Aty H, Simonetti O, Friedrich MG. T2-weighted cardiovascular magnetic resonance imaging. J Magn Reson Imaging. 2007;26:452–9.
Kim HW, Van Assche L, Jennings RB, Wince WB, Jensen CJ, Rehwald WG, Wendell DC, Bhatti L, Spatz DM, Parker MA, et al. Relationship of T2-weighted MRI myocardial Hyperintensity and the ischemic area-at-risk. Circ Res. 2015;117:254–65.
Lauenstein TC, Sharma P, Hughes T, Heberlein K, Tudorascu D, Martin DR. Evaluation of optimized inversion-recovery fat-suppression techniques for T2-weighted abdominal MR imaging. J Magn Reson Imaging. 2008;27:1448–54.
Kellman P, Aletras AH, Mancini C, McVeigh ER, Arai AE. T2-prepared SSFP improves diagnostic confidence in edema imaging in acute myocardial infarction compared to turbo spin echo. Magn Reson Med. 2007;57:891–7.
Payne AR, Casey M, McClure J, McGeoch R, Murphy A, Woodward R, Saul A, Bi XM, Zuehlsdorff S, Oldroyd KG, et al. Bright-blood T2-weighted MRI has higher diagnostic accuracy than dark-blood short tau inversion recovery MRI for detection of acute myocardial infarction and for assessment of the ischemic area at risk and myocardial salvage. Circ Cardiovasc Imaging. 2011;4:210–9.
Viallon M, Mewton N, Thuny F, Guehring J, O'Donnell T, Stemmer A, Bi X, Rapacchi S, Zuehlsdorff S, Revel D, Croisille P. T2-weighted cardiac MR assessment of the myocardial area-at-risk and salvage area in acute reperfused myocardial infarction: comparison of state-of-the-art dark blood and bright blood T2-weighted sequences. J Magn Reson Imaging. 2012;35:328–39.
Aletras AH, Kellman P, Derbyshire JA, Arai AE. ACUT2E TSE-SSFP: a hybrid method for T2-weighted imaging of edema in the heart. Magn Reson Med. 2008;59:229–35.
Giri S, Chung YC, Merchant A, Mihai G, Rajagopalan S, Raman SV, Simonetti OP. T2 quantification for improved detection of myocardial edema. J Cardiovasc Magn Reson. 2009;11:56.
Giri S, Shah S, Xue H, Chung YC, Pennell ML, Guehring J, Zuehlsdorff S, Raman SV, Simonetti OP. Myocardial T(2) mapping with respiratory navigator and automatic nonrigid motion correction. Magn Reson Med. 2012;68:1570–8.
Baessler B, Schaarschmidt F, Stehning C, Schnackenburg B, Maintz D, Bunck AC. Cardiac T2-mapping using a fast gradient echo spin echo sequence - first in vitro and in vivo experience. J Cardiovasc Magn Reson. 2015;17:67.
Basha TA, Bellm S, Roujol S, Kato S, Nezafat R. Free-breathing slice-interleaved myocardial T2 mapping with slice-selective T2 magnetization preparation. Magn Reson Med. 2016;76:555–65.
van Heeswijk RB, Feliciano H, Bongard C, Bonanno G, Coppo S, Lauriers N, Locca D, Schwitter J, Stuber M. Free-breathing 3 T magnetic resonance T2-mapping of the heart. JACC Cardiovasc Imaging. 2012;5:1231–9.
von Knobelsdorff-Brenkenhoff F, Prothmann M, Dieringer MA, Wassmuth R, Greiser A, Schwenke C, Niendorf T, Schulz-Menger J. Myocardial T1 and T2 mapping at 3 T: reference values, influencing factors and implications. J Cardiovasc Magn Reson. 2013;15:53.
Akcakaya M, Basha TA, Weingartner S, Roujol S, Berg S, Nezafat R. Improved quantitative myocardial T2 mapping: impact of the fitting model. Magn Reson Med. 2014.
Bano W, Feliciano H, Coristine AJ, Stuber M, van Heeswijk RB. On the accuracy and precision of cardiac magnetic resonance T-2 mapping: a high-resolution radial study using adiabatic T-2 preparation at 3T. Magn Reson Med. 2017;77:159–69.
Contrast ratio [https://en.wikipedia.org/wiki/Contrast_ratio].
Ugander M, Bagi PS, Oki AJ, Chen B, Hsu LY, Aletras AH, Shah S, Greiser A, Kellman P, Arai AE. Myocardial edema as detected by pre-contrast T1 and T2 CMR delineates area at risk associated with acute myocardial infarction. JACC Cardiovasc Imaging. 2012;5:596–603.
Bulluck H, White SK, Rosmini S, Bhuva A, Treibel TA, Fontana M, Abdel-Gadir A, Herrey A, Manisty C, Wan SM, et al. T1 mapping and T2 mapping at 3T for quantifying the area-at-risk in reperfused STEMI patients. J Cardiovasc Magn Reson. 2015;17:73.
Kumar A, Beohar N, Arumana JM, Larose E, Li D, Friedrich MG, Dharmakumar R. CMR imaging of edema in myocardial infarction using cine balanced steady-state free precession. JACC Cardiovasc Imaging. 2011;4:1265–73.
Nordlund D, Klug G, Heiberg E, Koul S, Larsen TH, Hoffmann P, Metzler B, Erlinge D, Atar D, Aletras AH, et al. Multi-vendor, multicentre comparison of contrast-enhanced SSFP and T2-STIR CMR for determining myocardium at risk in ST-elevation myocardial infarction. Eur Heart J Cardiovasc Imaging. 2016;17:744–53.
Zhu Y, Yang D, Chen Y, Liang D, Liu X, Chung Y-C: Myocardial Edema Imaging using Single-shot T2STIR Prepared bSSFP. In Proceedings of the 25th Annual Meeting of ISMRM, ; Honolulu, HI, USA. 2017: p.3253.
Rakow-Penner R, Daniel B, Yu H, Sawyer-Glover A, Glover GH. Relaxation times of breast tissue at 1.5T and 3T measured using IDEAL. J Magn Reson Imaging. 2006;23:87–91.
Nezafat R, Ouwerkerk R, Derbyshire AJ, Stuber M, McVeigh ER. Spectrally selective B1-insensitive T2 magnetization preparation sequence. Magn Reson Med. 2009;61:1326–35.
Jenista ER, Rehwald WG, Chen EL, Kim HW, Klem I, Parker MA, Kim RJ. Motion and flow insensitive adiabatic T2 -preparation module for cardiac MR imaging at 3 tesla. Magn Reson Med. 2013;70:1360–8.
Deshpande VS, Chung YC, Zhang Q, Shea SM, Li D. Reduction of transient signal oscillations in true-FISP using a linear flip angle series magnetization preparation. Magn Reson Med. 2003;49:151–7.
Kellman P, Wilson JR, Xue H, Ugander M, Arai AE. Extracellular volume fraction mapping in the myocardium, part 1: evaluation of an automated method. J Cardiovasc Magn Reson. 2012;14:63.
Kali A, Choi EY, Sharif B, Kim YJ, Bi X, Spottiswoode B, Cokic I, Yang HJ, Tighiouart M, Conte AH, et al. Native T1 mapping by 3-T CMR imaging for characterization of chronic myocardial infarctions. JACC Cardiovasc Imaging. 2015;8:1019–30.
Basha TA, Tang MC, Tsao C, Tschabrunn CM, Anter E, Manning WJ, Nezafat R. Improved dark blood late gadolinium enhancement (DB-LGE) imaging using an optimized joint inversion preparation and T2 magnetization preparation. Magn Reson Med. 2018;79:351–60.
Kellman P, Xue H, Olivieri LJ, Cross RR, Grant EK, Fontana M, Ugander M, Moon JC, Hansen MS. Dark blood late enhancement imaging. J Cardiovasc Magn Reson. 2016;18:77.
Wong EC, Liu TT, Luh WM, Frank LR, Buxton RB. T(1) and T(2) selective method for improved SNR in CSF-attenuated imaging: T(2)-FLAIR. Magn Reson Med. 2001;45:529–32.
Liu CY, Wieben O, Brittain JH, Reeder SB. Improved delayed enhanced myocardial imaging with T2-prep inversion recovery magnetization preparation. J Magn Reson Imaging. 2008;28:1280–6.
Liu CY, Bley TA, Wieben O, Brittain JH, Reeder SB. Flow-independent T(2)-prepared inversion recovery black-blood MR imaging. J Magn Reson Imaging. 2010;31:248–54.
Deshpande VS, Shea SM, Li D. Artifact reduction in true-FISP imaging of the coronary arteries by adjusting imaging frequency. Magn Reson Med. 2003;49:803–9.
Thavendiranathan P, Walls M, Giri S, Verhaert D, Rajagopalan S, Moore S, Simonetti OP, Raman SV. Improved detection of myocardial involvement in acute inflammatory cardiomyopathies using T2 mapping. Circ Cardiovasc Imaging. 2012;5:102–10.
Kellman P, McVeigh ER. Image reconstruction in SNR units: a general method for SNR measurement. Magn Reson Med. 2005;54:1439–47.
Simonetti OP, Kim RJ, Fieno DS, Hillenbrand HB, Wu E, Bundy JM, Finn JP, Judd RM. An improved MR imaging technique for the visualization of myocardial infarction. Radiology. 2001;218:215–23.
Ferreira VM, Piechnik SK, Dall'Armellina E, Karamitsos TD, Francis JM, Choudhury RP, Friedrich MG, Robson MD, Neubauer S. Non-contrast T1-mapping detects acute myocardial edema with high diagnostic accuracy: a comparison to T2-weighted cardiovascular magnetic resonance. J Cardiovasc Magn Reson. 2012;14:42.
The authors would like to thank Dr. Raymond Kwong from Brigham and Women’s Hospital for his insightful comments on the manuscript.
This work was supported in part by the National Natural Science Foundation of China (No. 61771463, 81971611, 81830056) and the National Key Research and Development Program of China (No. 2017YFC0112903).
Ethics approval and consent to participate
The healthy subject study was approved by the institutional review board (IRB) at Shenzhen Institutes of Advanced Technology, and the patient study was approved by the IRB of West China Hospital.
Consent for publication
Written informed consent was obtained from all subjects for publication of their individual details and accompanying images in this manuscript.
Dr. Yiu-Cho Chung is an employee of Siemens Healthcare, Singapore. The other authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Zhu, Y., Yang, D., Zou, L. et al. T2STIR preparation for single-shot cardiovascular magnetic resonance myocardial edema imaging. J Cardiovasc Magn Reson 21, 72 (2019) doi:10.1186/s12968-019-0583-y
- Single-shot imaging
- Short tau inversion recovery