Four-dimensional flow cardiovascular magnetic resonance in tetralogy of Fallot: a systematic review
Journal of Cardiovascular Magnetic Resonance volume 23, Article number: 59 (2021)
Patients with repaired Tetralogy of Fallot (rTOF) often develop cardiovascular dysfunction and require regular imaging to evaluate deterioration and time interventions such as pulmonary valve replacement. Four-dimensional flow cardiovascular magnetic resonance (4D flow CMR) enables detailed assessment of flow characteristics in all chambers and great vessels. We performed a systematic review of intra-cardiac 4D flow applications in rTOF patients, to examine clinical utility and highlight optimal methods for evaluating rTOF patients.
A comprehensive literature search was undertaken in March 2020 on Google Scholar and Scopus. A modified version of the Critical Appraisal Skills Programme (CASP) tool was used to assess and score the applicability of each study. Important clinical outcomes were assessed including similarities and differences.
Of the 635 articles identified, 26 studies met eligibility for systematic review. None of these were below 59% applicability on the modified CASP score. Studies could be broadly classified into four groups: (i) pilot studies, (ii) development of new acquisition methods, (iii) validation and (vi) identification of novel flow features. Quantitative comparison with other modalities included 2D phase contrast CMR (13 studies) and echocardiography (4 studies). The 4D flow study applications included stroke volume (18/26;69%), regurgitant fraction (16/26;62%), relative branch pulmonary artery flow(4/26;15%), systolic peak velocity (9/26;35%), systemic/pulmonary total flow ratio (6/26;23%), end diastolic and end systolic volume (5/26;19%), kinetic energy (5/26;19%) and vorticity (2/26;8%).
4D flow CMR shows potential in rTOF assessment, particularly in retrospective valve tracking for flow evaluation, velocity profiling, intra-cardiac kinetic energy quantification, and vortex visualization. Protocols should be targeted to pathology. Prospective, randomized, multi-centered studies are required to validate these new characteristics and establish their clinical use.
Tetralogy of Fallot (TOF) is a common serious form of congenital heart disease (CHD) and one of the first to be formally described historically [1,2,3,4]. The four main morphological features are (i) ventricular septal defect (VSD), (ii) right ventricular (RV) outflow tract obstruction, (iii) overriding aorta and iv) RV hypertrophy. Primary repair includes closure of the VSD, resection of infundibular muscle obstruction and relief of pulmonary stenosis. Additional repair of a stenosed pulmonary artery may be performed using a patch reconstruction. Residual anatomic and hemodynamic abnormalities in repaired TOF (rTOF) patients are highly prevalent, with pulmonary stenosis and pulmonary regurgitation (PR) being common. Chronic RV volume overload, akinesis or dyskinesis of the RV outflow tract wall, and a nearly universal right bundle branch block, trigger a sequence of pathophysiologic sequelae that lead to RV dilatation, and ultimately dysfunction and right sided heart failure [5, 6]. The long-term prognosis for TOF patients has improved in the past 80 years, however life expectancy is still lower in comparison with age-matched controls [2, 7]. A growing number of patients need continuous monitoring to determine whether further intervention, such as pulmonary valve replacement, is necessary [5, 7].
Longitudinal monitoring of cardiac output, RV volumes and pulmonic valve regurgitant fraction is typically performed using echocardiography and cardiovascular magnetic resonance (CMR). Although echocardiography is widely available and often used as a reference standard for blood flow velocity analysis , CMR has high signal-to-noise ratio and enables coverage of all anatomical regions [9,10,11]. Quantitative blood flow analysis using phase-contrast CMR of velocity (PC-CMR) enables precise quantification of velocity in the phase of each pixel of the CMR image. Four-dimensional PC-CMR (4D flow) enables evaluation of flow in multiple vessels as well as within the heart chambers in complex CHD [12,13,14].
Recently, a number of studies have reported 4D flow CMR in cardiac applications, including a systematic review  and a consensus statement . We aimed to provide a systematic review of 4D flow applications in rTOF patients, to highlight clinical utility of 4D flow in rTOF, review validation data, compare acquisition parameters, and identify the differences and similarities between studies. Recommended sequences and techniques are highlighted for specific applications in rTOF patients, with the aim of encouraging future multi-center studies and more standardization in clinical protocols.
The Preferred Reporting Items for Systematic Reviews and Meta-Analysis (PRISMA) checklist was adhered to when structuring this article . A comprehensive search was undertaken in March 2020 on Google Scholar and Scopus (this database enables a complete search of both MEDLINE and EMBASE). The search limitations included ‘Humans’ and ‘English Language’, with no time limitations. The search scope included the reference lists of included articles, citation tracking and manual reference searching.
The key words of the search were broken down to several searches to cover all the possible variations of the modality and the pathology (Tetralogy of Fallot), for example: (4D) OR (four-dimensional) AND (flow) OR [(3D) OR (three-dimensional)) AND ((cine) OR (time-resolved)) AND (PC) OR (phase contrast)] AND (CMR) (cardiovascular magnetic resonance) OR (MRI) OR (magnetic resonance imaging).
Article screening and eligibility criteria
The study selection process was performed by a single reviewer (AE) and reviewed independently (KG, MS) before a final review (AY, BC, KP). Disagreement in the articles were all resolved with the revision process. Once duplicates were removed, the titles and abstracts of the search results were assessed using a screening algorithm based on eligibility criteria as shown in Fig. 1. The studies that adhered to this screening had their full texts evaluated. Further exclusion was done based on the methodologies and study groups where these were not detailed in the abstracts.
Characteristics of studies, authors, year, and participants were noted, if available. The study type and aim were determined, and classified into pilot, diagnostic and mechanistic studies. Study methodology details, including the study design, outcomes and conclusions, were evaluated in particular for the 4D flow protocol and analysis. A summary is presented in Table 1.
The quality of the included studies was assessed by AE using a modified Critical Appraisal Skills Programme (CASP) tool, provided in Table 2. The questions were modified and set criteria designed for the purpose of estimation of the applicability of the studies selected, rather than criticism of the proposed methods. The presence or absence of criteria are not relevant to the strength of the presented work but to the completeness for reapplication of the same methodologies to similar study groups.
Answers of ‘yes’ scored 1 point, whereas answers of ‘no’ or ‘indeterminate’ scored 0 points. Total scores were converted to percentages and studies were allocated to one of three categories; ‘highly clinical applicable’ for a score of 67–100%, ‘potentially clinically applicable’ for 34–66% and ‘less clinically applicable’ for 0–33%.
A generalized meta-analysis was not possible for this systematic review as much of the research is exploratory and preliminary, with considerable heterogeneity in the study outcomes. However, analysis of the similarities and differences was performed. A narrative review is provided.
A search of the electronic databases produced 635 articles. After removing duplication, 570 articles remained. Based on the eligibility criteria, title and/or abstract were initially screened, and 30 articles remained which met the initial search criteria as potentially relevant to the current study. After reading the full-text articles, 4 articles were excluded. Therefore, the final selection of studies included 26 articles. A summary of the process is presented in Fig. 1.
Description of the included studies
Twenty-six studies were included in this systematic review, the details are described in Table 1. None were below 59% applicability thus no studies had low applicability. Of the 26 studies, 6 scored above 80%, 6 studies scored 70–80% and the rest were between 59 and 70%. Studies that had been performed more recently yielded higher scores, reflecting a more targeted application of the technique.
Aim of the studies
Studies could be broadly classified into:
Identification of novel flow features such as vortices in the right side of the heart in rTOF patients compared with normal populations [12, 19] and the effects of different surgeries on vortices in the heart  as early prognostic factors in ventricular changes and valvular malfunction.
Imaging validation methods
Fourteen of 26 (54%) studies had a quantitative comparison with other modalities. All these studies concluded that 4D flow results demonstrated a high agreement via several statistical tests and no significant differences in flow except in a few cases, with rational explanations for these results. Four studies involved comparison with echocardiography. Echocardiography has high availability and good temporal resolution; however, flow quantification is limited by the geometry and angulation of the transducer as well as the availability of acoustic windows . In a study that focused on the tricuspid valve, 38.5% of patients were classified differently by at least one grade using quantitative 4D flow CMR when compared to the echocardiographic assessment .
Several studies compared 4D flow with 2D PC-CMR, since 2D acquisition is historically well validated and standardized [9,10,11]. However, 2D slices need to be accurately localized and the results are often inconsistent (e.g. the flows in the left and right pulmonary branches may not add up to the flow in the main pulmonary artery) due to separate breath-hold acquisitions. An alternative comparison can be made between stroke volume integrated from ventricular outflow and volumetric stroke volume calculated from anatomical cine balanced steady state free precession (bSSFP) planimetry.
Several studies compared the patient population against a normal population that had no known history of cardiovascular disease. In earlier studies there was a significant difference in age between the groups (Fig. 2), with older volunteers. However later studies had more age-matched study populations [26, 31,32,33,34,35] (Fig. 2).
Quantified parameters included arterial or venous flow such as systolic peak velocities, or stroke volume from net flow across outflow valves, or evaluation of the valvular efficiency through regurgitant fraction (Table 1). Qualitative analysis of the flow and vorticity in different areas of the heart and great vessels were also evaluated [12, 18, 19, 35]. In addition to the measurements of basic flow volumes and velocities, the estimation of derived hemodynamic biomarkers such as wall shear forces and a resistance index  and ventricular kinetic energy measurements [31, 36, 37] have been proposed. Ventricular kinetic energy (KE) and its applications have been more recently investigated [33, 34, 38]. In some studies, end-diastolic (EDV) and end-systolic volumes (ESV) were quantified from the 4D flow data. The following is a summary of the main findings with respect to these parameters: In total there were 18/26 (69%) studies on stroke volume, 16/26 (62%) on regurgitant fraction, 4/26 (15%) on right versus left pulmonary artery flow, 9/26 (35%) on systolic peak velocity, 6/26 (23%) on Qp:Qs, 5/26 (19%) on EDV and ESV, 5/26 (19%) on KE, and 4/26 (15.3%) on vorticity. Commonly examined flow parameters are illustrated in Fig. 3.
Stroke volume was quantified as volume of effective antegrade flow per heartbeat, i.e. the difference between forward and backward flow volume above the aortic or pulmonary valve [11, 13] in the absence of regurgitant volumes. Good agreement between 2 and 4D was found in several studies. The average mean of differences represented < 1% of the overall mean value for arterial and < 3% for venous stroke volume [9, 11, 26, 27]. Other studies found good internal consistency through indirect measurement of cardiac output using stroke volume from cine slice planimetry. Nevertheless, some studies reported some significant bias, for example Giese et al . reported an underestimation of stroke volume by 2.5 ± 8.4 ml with 4D flow corresponding to 5.6 ± 14.9% with respect to the stroke volumes derived from the 2D flow datasets.
Internal validation of stroke volumes at different levels of the same vessel at different levels in the ascending aorta and pulmonary trunk did not show significant differences . However, there was a significantly higher variation in patients compared to healthy subjects, which may be due to patients having complex flow patterns leading to signal loss in the presence of some turbulent flows .
No statistically significant difference was observed between 4D flow (magnitude images) and bSSFP measurements of EDV, ESV and stroke volume in some studies using contrast [8, 23, 27, 36]. However, Hanneman et al.  observed that RVEDV and ESV were underestimated by 4D flow compared to bSSFP and attributed to differences in basal slice selection. Conversely, they found no significant difference in left ventricular (LV) or RV mass quantification with 4D flow compared to bSSFP acquisitions in the same study.
Regurgitant fraction is the diastolic reversed flow expressed as a percentage of forward flow or the percentage of backward flow volume into the corresponding ventricle during diastole [13, 39]. Regurgitant fraction is an important parameter in rTOF patients because of common pulmonary valve insufficiency . Due to the presence of vortices in the pulmonary artery, antegrade and retrograde flows can occur simultaneously in the same cardiac phase. Therefore, PR fraction computed using pixel-wise integration can be greater than that calculated by average velocity integration . PR volume is also important and may reflect the severity of regurgitation better than regurgitant fraction .
Several studies reported no significant differences between 2 and 4D flow CMR for both antegrade and retrograde flow [11, 21, 25]. However, van der Hulst et al.  observed significant differences in pulmonary valve backward flow volumes between 2D PC-CMR and 4D flow but no significant differences between the 4D flow and planimetric estimates (RV stroke volume minus LV stroke volume). On the other hand, a significant difference was seen between patient and normal populations . Chelu et al. reported that 4D flow had a sensitivity of 83% (95%CI: 36–100%), a specificity of 98% (95% CI: 88–100%), and an accuracy of 96% to identify patients with RF > 20% . Four studies observed ejection fractions concluding good correlations between 2D flow and 4D flow [12, 29, 42, 43].
The tricuspid valve is also of importance when assessing right sided CHDs. Tricuspid regurgitation (TR) is independently associated with both increased morbidity and increased mortality in CHD patients . The degree of TR is an important diagnostic parameter in the decision making of valve replacement in rTOF patients . The tricuspid valve has range of motion which is increased in the setting of severe TR  and quantification of the regurgitation fraction is difficult due to the increased in-plane motion of the valve annulus. TR volume can also be derived from the difference in planimetric RV stroke volume and direct flow measurement across the pulmonic valve .
Altering geometry and valve structure changes flow patterns, increase the difficulty of TR assessment. Some studies have validated tricuspid flow measurement by 4D flow in patients without TR  and patients with TR and complex RV geometry [23, 26]. Both studies demonstrated that quantifying tricuspid valve flow using retrospective valve tracking was possible in 100% of the imaged patients. Effective flow was reported to be significantly overestimated by 2D flow compared to both other methods, and regurgitant fraction was significantly underestimated by 2D flow in both studies. In one study 39% of TR patients were classified differently by at least 1 grade using 4D flow compared to echocardiography . Strong intra /inter-observer agreement was also demonstrated with 4D flow ICC > 0.91. Earlier studies had reported an agreement with an ICC = 0.93 to 0.94 [45, 46]. The difference between 4D flow and 2D PC-CMR flow was also attributed to motion of the valve annulus through the plane of the image with 2D, and the breathing artifact if 4D was obtained during free breathing, and finally interscan variability was also considered [23, 26].
Flow and peak velocity
Aortic and pulmonic flow rates showed good agreement between 4 and 2D flow using Bland–Altman evaluation in most of the studies. One study compared several 4D flow segmentation approaches with 2D, showing differences in flow rate of 1–2% and 0.3–0.4 L/min or 6–8% in cardiac output . The results of this study group are similar to those of a study on healthy subjects, where estimated stroke volumes were on average 3% greater with 4D flow than with 2D PC-CMR . Nevertheless, a 12% difference with a higher 2D estimate was also observed . Chelu et al.  reported that 4D flow underestimated peak systolic velocities, whereas Nordmeyer et al.  reported the opposite, with 10% higher peak velocities with 4D flow compared to 2D PC-CMR at the level of highest velocity. The same studies also noted better correlation with echocardiography with 4D flow than 2D PC-CMR. This was also observed by Gabbour et al.  who found that 2D PC-MR significantly underestimated aortic and main pulmonary artery peak systolic velocities compared to volumetric 4D flow and echo. Their study suggested that these findings were due to the 4D flow assessment of the entire vessel volume.
It has been suggested that the conversion of rotational flow into helical flow in the right atrium may be a method of conserving atrial KE during RV filling . This is supported by studies that observed a clockwise right atrium vortex in all their healthy subjects [12, 19]. A right handed helical flow was also observed between the right atrium and the RV during tricuspid valve diastolic filling, which was suggested to have an influence on the KE used by the atrium [31, 36], also described as an inflow jet with an adjacent vortex formation [12, 49]. rTOF patients showed abnormal timing of right atrium filling, more in diastole than in systole  with additional anticlockwise vortices .
Following the vortex observed in the tricuspid valve, parts of the intraventricular flow were instantaneously directed towards the RV outflow tract during diastole to be followed by slight helical flow in the RV outflow tract during systole . Vortices and changes in flow were minimal at the apex. In rTOF patients flow features were very heterogeneous in direction, with large PR jets also directed toward the RV apex, and increased number of vortices [12, 19].
The pulmonary artery and its branches (right and left pulmonary arteries)
Whereas flow in healthy subjects is relatively uniform, rTOF patients exhibit asymmetric flow distribution in the pulmonary arteries, severe regurgitation and pronounced vortices  consisting of increased helical and vertical flow patterns. High peak velocities > 1.5 m/s have been observed in patient populations but not in normal control groups [18, 19].
Superior/inferior vena cava
Two studies observed flow in the venae cavae. The flow from the superior vena cava and inferior vena cava into the right atrium was considered normal when greater flow occurred during systole than diastole. This occurred in almost all normal subjects (although it was reversed in 10%, attributed to intrathoracic and intraperitoneal pressure fluctuations) and ~ 20% of subjects with rTOF [18, 19].
Pulse sequence parameters
Table 2 shows a comparison of sequence parameters between the studies included in this review. Some parameters were very similar between studies whereas others were adjusted to optimize the visualization of the structure of interest. The following sections summarize the main differences. Of note is that 16 studies used contrast to provide enhanced signal-to-noise and higher acceleration . Most of these studies used gadolinium, although 2 used ferumoxytol (Table 2).
The types of CMRI units were variable with 16 studies using 1.5 T, 5 studies using 3 T, and 6 studies incorporating both field strengths. No studies compared 1.5 vs 3 T in accuracy or visualization of flow parameters.
The acceleration type and factor and scan duration
The clinical applicability of 4D flow is aided by shortening the scan duration . Improvements of reconstruction algorithms dedicated to phase-contrast imaging enable acceleration factors greater than 5. Parallel imaging acceleration was common with the shortest scan duration being 3.5 min . Parallel imaging compressed sensing (PICS), echo planar imaging (EPI) and isotropic voxel radial projection imaging (VIPR) were often used in additional to conventional parallel imaging methods (SENSE and GRAPPA). The average acceleration factor was 2 (in-plane acceleration factor). The average scan time across studies was 11 min, although this was highly dependent on heart rate.
Use of respiratory gating with either navigator or bellows was mixed, with 14 studies not using respiratory gating. Several studies examined non-gated 4D flow CMR against other methods, suggesting good agreement with conventional breath-hold 2D and respiratory gated 4D flow CMR methods [11, 50]. Respiratory compensation with k-space phase reordering (EXORCIST, General Electric Healthcare, Milwaukee, Wisconsin, USA), was used in some studies [9, 10]. Adaptive diaphragm navigator gating was used by others although the acceptance window was typically not reported (with the exception of 3–5 mm in ), and several studies used respiratory bellows [19, 37].
Most studies used retrospective cardiac gating in the acquisition. Nevertheless, three studies conducted prospective gating [8, 12, 18], and vector electrocardiogram (ECG) gating was used by several studies [9, 10, 20, 28]. One study implemented a semi-retrospective gating .
Velocity encoding (VENC)
VENC is defined as the (positive or negative) maximum velocity that can be detected without phase wrap. Prior approximate knowledge of peak velocity expected in the vessel of interest is thus essential. If the VENC is too low, either the scan must be repeated or antialiasing correction needs to be performed. Decreasing VENC corresponds to increasing the strengths and duration of the velocity encoding gradients, leading to longer echo and repetition times [16, 51]. However, VENC should be set as low as possible to achieve optimal phase signal-to-noise ratio. Most studies used 150 cm/s (range 40–500). Lower VENC is appropriate for intraventricular vortices  or KE  quantification. The highest VENC was 500 cm/s which may be necessary for peak systolic velocities in jets [27, 42]. Anti-aliasing pre-processing was performed by some studies to ameliorate this issue ; however, post processing antialiasing algorithms are becoming more common.
Clinical application of 4D flow in rTOF
This systematic review emphasized the applicability of the 4D flow CMR in rTOF clinical evaluations. Velocity and flow volume measurements are required for evaluating severity of disease, and have been validated against 2D PC-CMR [8, 10, 11, 23,24,25,26,27, 30, 53] and echocardiography [8, 23, 27, 28]. Visualization of anatomy and classification of pulmonary blood supply is necessary to plan appropriate surgical management in many forms of CHD . Applicability was highly promising, especially where studies addressed the diagnostic needs of more complex cases requiring quantitative flow measurements on both the venous and arterial sides [10, 27]. Accuracy was internally validated through employing the ‘conservation of mass’ principle, comparing volumes that are expected to be equal in absence of valvular malfunction or shunts . This was done by comparison of input venous return and output arterial volumes , the comparison between the sum of branches to the main pulmonary artery , as well as the systemic versus the pulmonary circulation [10, 22]. Superiority of 4D flow CMR in rTOF was evident through higher correlations and lower mean differences than 2D PC-CMR. This is in agreement with studies in other patient groups [55, 56], which also found greater inconsistency in 2D PC-CMR, likely due to changing output during different breath-holds [25, 57].
Visualization of the pulmonary arteries is generally difficult to evaluate using echocardiography, due to acoustic-window limitations, but was feasible and recommended by 4D flow . Further, mean velocities and net flow calculations with echo is dependent on flow profile and vessel cross-sectional area, which may be less accurate with complex arterial flow and distorted anatomy . Complex flow patterns are particularly relevant in patients with dilated main pulmonary arteries, as the accuracy of flow measurements is compromised in the presence of focal turbulence, through-plane motion of stenotic semilunar valves, or the postoperative distorted anatomy itself .
Quantification of PR holds particular importance in the timing of pulmonary valve replacement . 4D flow is recommended over multiple 2D PC-CMR acquisitions for evaluating residual pulmonary stenosis, branch artery stenosis, and PR in candidates for pulmonary valve replacement [12, 18].
Retrospective valve tracking for valvular flow quantification using 4D flow CMR had high clinical applicability in studies that covered the whole heart to investigate several pathologies . For example, flow measurements have been recommended to be performed not only at the pulmonary but also at the aortic valve [28, 58] in the rTOF group. Early TOF repair may cause a decrease in aortic compliance and increase in wall shear stress which would increase the risk of late aortic complications in the rTOF adult . Tricuspid valve  flow visualization and anatomical details were also found to be essential to improve preoperative selection of the type of procedure, for example patients that would benefit from tricuspid valve annuloplasty [59, 60]. 4D flow has better spatial coverage in all dimensions enabling visualization of flow jets through stenotic valves . However, in children, high steady-state heart rate and respiration frequency  potentially influence flow measurement accuracy .
Interobserver agreement were moderate to high in 4D flow and bSSFP estimates of ventricular volume, ejection fraction, and mass, with no significant differences in agreement between techniques found. However, agreement with 4D flow CMR was higher for the LV than the RV, likely due to the complexity of RV especially in CHD patients [12, 29].
4D flow can reduce the total time of CMR, with an average scan time across studies of 12 min (range from 4 to 20 according to the acceleration type). This has reduced the duration of anesthesia required for younger pediatric patients  or patients with decompensated heart failure .
Novel quantitative and qualitative parameters
4D flow enables greater understanding of pathophysiology and several studies in this systematic review have examined new diagnostic and prognostic flow parameters . These augment conventional measures of velocities and regurgitant fractions with new measures of flow characteristics and energetics, which could aid prediction of future outcomes. Qualitative assessment of blood flow patterns is a distinctive feature of 4D flow and further study is required to relate these measures with outcomes [12, 18]. Studies have covered a wide range of pathology, from rTOF pediatric and adult cohorts to adult and pediatric healthy subjects. Vortex flow has been corelated with pulmonary hypertension , and turbulent flow may contribute to the development of RV remodeling, RV outflow tract fibrosis, and other complications seen in patients with rTOF . Furthermore, surgical outcomes may be additionally affected by postoperative vessel dilation due to vortices, requiring standardization and automated quantification . Flow vortices may provide independent prediction of surgical outcome [18, 19, 63, 64] and further research on automated vortex extraction is required [15, 65].
Diastolic RV function and biventricular hemodynamic forces may be a useful prognostic tool . A restrictive right ventricular filling pattern has been correlated postoperatively with slower recovery after repair . With 4D flow CMR, summation of pulmonary valve flow and tricuspid valve flow enables assessment of RV diastolic function in the presence of PR . Altered flow hemodynamic forces have also been suggested as a feature of mechanical myocardial dyssynchrony in heart failure, cardiomyopathies, and TOF .
Other studies observed ventricular KE as a promising non-invasive early indicator of ventricular efficiency [31, 33, 34, 36, 37]. A correlation between pathologically turbulent flow and abnormal vascular remodeling  is consistent with greater energy loss in rTOF with increased ventricular remodeling [30, 36]. Turbulent kinetic energy patterns and anatomical location visualization in the RV in patients with more severe PR suggest that these reflect the severity of PR . RV KE was also found to be higher in rTOF than healthy subjects  indicating that more work needs to be done by the RV in rTOF to generate the same cardiac output as in normal subjects . Future studies are needed to determine the predictive power of KE measurements for RV dysfunction and thus the timing of reintervention .
Pulse sequence parameter recommendations based on the systemic review outcome for rTOF patient scan
These recommendations are based on the results of the systematic review, in comparison with the 2015 SCMR 4D flow consensus statement . Although parameters change according to the different clinical indications, future multi-center clinical studies would benefit from standardization of pulse sequence parameters.
Respiratory and cardiac gating
Retrospective ECG synchronization had been recommended  to avoid overlooking data in the last 10–20% of the cardiac cycle. Prospective gating can be used when quantifying parameters such as peak systolic velocity or net flow which are low at the end of the cycle . This was supported by high correlation between 2D PC-CMR and 4D flow CMR results. However, regurgitant fraction had lower Bland Altman agreement, supporting the recommendation of retrospective ECG gating. Respiratory gating on the other hand, although recommended , was not done by more than half the studies included. Free-breathing 4D flow CMR without respiratory gating decreases the scan time and in some cases had had negligible effect on accuracy . However, future studies should endeavour to apply both retrospective ECG gating and some form of respiratory gating or motion correction.
4D flow utilizes bright blood spoiled gradient echo sequences, enabling the generation of PC angiograms without the need for an external contrast agent. This allows application to patients that are either contraindicated or are at risk . More than half the included rTOF studies used gadolinium, often because it was required for other reasons. T1 shortening contrast agents enable larger flip angles and reduced signal loss due to saturation in 4D flow acquisitions, and are commonly available in centers . This agrees with the consensus statement ; however, timing of administration and type used may cause variable outcomes.
Velocity encoding (VENC)
Standard 4D flow acquisitions have required a single VENC adjusted to the highest expected velocity within the chosen 4D volume . This prevents aliasing but poses a potential inaccuracy for quantification of low velocities in venous or intra-cardiac flows. For rTOF studies a VENC of 150 cm/s was a common compromise. Although some studies found an average error of less than 3% of measured flow volumes , VENC may need to be customized according to the needs of each case . Dual VENC sequences have been proposed to avoid aliasing in fast velocities and retain high signal to noise ratio in low flow regions . Recently, a triple VENC  method has been developed. The recommended VENC of 150 cm/s has been used by studies that evaluated vortices [12, 18]. However, most studies recorded a range and it was not always clear what VENC was used for what purpose. It is recommended that future studies report these details for ease of replication, with a minimum of one VENC set at 150 cm/s unless otherwise indicated.
Temporal and special resolutions
It has been recommended that at least 5–6 voxels across the vessel lumen are essential for quantitative volume accuracy, and 2.5 × 2.5 × 2.5 mm3 was recommended for the aorta or pulmonary artery in the consensus statement . However, the reviewed studies had a tendency to be lower (0.8 mm at lowest) probably to have better visualization of branch pulmonaries or veins. Also the echo time should be minimized to reduce signal loss due to incoherent flow patterns as shorter echo times have been suggested to further improve precision of flow measurements .
Temporal resolution < 40 ms  with a maximum voxel size of 2.5 × 2.5 × 2.5 mm3 is recommended, which was in agreement with most studies included. Higher temporal resolution imaging techniques were suggested to improve future studies that aim to further clarify fluid-tissue interaction relationships .
Eddy-current correction and Maxwell correction techniques should be employed to reduce background phase-offsets [9, 29, 43]. Phase unwrapping also affects the accuracy of velocity measurements at peaks and should be employed. This is in agreement with the consensus statement  and was employed by all the studies reviewed.
A variety of software packages were used across studies; some in-house tools have been developed [28, 42] but more recent studies typically used commercially available packages [24, 25] that are either provided on the scanner or on post-processing workstations, utilizing the DICOM format. Consistency between vendors is highly recommended but currently not well characterized. Ideally, open benchmark datasets should be used to evaluate consistency between software solutions.
4D flow CMR is a rapidly evolving field, and current study designs have considerable methodological heterogeneity. Until recently, most software used in-house code which was not commercially available. Closed source analysis software makes it difficult to evaluate the accuracy of results or the details of methodology. Acquisition durations are still lengthy, and the use of contrast is common, although advances in machine learning show promise in enhancing 4D flow data [69, 70]. Almost half the number of studies had missing information that would impact the reproducibility of the work, e.g. the number of phases was not reported in 12/26 studies. There was also a lack of clinically relevant endpoints correlated with 4D flow. Future research is required to develop multicenter studies. The nature of systematic reviews only allows use of existing literature, and biases and limitations of included studies affects the reliability of the review.
4D flow is highly applicable to rTOF patients in a single free-breathing 10–15 min acquisition. Particular strengths are retrospective valve tracking, velocity profiling, and volumetric quantification. Novel parameters such as KE and vortex quantification offer new insights into mechanisms of disease. Prospective, randomized, multi-centered studies are required to investigate the application of these methods in patient management.
Availability of data and materials
This research has been conducted using published studies.
Balanced steady state free precession
Critical Appraisal Skills Programme
Congenital heart disease
Cardiovascular magnetic resonance
Left ventricle/left ventricular
Parallel imaging compressed sensing
Repaired tetralogy of Fallot
Right ventricle/right ventricular
Tetralogy of Fallot
Isotropic voxel radial projection imaging
Ventricular septal defect
Restivo A, Anderson RH, Carletti R, Di Gioia CRT. Correlating the morphological features of tetralogy of Fallot and the Eisenmenger malformation. Cardiol Young. 2017;27(1):161–72. https://doi.org/10.1017/S1047951116000287.
Nollert G, Fischlein T, Bouterwek S, Böhmer C, Klinner W, Reichart B. Long-term survival in patients with repair of tetralogy of Fallot: 36-year follow-up of 490 survivors of the first year after surgical repair. J Am Coll Cardiol. 1997;30(5):1374–84. https://doi.org/10.1016/S0735-1097(97)00318-5.
Karl TR, Stocker C. Tetralogy of fallot and its variants. Pediatr Crit Care Med. 2016;17(8):S330–6. https://doi.org/10.1097/PCC.0000000000000831.
Hrusca A, Rachisan AL, Gach P, et al. Detection of pulmonary and coronary artery anomalies in tetralogy of Fallot using non-ECG-gated CT angiography. Diagn Interv Imaging. 2016;97(5):543–8. https://doi.org/10.1016/j.diii.2016.03.010.
Geva T. Repaired tetralogy of Fallot: the roles of cardiovascular magnetic resonance in evaluating pathophysiology and for pulmonary valve replacement decision support. J Cardiovasc Magn Reson. 2011;13(1):9. https://doi.org/10.1186/1532-429X-13-9.
Geva T. Indications for pulmonary valve replacement in repaired tetralogy of fallot: the quest continues. Circulation. 2013. https://doi.org/10.1161/CIRCULATIONAHA.113.005878.
Vaujois L, Gorincour G, Alison M, Déry J, Poirier N, Lapierre C. Imaging of postoperative tetralogy of Fallot repair. Diagn Interv Imaging. 2016;97(5):549–60. https://doi.org/10.1016/j.diii.2016.02.007.
Gabbour M, Schnell S, Jarvis K, Robinson JD, Markl M, Rigsby CK. 4-D flow magnetic resonance imaging: blood flow quantification compared to 2-D phase-contrast magnetic resonance imaging and Doppler echocardiography. Pediatr Radiol. 2015;45(6):804–13. https://doi.org/10.1007/s00247-014-3246-z.
Hsiao A, Lustig M, Alley MT, et al. Rapid pediatric cardiac assessment of flow and ventricular volume with compressed sensing parallel imaging volumetric cine phase-contrast MRI. Am J Roentgenol. 2012;198(3):250–9. https://doi.org/10.2214/AJR.11.6969.
Hsiao A, Alley MT, Massaband P, Herfkens RJ, Chan FP, Vasanawala SS. Improved cardiovascular flow quantification with time-resolved volumetric phase-contrast MRI. Pediatr Radiol. 2011;41(6):711–20. https://doi.org/10.1007/s00247-010-1932-z.
Nordmeyer S, Riesenkampff E, Crelier G, et al. Flow-sensitive four-dimensional cine magnetic resonance imaging for offline blood flow quantification in multiple vessels: a validation study. J Magn Reson Imaging. 2010;32(3):677–83. https://doi.org/10.1002/jmri.22280.
Hirtler D, Garcia J, Barker AJ, Geiger J. Assessment of intracardiac flow and vorticity in the right heart of patients after repair of tetralogy of Fallot by flow-sensitive 4D MRI. Eur Radiol. 2016;26(10):3598–607. https://doi.org/10.1007/s00330-015-4186-1.
Köhler B, Preim U, Grothoff M, Gutberlet M, Fischbach K, Preim B. Robust cardiac function assessment in 4D PC-MRI data of the aorta and pulmonary artery. Comput Graph Forum. 2016;35(1):32–43. https://doi.org/10.1111/cgf.12669.
Kohler B, Gasteiger R, Preim U, Theisel H, Gutberlet M, Preim B. Semi-automatic vortex extraction in 4D PC-MRI cardiac blood flow data using line predicates. IEEE Trans Vis Comput Graph. 2013;19(12):2773–82. https://doi.org/10.1109/TVCG.2013.189.
Crandon S, Elbaz MSM, Westenberg JJM, van der Geest RJ, Plein S, Garg P. Clinical applications of intra-cardiac four-dimensional flow cardiovascular magnetic resonance: a systematic review. Int J Cardiol. 2017;249:486–93. https://doi.org/10.1016/j.ijcard.2017.07.023.
Dyverfeldt P, Bissell M, Barker AJ, et al. 4D flow cardiovascular magnetic resonance consensus statement. J Cardiovasc Magn Reson. 2015;17(1):72. https://doi.org/10.1186/s12968-015-0174-5.
Moher D, Liberati A, Tetzlaff J, et al. Preferred reporting items for systematic reviews and meta-analyses: the PRISMA statement. PLoS Med. 2009. https://doi.org/10.1371/journal.pmed.1000097.
Geiger J, Markl M, Jung B, et al. 4D-MR flow analysis in patients after repair for tetralogy of Fallot. Eur Radiol. 2011;21(8):1651–7. https://doi.org/10.1007/s00330-011-2108-4.
François CJ, Srinivasan S, Schiebler ML, et al. 4D cardiovascular magnetic resonance velocity mapping of alterations of right heart flow patterns and main pulmonary artery hemodynamics in tetralogy of Fallot. J Cardiovasc Magn Reson. 2012;14(1):1–12. https://doi.org/10.1186/1532-429X-14-16.
Tariq U, Hsiao A, Alley M, Zhang T, Lustig M, Vasanawala SS. Venous and arterial flow quantification are equally accurate and precise with parallel imaging compressed sensing 4D phase contrast MRI. J Magn Reson Imaging. 2013;37(6):1419–26. https://doi.org/10.1002/jmri.23936.
Gabbour M, Rigsby C, Markl M, et al. Comparison of 4D flow and 2D PC MRI blood flow quantification in children and young adults with congenital heart disease. J Cardiovasc Magn Reson. 2013;15(S1):E90. https://doi.org/10.1186/1532-429X-15-S1-E90.
Giese D, Wong J, Greil GF, Buehrer M, Schaeffter T, Kozerke S. Towards highly accelerated Cartesian time-resolved 3D flow cardiovascular magnetic resonance in the clinical setting. J Cardiovasc Magn Reson. 2014;16:42. https://doi.org/10.1186/1532-429X-16-42.
Driessen MMP, Schings MA, Sieswerda GT, et al. Tricuspid flow and regurgitation in congenital heart disease and pulmonary hypertension: comparison of 4D flow cardiovascular magnetic resonance and echocardiography. J Cardiovasc Magn Reson. 2018;20(1):5. https://doi.org/10.1186/s12968-017-0426-7.
Jacobs KG, Chan FP, Cheng JY, Vasanawala SS, Maskatia SA. 4D flow vs 2D cardiac MRI for the evaluation of pulmonary regurgitation and ventricular volume in repaired tetralogy of Fallot: a retrospective case control study. Int J Cardiovasc Imaging. 2020. https://doi.org/10.1007/s10554-019-01751-1.
Isorni MA, Martins D, Ben Moussa N, et al. 4D flow MRI versus conventional 2D for measuring pulmonary flow after tetralogy of Fallot repair. Int J Cardiol. 2019. https://doi.org/10.1016/j.ijcard.2019.10.030.
van der Hulst AE, Westenberg JJM, Kroft LJM, et al. Tetralogy of fallot: 3D velocity-encoded MR imaging for evaluation of right ventricular valve flow and diastolic function in patients after correction. Radiology. 2010;256(3):724–34. https://doi.org/10.1148/radiol.10092269.
Nordmeyer S, Riesenkampff E, Messroghli D, et al. Four-dimensional velocity-encoded magnetic resonance imaging improves blood flow quantification in patients with complex accelerated flow. J Magn Reson Imaging. 2013;37(1):208–16. https://doi.org/10.1002/jmri.23793.
Hsiao A, Tariq U, Alley MT, Lustig M, Vasanawala SS. Inlet and outlet valve flow and regurgitant volume may be directly and reliably quantified with accelerated, volumetric phase-contrast MRI. J Magn Reson Imaging. 2015;41(2):376–85. https://doi.org/10.1002/jmri.24578.
Hanneman K, Kino A, Cheng JY, Alley MT, Vasanawala SS. Assessment of the precision and reproducibility of ventricular volume, function, and mass measurements with ferumoxytol-enhanced 4D flow MRI. J Magn Reson Imaging. 2016;44(2):383–92. https://doi.org/10.1002/jmri.25180.
Chelu RG, Wanambiro KW, Hsiao A, et al. Cloud-processed 4D CMR flow imaging for pulmonary flow quantification. Eur J Radiol. 2016;85(10):1849–56. https://doi.org/10.1016/j.ejrad.2016.07.018.
Fredriksson A, Trzebiatowska-Krzynska A, Dyverfeldt P, Engvall J, Ebbers T, Carlhäll CJ. Turbulent kinetic energy in the right ventricle: potential MR marker for risk stratification of adults with repaired tetralogy of Fallot. J Magn Reson Imaging. 2017. https://doi.org/10.1002/jmri.25830.
Sjöberg P, Töger J, Hedström E, et al. Altered biventricular hemodynamic forces in patients with repaired tetralogy of fallot and right ventricular volume overload because of pulmonary regurgitation. Am J Physiol Heart Circ Physiol. 2018;315(6):H1691–702. https://doi.org/10.1152/ajpheart.00330.2018.
Sjöberg P, Bidhult S, Bock J, et al. Disturbed left and right ventricular kinetic energy in patients with repaired tetralogy of Fallot: pathophysiological insights using 4D-flow MRI. Eur Radiol. 2018;28(10):4066–76. https://doi.org/10.1007/s00330-018-5385-3.
Robinson JD, Rose MJ, Joh M, et al. 4-D flow magnetic-resonance-imaging-derived energetic biomarkers are abnormal in children with repaired tetralogy of Fallot and associated with disease severity. Pediatr Radiol. 2019;49(3):308–17. https://doi.org/10.1007/s00247-018-4312-8.
Schäfer M, Barker AJ, Morgan GJ, et al. Increased systolic vorticity in the left ventricular outflow tract is associated with abnormal aortic flow formations in tetralogy of Fallot. Int J Cardiovasc Imaging. 2020. https://doi.org/10.1007/s10554-019-01764-w.
Jeong D, Anagnostopoulos PV, Roldan-Alzate A, et al. Ventricular kinetic energy may provide a novel noninvasive way to assess ventricular performance in patients with repaired tetralogy of Fallot. J Thorac Cardiovasc Surg. 2015;149(5):1339–47. https://doi.org/10.1016/j.jtcvs.2014.11.085.
Hussaini SF, Rutkowski DR, Roldán-Alzate A, François CJ. Left and right ventricular kinetic energy using time-resolved versus time-average ventricular volumes. J Magn Reson Imaging. 2017;45(3):821–8. https://doi.org/10.1002/jmri.25416.
Lawley CM, Broadhouse KM, Callaghan FM, Winlaw DS, Figtree GA, Grieve SM. 4D flow magnetic resonance imaging: role in pediatric congenital heart disease. Asian Cardiovasc Thorac Ann. 2018. https://doi.org/10.1177/0218492317694248.
Davlouros PA, Kilner PJ, Hornung TS, et al. Right ventricular function in adults with repaired tetralogy of Fallot assessed with cardiovascular magnetic resonance imaging: detrimental role of right ventricular outflow aneurysms or akinesia and adverse right-to-left ventricular interaction. J Am Coll Cardiol. 2002;40(11):2044–52. https://doi.org/10.1016/S0735-1097(02)02566-4.
Sotelo J, Bächler P, Urbina J, et al. Quantification of pulmonary regurgitation in patients with repaired tetralogy of Fallot by 2D phase-contrast MRI: differences between the standard method of velocity averaging and a pixel-wise analysis. JRSM Cardiovasc Dis. 2017. https://doi.org/10.1177/2048004017731986.
Wald RM, Redington AN, Pereira A, et al. Refining the assessment of pulmonary regurgitation in adults after tetralogy of Fallot repair: should we be measuring regurgitant fraction or regurgitant volume? Eur Heart J. 2009. https://doi.org/10.1093/eurheartj/ehn595.
Hsiao A, Lustig M, Alley MT, Murphy MJ, Vasanawala SS. Evaluation of valvular insufficiency and shunts with parallel-imaging compressed-sensing 4D phase-contrast MR imaging with stereoscopic 3D velocity-fusion volume-rendered visualization. Radiology. 2012;265(1):87–95. https://doi.org/10.1148/radiol.12120055.
Chelu RG, van den Bosch AE, van Kranenburg M, et al. Qualitative grading of aortic regurgitation: a pilot study comparing CMR 4D flow and echocardiography. Int J Cardiovasc Imaging. 2016;32(2):301–7. https://doi.org/10.1007/s10554-015-0779-7.
Mauger C, Gilbert K, Lee AM, et al. Right ventricular shape and function: cardiovascular magnetic resonance reference morphology and biventricular risk factor morphometrics in UK Biobank. J Cardiovasc Magn Reson. 2019;21(1):1–13. https://doi.org/10.1186/s12968-019-0551-6.
Roes SD, Hammer S, van der Geest RJ, et al. Flow assessment through four heart valves simultaneously using 3-dimensional 3-directional velocity-encoded magnetic resonance imaging with retrospective valve tracking in healthy volunteers and patients with valvular regurgitation. Invest Radiol. 2009;44(10):669–75. https://doi.org/10.1097/RLI.0b013e3181ae99b5.
Westenberg JJM, Roes SD, Ajmone Marsan N, et al. Mitral valve and tricuspid valve blood flow: accurate quantification with 3D velocity-encoded MR imaging with retrospective valve tracking. Radiology. 2008;249(3):792–800. https://doi.org/10.1148/radiol.2492080146.
Brix L, Ringgaard S, Rasmusson A, Sørensen TS, Kim WY. Three dimensional three component whole heart cardiovascular magnetic resonance velocity mapping: comparison of flow measurements from 3D and 2D acquisitions. J Cardiovasc Magn Reson. 2009. https://doi.org/10.1186/1532-429X-11-3.
Arvidsson PM, Toger J, Heiberg E, Carlsson M, Arheden H. Quantification of left and right atrial kinetic energy using four-dimensional intracardiac magnetic resonance imaging flow measurements. J Appl Physiol. 2013;114(10):1472–81. https://doi.org/10.1152/japplphysiol.00932.2012.
Gharib M, Rambod E, Kheradvar A, Sahn DJ, Dabiri JO. Optimal vortex formation as an index of cardiac health. Proc Natl Acad Sci. 2006;103(16):6305–8. https://doi.org/10.1073/pnas.0600520103.
Uribe Arancibia S, Beerbaum P, Rasmusson A, Sorensen T, Razavi R, Schaeffter T. 4D flow of the whole heart and great vessels using a real time self respiratory gating technique: a validation study. Proc Int Soc Mag Reson Med. 2009;17:319. https://doi.org/10.1186/1532-429X-11-S1-P3.
Markl M, Frydrychowicz A, Kozerke S, Hope M, Wieben O. 4D flow MRI. J Magn Reson Imaging. 2012;36(5):1015–36. https://doi.org/10.1002/jmri.23632.
Jeong D, Roldán-alzate A, Francois CJ. Right ventricular kinetic energy: 4D flow MRI analysis of healthy volunteers and repaired tetralogy of Fallot. J Cardiovasc Magn Reson. 2014;16(Suppl 1):O46. https://doi.org/10.1186/1532-429X-16-S1-O46.
Giese D, Greil G, Schaeffter T, Kozerke S. Highly accelerated 4D MR flow measurements in congenital heart disease. Proc Int Soc Magn Reson Med. 2012;20:665.
Griselli M, McGuirk SP, Winlaw DS, et al. The influence of pulmonary artery morphology on the results of operations for major aortopulmonary collateral arteries and complex congenital heart defects. J Thorac Cardiovasc Surg. 2004. https://doi.org/10.1016/j.jtcvs.2003.08.052.
Hanneman K, Sivagnanam M, Nguyen ET, et al. Magnetic resonance assessment of pulmonary (QP) to systemic (QS) flows using 4D phase-contrast imaging: pilot study comparison with standard through-plane 2D phase-contrast imaging. Acad Radiol. 2014. https://doi.org/10.1016/j.acra.2014.04.012.
Wentland AL, Grist TM, Wieben O. Repeatability and internal consistency of abdominal 2D and 4D phase contrast MR flow measurements. Acad Radiol. 2013. https://doi.org/10.1016/j.acra.2012.12.019.
Rutz T, Meierhofer C, Naumann S, et al. Comparison of MR flow quantification in peripheral and main pulmonary arteries in patients after right ventricular outflow tract surgery: a retrospective study. J Magn Reson Imaging. 2017;46(6):1839–45. https://doi.org/10.1002/jmri.25701.
Lee S, Kim YJ, Jung JW, et al. Evaluation of flow pattern in the ascending aorta in patients with repaired tetralogy of fallot using four-dimensional flow magnetic resonance imaging. Korean J Radiol. 2019;20(9):1334–41. https://doi.org/10.3348/kjr.2019.0096.
Baumgartner H, Taylor J. The 2010 version of the ESC guidelines for the management of grown-up adult congenital heart disease are discussed by guidelines task force chairman H Baumgartner. Eur Heart J. 2010;31(23):2825–6. https://doi.org/10.1093/eurheartj/ehq309.
Cramer JW, Ginde S, Hill GD, et al. Tricuspid repair at pulmonary valve replacement does not alter outcomes in tetralogy of fallot. Ann Thorac Surg. 2015;99(3):899–904. https://doi.org/10.1016/j.athoracsur.2014.09.086.
Reiter G, Reiter U, Kovacs G, et al. Magnetic resonance-derived 3-dimensional blood flow patterns in the main pulmonary artery as a marker of pulmonary hypertension and a measure of elevated mean pulmonary arterial pressure. Circ Cardiovasc Imaging. 2008;1(1):23–30. https://doi.org/10.1161/CIRCIMAGING.108.780247.
Calkoen EE, Elbaz MSM, Westenberg JJM, et al. Altered left ventricular vortex ring formation by 4-dimensional flow magnetic resonance imaging after repair of atrioventricular septal defects. J Thorac Cardiovasc Surg. 2015;150(5):1233–40. https://doi.org/10.1016/j.jtcvs.2015.07.048.
Frydrychowicz A, Berger A, Russe MF, et al. Time-resolved magnetic resonance angiography and flow-sensitive 4-dimensional magnetic resonance imaging at 3 Tesla for blood flow and wall shear stress analysis. J Thorac Cardiovasc Surg. 2008;136(2):400–7. https://doi.org/10.1016/j.jtcvs.2008.02.062.
Pedrizzetti G, La Canna G, Alfieri O, Tonti G. The vortex—an early predictor of cardiovascular outcome? Nat Rev Cardiol. 2014;11(9):545–53. https://doi.org/10.1038/nrcardio.2014.75.
Elbaz MSM, Calkoen EE, Westenberg JJM, Lelieveldt BPF, Roest AAW, van der Geest RJ. Vortex flow during early and late left ventricular filling in normal subjects: quantitative characterization using retrospectively-gated 4D flow cardiovascular magnetic resonance and three-dimensional vortex core analysis. J Cardiovasc Magn Reson. 2014;16:78. https://doi.org/10.1186/s12968-014-0078-9.
Sachdev MS, Bhagyavathy A, Varghese R, Coelho R, Kumar RS. Right ventricular diastolic function after repair of tetralogy of fallot. Pediatr Cardiol. 2006;27(2):250–5. https://doi.org/10.1007/s00246-005-1186-y.
Hope MD, Hope TA, Crook SES, et al. 4D flow CMR in assessment of valve-related ascending aortic disease. JACC Cardiovasc Imaging. 2011;4(7):781–7. https://doi.org/10.1016/j.jcmg.2011.05.004.
Ma LE, Markl M, Chow K, Vali A, Wu C, Schnell S. Efficient triple-VENC phase-contrast MRI for improved velocity dynamic range. Magn Reson Med. 2020. https://doi.org/10.1002/mrm.27943.
Ferdian E, Suinesiaputra A, Dubowitz DJ, et al. 4DFlowNet: super-resolution 4D flow MRI using deep learning and computational fluid dynamics. Front Phys. 2020. https://doi.org/10.3389/fphy.2020.00138.
Berhane H, Scott M, Elbaz M, et al. Fully automated 3D aortic segmentation of 4D flow MRI for hemodynamic analysis using deep learning. Magn Reson Med. 2020. https://doi.org/10.1002/mrm.28257.
This work was funded by the New Zealand Heart Foundation. AY acknowledges the support of the Health Research Council of New Zealand grants 17/234 and 17/608 and NIH R01HL121754 from the National Heart, Lung, and Blood Institute.
Ethics approval and consent to participate
Consent for publication
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Elsayed, A., Gilbert, K., Scadeng, M. et al. Four-dimensional flow cardiovascular magnetic resonance in tetralogy of Fallot: a systematic review. J Cardiovasc Magn Reson 23, 59 (2021). https://doi.org/10.1186/s12968-021-00745-0